A prototype of a luminometer, designed for a future e+e- collider detector,
and consisting at present of a four-plane module, was tested in the CERN PS
accelerator T9 beam. The objective of this beam test was to demonstrate a
multi-plane tungsten/silicon operation, to study the development of the
electromagnetic shower and to compare it with MC simulations. The Moli\`ere
radius has been determined to be 24.0 +/- 0.6 (stat.) +/- 1.5 (syst.) mm using
a parametrization of the shower shape. Very good agreement was found between
data and a detailed Geant4 simulation.
It is the hybrid pixel detector technology which brought to the X-ray imaging
a low noise level at a high spatial resolution, thanks to the single photon
counting. However, silicon as the most widespread detector material is
marginally sensitive to photons with energy more than 30 keV. That's why the
high-Z alternatives to silicon such as gallium arsenide and cadmium telluride
are increasingly attracting attention of the community for the development of
X-ray imaging systems in recent years. We present in this work the results of
our investigations of the Timepix detectors bump-bonded with sensors made of
gallium arsenide compensated by chromium (GaAs:Cr). The properties which are
mostly important from the practical point of view: IV characteristics, charge
transport characteristics, operational stability, homogeneity, temperature
dependence as well as energy and spatial resolution are considered.
Applicability of these detectors for spectroscopic X-ray imaging is discussed.
The response of the Timepix detector to neon ions with kinetic energy 77 and
158.4 MeV has been studied at the cyclotron U-400M of the JINR Flerov
Laboratory of Nuclear Reaction. Sensors produced from gallium arsenide
compensated by chromium and from silicon are used for these measurements. While
in Timepix detector with Si sensor the well-known so-called "volcano effect"
observed, in Timepix detector with GaAs:Cr sensor such effect was completely
In the work the behavior of the Timepix detector with GaAs:Cr sensor under
irradiation with heavy ions is described in comparison with the detector based
on Si sensor. Also the possible reason for absence of "volcano" effect in
GaAs:Cr detector is proposed.
Detector-plane prototypes of the very forward calorimetry of a future
detector at an e+e- collider have been built and their performance was measured
in an electron beam. The detector plane comprises silicon or GaAs pad sensors,
dedicated front-end and ADC ASICs, and an FPGA for data concentration.
Measurements of the signal-to-noise ratio and the response as a function of the
position of the sensor are presented. A deconvolution method is successfully
applied, and a comparison of the measured shower shape as a function of the
absorber depth with a Monte-Carlo simulation is given.
The MARS scanner is designed for the x-ray spectroscopic study of samples
with the aid of computer tomography methods. Computer tomography allows the
reconstruction of slices of an investigated sample using a set of shadow
projections obtained for different angles. Projections in the MARS scanner are
produced using a cone x-ray beam geometry. Correct reconstruction in this
scheme requires precise knowledge of several geometrical parameters of a
tomograph, such as displacement of a rotation axis, x-ray source position with
respect to a camera, and camera inclinations. Use of inaccurate parameters
leads to a poor sample reconstruction. Non-ideal positioning of camera, x-ray
source and cylindrical rotating frame (gantry) itself on which these parts are
located, leads to the need for tomograph alignment. In this note we describe
the alignment procedure that was used to get different geometrical corrections
for the reconstruction. Also, several different estimations of the final
spatial resolution for reconstructed images are presented.