• ### When "Optimal Filtering" Isn't(1611.07856)

The so-called "optimal filter" analysis of a microcalorimeter's x-ray pulses is statistically optimal only if all pulses have the same shape, regardless of energy. The shapes of pulses from a nonlinear detector can and do depend on the pulse energy, however. A pulse-fitting procedure that we call "tangent filtering" accounts for the energy dependence of the shape and should therefore achieve superior energy resolution. We take a geometric view of the pulse-fitting problem and give expressions to predict how much the energy resolution stands to benefit from such a procedure. We also demonstrate the method with a case study of K-line fluorescence from several 3d transition metals. The method improves the resolution from 4.9 eV to 4.2 eV at the Cu K$\alpha$ line (8.0keV).
• ### First application of superconducting transition-edge-sensor microcalorimeters to hadronic-atom x-ray spectroscopy(1608.05436)

High-resolution pionic-atom x-ray spectroscopy was performed with an x-ray spectrometer based on a 240-pixel array of superconducting transition-edge-sensor (TES) microcalorimeters at the piM1 beam line of the Paul Scherrer Institute. X-rays emitted by pionic carbon via the 4f->3d transition and the parallel 4d->3p transition were observed with a full-width-at-half-maximum energy resolution of 6.8 eV at 6.4 keV. Measured x-ray energies are consistent with calculated electromagnetic values which considered the strong-interaction effect assessed via the Seki-Masutani potential for the 3p energy level, and favor the electronic population of two filled 1s electrons in the K-shell. Absolute energy calibration with an uncertainty of 0.1 eV was demonstrated under a high-rate hadron beam condition of 1.45 MHz. This is the first application of a TES spectrometer to hadronic-atom x-ray spectroscopy and is an important milestone towards next-generation high-resolution kaonic-atom x-ray spectroscopy.
• ### The Practice of Pulse Processing(1511.03950)

Nov. 12, 2015 physics.ins-det
The analysis of data from x-ray microcalorimeters requires great care; their excellent intrinsic energy resolution cannot usually be achieved in practice without a statistically near-optimal pulse analysis and corrections for important systematic errors. We describe the essential parts of a pulse-analysis pipeline for data from x-ray microcalorimeters, including steps taken to reduce systematic gain variation and the unwelcome dependence of filtered pulse heights on the exact pulse-arrival time. We find these steps collectively to be essential tools for getting the best results from a microcalorimeter-based x-ray spectrometer.
• ### Development of holmium-163 electron-capture spectroscopy with transition-edge sensors(1510.03874)

Oct. 20, 2015 nucl-ex, physics.ins-det
Calorimetric decay energy spectroscopy of electron-capture-decaying isotopes is a promising method to achieve the sensitivity required for electron neutrino mass measurement. The very low total nuclear decay energy (QEC < 3 keV) and short half-life (4570 y) of 163Ho make it attractive for high-precision electron capture spectroscopy (ECS) near the kinematic endpoint, where the neutrino momentum goes to zero. In the ECS approach, an electron-capture-decaying isotope is embedded inside a microcalorimeter designed to capture and measure the energy of all the decay radiation except that of the escaping neutrino. We have developed a complete process for proton-irradiation-based isotope production, isolation, and purification of 163Ho. We have developed transition-edge sensors for this measurement and methods for incorporating 163Ho into high-resolution microcalorimeters, and have measured the electron-capture spectrum of 163Ho. We present our work in these areas and discuss the measured spectrum and its comparison to current theory.
• ### High-resolution kaonic-atom x-ray spectroscopy with transition-edge-sensor microcalorimeters(1401.8039)

Feb. 19, 2014 nucl-ex, physics.ins-det
We are preparing for an ultra-high resolution x-ray spectroscopy of kaonic atoms using an x-ray spectrometer based on an array of superconducting transition-edge-sensor microcalorimeters developed by NIST. The instrument has excellent energy resolutions of 2 - 3 eV (FWHM) at 6 keV and a large collecting area of about 20 mm^2. This will open new door to investigate kaon-nucleus strong interaction and provide new accurate charged-kaon mass value.
• ### SCUBA-2: The 10000 pixel bolometer camera on the James Clerk Maxwell Telescope(1301.3650)

