• ### Preliminary results of Resistive Plate Chambers operated with eco-friendly gas mixtures for application in the CMS experiment(1605.08172)

May 26, 2016 hep-ex, physics.ins-det
The operations of Resistive Plate Chambers in LHC experiments require Fluorine based (F-based) gases for optimal performance. Recent European regulations demand the use of environmentally unfriendly F-based gases to be limited or banned. In view of the CMS experiment upgrade, several tests are ongoing to measure the performance of the detector with these new ecological gas mixtures, in terms of efficiency, streamer probability, induced charge and time resolution. Prototype chambers with readout pads and with the standard CMS electronic setup are under test. In this paper preliminary results on performance of RPCs operated with a potential eco-friendly gas candidate 1,3,3,3-Tetrafluoropropene, commercially known as HFO-1234ze, with CO2 and CF3I based gas mixtures are presented and discussed for the possible application in the CMS experiment.
• ### Eco-friendly gas mixtures for Resistive Plate Chambers based on Tetrafluoropropene and Helium(1605.01691)

May 4, 2016 hep-ex, physics.ins-det
Due to the recent restrictions deriving from the application of the Kyoto protocol, the main components of the gas mixtures presently used in the Resistive Plate Chambers systems of the LHC experiments will be most probably phased out of production in the coming years. Identifying possible replacements with the adequate characteristics requires an intense R&D, which was recently started, also in collaborations across the various experiments. Possible candidates have been proposed and are thoroughly investigated. Some tests on one of the most promising candidate - HFO-1234ze, an allotropic form of tetrafluoropropane- have already been reported. Here an innovative approach, based on the use of Helium, to solve the problems related to the too elevate operating voltage of HFO-1234ze based gas mixtures, is discussed and the relative first results are shown.
• ### Characterization of the GEM foil materials(1512.08621)

Dec. 29, 2015 physics.ins-det
Systematic studies on the GEM foil material are performed to measure the moisture diffusion rate and saturation level. These studies are important because the presence of this compound inside the detector's foil can possibly change its mechanical and electrical properties and, in such a way, the detector performance can be affected. To understand this phenomenon, a model is developed with COMSOL Multhiphysics v. 4.3, which described the adsorption and diffusion within the geometry of GEM foil, the concentration profiles and the time required to saturate the foil. The COMSOL model is verified by experimental observations on a GEM foil sample. This note will describe the model and its experimental verification results.
• ### Fiber Bragg Grating sensors for deformation monitoring of GEM foils in HEP detectors(1512.08629)

Dec. 29, 2015 physics.ins-det
Fiber Bragg Grating (FBG) sensors have been so far mainly used in high energy physics (HEP) as high precision positioning and re-positioning sensors and as low cost, easy to mount, radiation hard and low space- consuming temperature and humidity devices. FBGs are also commonly used for very precise strain measurements. In this work we present a novel use of FBGs as flatness and mechanical tensioning sensors applied to the wide Gas Electron Multiplier (GEM) foils of the GE1/1 chambers of the Compact Muon Solenoid (CMS) experiment at Large Hadron Collider (LHC) of CERN. A network of FBG sensors has been used to determine the optimal mechanical tension applied and to characterize the mechanical stress applied to the foils. The preliminary results of the test performed on a full size GE1/1 final prototype and possible future developments will be discussed.
• ### Fiber Bragg Grating (FBG) sensors as flatness and mechanical stretching sensors(1512.08481)

Dec. 28, 2015 physics.ins-det
A novel approach which uses Fibre Bragg Grating (FBG) sensors has been utilised to assess and monitor the flatness of Gaseous Electron Multipliers (GEM) foils. The setup layout and preliminary results are presented.
• ### A novel application of Fiber Bragg Grating (FBG) sensors in MPGD(1512.08529)

Dec. 28, 2015 physics.ins-det
We present a novel application of Fiber Bragg Grating (FBG) sensors in the construction and characterisation of Micro Pattern Gaseous Detector (MPGD), with particular attention to the realisation of the largest triple (Gas electron Multiplier) GEM chambers so far operated, the GE1/1 chambers of the CMS experiment at LHC. The GE1/1 CMS project consists of 144 GEM chambers of about 0.5 m2 active area each, employing three GEM foils per chamber, to be installed in the forward region of the CMS endcap during the long shutdown of LHC in 2108-2019. The large active area of each GE1/1 chamber consists of GEM foils that are mechanically stretched in order to secure their flatness and the consequent uniform performance of the GE1/1 chamber across its whole active surface. So far FBGs have been used in high energy physics mainly as high precision positioning and re-positioning sensors and as low cost, easy to mount, low space consuming temperature sensors. FBGs are also commonly used for very precise strain measurements in material studies. In this work we present a novel use of FBGs as flatness and mechanical tensioning sensors applied to the wide GEM foils of the GE1/1 chambers. A network of FBG sensors have been used to determine the optimal mechanical tension applied and to characterise the mechanical tension that should be applied to the foils. We discuss the results of the test done on a full-sized GE1/1 final prototype, the studies done to fully characterise the GEM material, how this information was used to define a standard assembly procedure and possible future developments.
• ### A study of HFO-1234ze (1,3,3,3-Tetrafluoropropene) as an eco-friendly replacement in RPC detectors(1505.01648)

