• Run 2 Upgrades to the CMS Level-1 Calorimeter Trigger(1511.05855)

Nov. 18, 2015 hep-ex, physics.ins-det
The CMS Level-1 calorimeter trigger is being upgraded in two stages to maintain performance as the LHC increases pile-up and instantaneous luminosity in its second run. In the first stage, improved algorithms including event-by-event pile-up corrections are used. New algorithms for heavy ion running have also been developed. In the second stage, higher granularity inputs and a time-multiplexed approach allow for improved position and energy resolution. Data processing in both stages of the upgrade is performed with new, Xilinx Virtex-7 based AMC cards.
• A joint measurement is presented of the branching fractions $B^0_s\to\mu^+\mu^-$ and $B^0\to\mu^+\mu^-$ in proton-proton collisions at the LHC by the CMS and LHCb experiments. The data samples were collected in 2011 at a centre-of-mass energy of 7 TeV, and in 2012 at 8 TeV. The combined analysis produces the first observation of the $B^0_s\to\mu^+\mu^-$ decay, with a statistical significance exceeding six standard deviations, and the best measurement of its branching fraction so far. Furthermore, evidence for the $B^0\to\mu^+\mu^-$ decay is obtained with a statistical significance of three standard deviations. The branching fraction measurements are statistically compatible with SM predictions and impose stringent constraints on several theories beyond the SM.
• The Online Calibration, Operation, and Performance of the CMS Pixel Detector(1101.1989)

Jan. 10, 2011 hep-ex, physics.ins-det
The CMS pixel detector consists of approximately 66 million silicon pixels whose analog signals are read out by 15,840 programmable Readout Chips. With the recent startup of the LHC, the detector is now collecting data used for precise vertexing and track-finding. In preparation for data taking, the detector's Readout Chips and their supporting readout and control electronics were calibrated. The calibration that has taken place since the detector's installation in the summer of 2008 will be described. These calibrations focused on the optimization of the Readout Chips' thresholds and analog response. The operation of the detector during the early running of the LHC will also be discussed. The calibrations that are performed on a regular basis and a mechanism to handle the readout of large beam background events will be described.