• A new pixel detector for the CMS experiment was built in order to cope with the instantaneous luminosities anticipated for the Phase~I Upgrade of the LHC. The new CMS pixel detector provides four-hit tracking with a reduced material budget as well as new cooling and powering schemes. A new front-end readout chip mitigates buffering and bandwidth limitations, and allows operation at low comparator thresholds. In this paper, comprehensive test beam studies are presented, which have been conducted to verify the design and to quantify the performance of the new detector assemblies in terms of tracking efficiency and spatial resolution. Under optimal conditions, the tracking efficiency is $99.95\pm0.05\,\%$, while the intrinsic spatial resolutions are $4.80\pm0.25\,\mu \mathrm{m}$ and $7.99\pm0.21\,\mu \mathrm{m}$ along the $100\,\mu \mathrm{m}$ and $150\,\mu \mathrm{m}$ pixel pitch, respectively. The findings are compared to a detailed Monte Carlo simulation of the pixel detector and good agreement is found.
• The degradation of signal in silicon sensors is studied under conditions expected at the CERN High-Luminosity LHC. 200 $\mu$m thick n-type silicon sensors are irradiated with protons of different energies to fluences of up to $3 \cdot 10^{15}$ neq/cm$^2$. Pulsed red laser light with a wavelength of 672 nm is used to generate electron-hole pairs in the sensors. The induced signals are used to determine the charge collection efficiencies separately for electrons and holes drifting through the sensor. The effective trapping rates are extracted by comparing the results to simulation. The electric field is simulated using Synopsys device simulation assuming two effective defects. The generation and drift of charge carriers are simulated in an independent simulation based on PixelAV. The effective trapping rates are determined from the measured charge collection efficiencies and the simulated and measured time-resolved current pulses are compared. The effective trapping rates determined for both electrons and holes are about 50% smaller than those obtained using standard extrapolations of studies at low fluences and suggests an improved tracker performance over initial expectations.
• ### Radiation hardness of CMS pixel barrel modules(1001.0666)

Jan. 5, 2010 hep-ex, physics.ins-det
• ### Mechanical Design and Material Budget of the CMS Barrel Pixel Detector(0904.4761)

April 30, 2009 hep-ex, physics.ins-det
The Compact Muon Solenoid experiment at the Large Hadron Collider at CERN includes a silicon pixel detector as its innermost component. Its main task is the precise reconstruction of charged particles close to the primary interaction vertex. This paper gives an overview of the mechanical requirements and design choices for the barrel pixel detector. The distribution of material in the detector as well as its description in the Monte Carlo simulation are discussed in detail.
• ### Signal height in silicon pixel detectors irradiated with pions and protons(0901.3422)

Jan. 22, 2009 hep-ex, physics.ins-det
Pixel detectors are used in the innermost part of multi purpose experiments at the Large Hadron Collider (LHC) and are therefore exposed to the highest fluences of ionising radiation, which in this part of the detectors consists mainly of charged pions. The radiation hardness of the detectors has thoroughly been tested up to the fluences expected at the LHC. In case of an LHC upgrade the fluence will be much higher and it is not yet clear up to which radii the present pixel technology can be used. In order to establish such a limit, pixel sensors of the size of one CMS pixel readout chip (PSI46V2.1) have been bump bonded and irradiated with positive pions up to 6E14 Neq/cm^2 at PSI and with protons up to 5E15 Neq/cm^2. The sensors were taken from production wafers of the CMS barrel pixel detector. They use n-type DOFZ material with a resistance of about 3.7kOhm cm and an n-side read out. As the performance of silicon sensors is limited by trapping, the response to a Sr-90 source was investigated. The highly energetic beta-particles represent a good approximation to minimum ionising particles. The bias dependence of the signal for a wide range of fluences will be presented.
• ### Building CMS Pixel Barrel Detectur Modules(physics/0702181)

Feb. 21, 2007 physics.ins-det
For the barrel part of the CMS pixel tracker about 800 silicon pixel detector modules are required. The modules are bump bonded, assembled and tested at the Paul Scherrer Institute. This article describes the experience acquired during the assembly of the first ~200 modules.
• ### CMS Barrel Pixel Detector Overview(physics/0702182)

Feb. 21, 2007 physics.ins-det
The pixel detector is the innermost tracking device of the CMS experiment at the LHC. It is built from two independent sub devices, the pixel barrel and the end disks. The barrel consists of three concentric layers around the beam pipe with mean radii of 4.4, 7.3 and 10.2 cm. There are two end disks on each side of the interaction point at 34.5 cm and 46.5 cm. This article gives an overview of the pixel barrel detector, its mechanical support structure, electronics components, services and its expected performance.
• ### Design and Performance of the CMS Pixel Detector Readout Chip(physics/0511166)

Feb. 24, 2006 physics.ins-det
The readout chip for the CMS pixel detector has to deal with an enormous data rate. On-chip zero suppression is inevitable and hit data must be buffered locally during the latency of the first level trigger. Dead-time must be kept at a minimum. It is dominated by contributions coming from the readout. To keep it low an analog readout scheme has been adopted where pixel addresses are analog coded. We present the architecture of the final CMS pixel detector readout chip with special emphasis on the analog readout chain. Measurements of its performance are discussed.
• ### Qualification Procedures of the CMS Pixel Barrel Modules(physics/0510165)

Oct. 18, 2005 physics.ins-det
The CMS pixel barrel system will consist of three layers built of about 800 modules. One module contains 66560 readout channels and the full pixel barrel system about 48 million channels. It is mandatory to test each channel for functionality, noise level, trimming mechanism, and bump bonding quality. Different methods to determine the bump bonding yield with electrical measurements have been developed. Measurements of several operational parameters are also included in the qualification procedure. Among them are pixel noise, gains and pedestals. Test and qualification procedures of the pixel barrel modules are described and some results are presented.