May 7, 2015 hep-ex, physics.ins-det
The operations of Resistive Plate Chambers in LHC experiments require F-based gases for optimal performance. Recent regulations demand the use of environmentally unfriendly F-based gases to be limited or banned. This report shows results of studies on performance of RPCs operated with a potential eco-friendly gas candidate 1,3,3,3-Tetrafluoropropene, commercially known as HFO-1234ze.
• ### Study of gas contaminants and interaction with materials in RPC closed loop system(1302.5225)

March 22, 2013 hep-ex, physics.ins-det
Resistive Plate Counters (RPC) detectors at the Large Hadron Collider (LHC) experiments use gas recirculation systems to cope with large gas mixture volumes and costs. In this paper a long-term systematic study about gas purifiers, gas contaminants and detector performance is discussed. The study aims at measuring the lifetime of purifiers with new and used cartridge material along with contaminants release in the gas system. During the data-taking the response of several RPC double-gap detectors was monitored in order to characterize the correlation between dark currents, filter status and gas contaminants.
• ### A study of gas contaminants and interaction with materials in RPC closed loop systems(1210.1819)

Oct. 5, 2012 hep-ex, physics.ins-det
Resistive Plate Counters (RPC) detectors at the Large Hadron Collider (LHC) experiments use gas recirculation systems to cope with large gas mixture volumes and costs. In this paper a long-term systematic study about gas purifiers, gas contaminants and detector performance is discussed. The study aims at measuring the lifetime of purifiers with unused and used cartridge material along with contaminants release in the gas system. During the data-taking the response of several RPC double-gap detectors was monitored in order to characterize the correlation between dark currents, filter status and gas contaminants.
• ### Performance of the Gas Gain Monitoring system of the CMS RPC muon detector and effective working point fine tuning(1209.3893)

Sept. 18, 2012 hep-ex, physics.ins-det
The Gas Gain Monitoring (GGM) system of the Resistive Plate Chamber (RPC) muon detector in the Compact Muon Solenoid (CMS) experiment provides fast and accurate determination of the stability in the working point conditions due to gas mixture changes in the closed loop recirculation system. In 2011 the GGM began to operate using a feedback algorithm to control the applied voltage, in order to keep the GGM response insensitive to environmental temperature and atmospheric pressure variations. Recent results are presented on the feedback method used and on alternative algorithms.
• ### A new approach in modeling the response of RPC detectors(1012.5508)

Dec. 26, 2010 hep-ex, physics.ins-det
The response of RPC detectors is highly sensitive to environmental variables. A novel approach is presented to model the response of RPC detectors in a variety of experimental conditions. The algorithm, based on Artificial Neural Networks, has been developed and tested on the CMS RPC gas gain monitoring system during commissioning.
• ### Study of gas purifiers for the CMS RPC detector(1012.5511)

Dec. 26, 2010 hep-ex, physics.ins-det
The CMS RPC muon detector utilizes a gas recirculation system called closed loop (CL) to cope with large gas mixture volumes and costs. A systematic study of CL gas purifiers has been carried out over 400 days between July 2008 and August 2009 at CERN in a low-radiation test area, with the use of RPC chambers with currents monitoring, and gas analysis sampling points. The study aimed to fully clarify the presence of pollutants, the chemistry of purifiers used in the CL, and the regeneration procedure. Preliminary results on contaminants release and purifier characterization are reported.
• ### Construction of the first full-size GEM-based prototype for the CMS high-$\eta$ muon system(1012.1524)

Dec. 9, 2010 hep-ex, physics.ins-det
In view of a possible extension of the forward CMS muon detector system and future LHC luminosity upgrades, Micro-Pattern Gas Detectors (MPGDs) are an appealing technology. They can simultaneously provide precision tracking and fast trigger information, as well as sufficiently fine segmentation to cope with high particle rates in the high-eta region at LHC and its future upgrades. We report on the design and construction of a full-size prototype for the CMS endcap system, the largest Triple-GEM detector built to-date. We present details on the 3D modeling of the detector geometry, the implementation of the readout strips and electronics, and the detector assembly procedure.
• ### Sensitivity and environmental response of the CMS RPC gas gain monitoring system(0812.1710)

Dec. 9, 2008 physics.ins-det
Results from the gas gain monitoring (GGM) system for the muon detector using RPC in the CMS experiment at the LHC is presented. The system is designed to provide fast and accurate determination of any shift in the working point of the chambers due to gas mixture changes.
• ### Micrometric Position Monitoring Using Fiber Bragg Grating Sensors in Silicon Detectors(physics/0512255)

Dec. 28, 2005 physics.ins-det
We show R&D results including long term stability, resolution, radiation hardness and characterization of Fiber Grating sensors used to monitor structure deformation, repositioning and surveying of silicon detector in High Energy Physics.