Jan. 16, 2013 physics.ins-det, astro-ph.IM
SCUBA-2 is an innovative 10000 pixel bolometer camera operating at submillimetre wavelengths on the James Clerk Maxwell Telescope (JCMT). The camera has the capability to carry out wide-field surveys to unprecedented depths, addressing key questions relating to the origins of galaxies, stars and planets. With two imaging arrays working simultaneously in the atmospheric windows at 450 and 850 microns, the vast increase in pixel count means that SCUBA-2 maps the sky 100-150 times faster than the previous SCUBA instrument. In this paper we present an overview of the instrument, discuss the physical characteristics of the superconducting detector arrays, outline the observing modes and data acquisition, and present the early performance figures on the telescope. We also showcase the capabilities of the instrument via some early examples of the science SCUBA-2 has already undertaken. In February 2012, SCUBA-2 began a series of unique legacy surveys for the JCMT community. These surveys will take 2.5 years and the results are already providing complementary data to the shorter wavelength, shallower, larger-area surveys from Herschel. The SCUBA-2 surveys will also provide a wealth of information for further study with new facilities such as ALMA, and future telescopes such as CCAT and SPICA.
• ### Current distribution and transition width in superconducting transition-edge sensors(1212.3537)

Present models of the superconducting-to-normal transition in transition-edge sensors (TESs) do not describe the current distribution within a biased TES. This distribution is complicated by normal-metal features that are integral to TES design. We present a model with one free parameter that describes the evolution of the current distribution with bias. To probe the current distribution experimentally, we fabricated TES devices with different current return geometries. Devices where the current return geometry mirrors current flow within the device have sharper transitions, thus allowing for a direct test of the current-flow model. Measurements from these devices show that current meanders through a TES low in the resistive transition but flows across the normal-metal features by 40% of the normal-state resistance. Comparison of transition sharpness between device designs reveals that self-induced magnetic fields play an important role in determining the width of the superconducting transition. [http://dx.doi.org/10.1063/1.4771984]
• ### Code-division multiplexing for x-ray microcalorimeters(1201.6289)

Jan. 30, 2012 physics.ins-det, astro-ph.IM
We demonstrate the code-division multiplexed (CDM) readout of eight transition-edge sensor microcalorimeters. The energy resolution is 3.0 eV (full width at half-maximum) or better at 5.9 keV, with a best resolution of 2.3 eV and a mean of 2.6 eV over the seven modulated detectors. The flux-summing CDM system is described and compared with similar time-division multiplexed (TDM) readout. We show that the sqrt(Npixels) multiplexing disadvantage associated with TDM is not present in CDM. This demonstration establishes CDM as both a simple route to higher performance in existing TDM microcalorimetric experiments and a long-term approach to reaching higher multiplexing factors.
• ### Optimization and analysis of code-division multiplexed TES microcalorimeters(1110.2531)

Oct. 11, 2011 physics.ins-det, astro-ph.IM
We are developing code-division multiplexing (CDM) systems for transition-edge sensor arrays with the goal of reaching multiplexing factors in the hundreds. We report on x-ray measurements made with a four-channel prototype CDM system that employs a flux-summing architecture, emphasizing data-analysis issues. We describe an empirical method to determine the demodulation matrix that minimizes cross-talk. This CDM system achieves energy resolutions of between 2.3 eV and 3.0 eV FWHM at 5.9 keV.
• ### Advanced code-division multiplexers for superconducting detector arrays(1110.1608)

Multiplexers based on the modulation of superconducting quantum interference devices are now regularly used in multi-kilopixel arrays of superconducting detectors for astrophysics, cosmology, and materials analysis. Over the next decade, much larger arrays will be needed. These larger arrays require new modulation techniques and compact multiplexer elements that fit within each pixel. We present a new in-focal-plane code-division multiplexer that provides multiplexing elements with the required scalability. This code-division multiplexer uses compact lithographic modulation elements that simultaneously multiplex both signal outputs and superconducting transition-edge sensor (TES) detector bias voltages. It eliminates the shunt resistor used to voltage bias TES detectors, greatly reduces power dissipation, allows different dc bias voltages for each TES, and makes all elements sufficiently compact to fit inside the detector pixel area. These in-focal-plane code-division multiplexers can be combined with multi-gigahertz readout based on superconducting microresonators to scale to even larger arrays.