
Central frequency is a key parameter of storage rings. This paper presents the measurement of central frequency of the HLSII storage ring using the sextupole modulation method. Firstly, the basis of central frequency measurement of the electron storage ring is briefly introduced. Then, the error sources and the optimized measurement method for the HLSII storage ring are discussed. The workflow of the selfcompiled Matlab script used in central frequency measurement is also described. In the end, the results achieved by using two methods to crosscheck each other are shown. The measured value of the central frequency demonstrates that the real circumference of the HLSII storage ring agrees well with the designed value.

A logarithm processing algorithm to measure beam transverse size and position is proposed and preliminary experimental results in Hefei Light Source II (HLS II) are given. The algorithm is based on only 4 successive channels of 16 anode channels of multianode photomultiplier tube (MAPMT) R5900U00L16 which has typical rise time of 0.6 ns and effective area of 0.8x16 mm for a single anode channel. In the paper, we firstly elaborate the simulation results of the algorithm with and without channel inconsistency. Then we calibrate the channel inconsistency and verify the algorithm using general current signal processor Libera Photon in lowspeed scheme. Finally we get turnbyturn beam size and position and calculate the vertical tune in highspeed scheme. The experimental results show that measured values fit well with simulation results after channel differences are calibrated and the fractional part of the tune in vertical direction is 0.3628 which is very close to the nominal value 0.3621.