• ### Design and commissioning of a 600 L Time Projection Chamber with Microbulk Micromegas(1804.02863)

April 9, 2018 physics.ins-det
We report the design, construction, and initial commissioning results of a large high pressure gaseous Time Projection Chamber (TPC) with Micromegas modules for charge readout. The detector vessel has an inner volume of about 600 L and an active volume of 270 L. At 10 bar operating pressure, the active volume contains about 20 kg of xenon gas and can image charged particle tracks. Drift electrons are collected by the charge readout plane, which accommodates a tessellation of seven Micromegas modules. Each of the Micromegas covers a square of 20 cm by 20 cm. A new type of Microbulk Micromegas is chosen for this application due to its good gain uniformity and low radioactive contamination. Initial commissioning results with 1 Micromegas module running with 1 bar argon and isobutane gas mixture and 5 bar xenon and trimethylamine (TMA) gas mixture are reported. We also recorded extended background tracks from cosmic ray events and highlighted the unique tracking feature of this gaseous TPC.
• ### PandaX-III: Searching for Neutrinoless Double Beta Decay with High Pressure $^{136}$Xe Gas Time Projection Chambers(1610.08883)

Oct. 28, 2016 hep-ex, nucl-ex, physics.ins-det
Searching for the Neutrinoless Double Beta Decay (NLDBD) is now regarded as the topmost promising technique to explore the nature of neutrinos after the discovery of neutrino masses in oscillation experiments. PandaX-III (Particle And Astrophysical Xenon Experiment III) will search for the NLDBD of $^{136}$Xe at the China Jin Ping underground Laboratory (CJPL). In the first phase of the experiment, a high pressure gas Time Projection Chamber (TPC) will contain 200 kg, 90% $^{136}$Xe enriched gas operated at 10 bar. Fine pitch micro-pattern gas detector (Microbulk Micromegas) will be used at both ends of the TPC for the charge readout with a cathode in the middle. Charge signals can be used to reconstruct tracks of NLDBD events and provide good energy and spatial resolution. The detector will be immersed in a large water tank to ensure $\sim$5 m of water shielding in all directions. The second phase, a ton-scale experiment, will consist of five TPCs in the same water tank, with improved energy resolution and better control over backgrounds.
• ### New pixelized Micromegas detector with low discharge rate for the COMPASS experiment(1111.3337)

Jan. 25, 2012 hep-ex, physics.ins-det
New Micromegas (Micro-mesh gaseous detectors) are being developed in view of the future physics projects planned by the COMPASS collaboration at CERN. Several major upgrades compared to present detectors are being studied: detectors standing five times higher luminosity with hadron beams, detection of beam particles (flux up to a few hundred of kHz/mm^{2}, 10 times larger than for the present Micromegas detectors) with pixelized read-out in the central part, light and integrated electronics, and improved robustness. Two solutions of reduction of discharge impact have been studied, with Micromegas detectors using resistive layers and using an additional GEM foil. Performance of such detectors has also been measured. A large size prototypes with nominal active area and pixelized read-out has been produced and installed at COMPASS in 2010. In 2011 prototypes featuring an additional GEM foil, as well as an resistive prototype, are installed at COMPASS and preliminary results from those detectors presented very good performance. We present here the project and report on its status, in particular the performance of large size prototypes with an additional GEM foil.
• ### New pixelized Micromegas detector for the COMPASS experiment(0909.5402)

Oct. 19, 2009 physics.ins-det
New Micromegas (Micro-mesh gaseous detectors) are being developed in view of the future physics projects planned by the COMPASS collaboration at CERN. Several major upgrades compared to present detectors are being studied: detectors standing five times higher luminosity with hadron beams, detection of beam particles (flux up to a few hundred of kHz/mm^2, 10 times larger than for the present detectors) with pixelized read-out in the central part, light and integrated electronics, and improved robustness. Studies were done with the present detectors moved in the beam, and two first pixelized prototypes are being tested with muon and hadron beams in real conditions at COMPASS. We present here this new project and report on two series of tests, with old detectors moved into the beam and with pixelized prototypes operated in real data taking condition with both muon and hadron beams.