• ### Feasibility study of the time reversal symmetry tests in decays of metastable positronium atoms with the J-PET detector(1804.07148)

April 19, 2018 hep-ex, physics.ins-det
This article reports on the feasibility of testing of the symmetry under reversal in time in a purely leptonic system constituted by positronium atoms using the J-PET detector. The present state of T symmetry tests is discussed with an emphasis on the scarcely explored sector of leptonic systems. Two possible strategies of searching for manifestations of T violation in non-vanishing angular correlations of final state observables in the decays of metastable triplet states of positronium available with J-PET are proposed and discussed. Results of a pilot measurement with J-PET and assessment of its performance in reconstruction of three-photon decays are shown along with an analysis of its impact on the sensitivity of the detector for the determination of T -violation sensitive observables.
• ### Towards 2+1 photon tomography: Energy-based selection of two 511 keV photons and a prompt photon with the J-PET scanner(1803.00996)

The possibility to separate signals caused by 511 keV photons created in annihilation of electron-positron pairs and the so-called prompt photons from nuclei de- excitation is investigated. It could potentially be used to improve the quality of reconstructed images in the J-PET scanner in 2+1 photon tomography. Firstly, a research is conducted for several radioisotopes that decay via \b{eta}+ decay followed by de-excitation of an excited nucleus. Efficiency, purity and false positive rate are calculated for each isotope as a function of energy deposited threshold, with a hypothesis that signals caused by 511 keV photons deposit smaller values of energy than 1 z 13the selected threshold, while prompt photons deposit larger energy than the threshold. Analysis of the results accompanied with physical properties of radioisotopes suggests using 44 Sc, which is the most promising candidate for medical applications. With the use of GATE and J-POS simulation software, in-phantom scattering was introduced and the best energy deposited threshold value was estimated to be approximately 375 keV. It corresponds to almost 100% efficiency for 511 keV signals, 75% purity for 511 keV photons, and approximately 70% efficiency and purity for prompt photons.
• ### Time calibration of the J-PET detector(1710.05598)

Oct. 16, 2017 nucl-ex, physics.ins-det
The Jagiellonian Positron Emission Tomograph (J-PET) project carried out in the Institute of Physics of the Jagiellonian University is focused on construction and tests of the first prototype of PET scanner for medical diagnostic which allows for the simultaneous 3D imaging of the whole human body using organic scintillators. The J-PET prototype consists of 192 scintillator strips forming three cylindrical layers which are optimized for the detection of photons from the electron-positron annihilation with high time- and high angular-resolutions. In this article we present time calibration and synchronization of the whole J-PET detection system by irradiating each single detection module with a 22Na source and a small detector providing common reference time for synchronization of all the modules.
• ### Analysis procedure of the positronium lifetime spectra for the J-PET detector(1710.05611)

Oct. 16, 2017 nucl-ex, physics.ins-det
Positron Annihilation Lifetime Spectroscopy (PALS) has shown to be a powerful tool to study the nanostructures of porous materials. Positron Emissions Tomography (PET) are devices allowing imaging of metabolic processes e.g. in human bodies. A newly developed device, the J-PET (Jagiellonian PET), will allow PALS in addition to imaging, thus combining both analyses providing new methods for physics and medicine. In this contribution we present a computer program that is compatible with the J-PET software. We compare its performance with the standard program LT 9.0 by using PALS data from hexane measurements at different temperatures. Our program is based on an iterative procedure, and our fits prove that it performs as good as LT 9.0.
• ### Multichannel FPGA based MVT system for high precision time (20~ps~RMS) and charge measurement(1707.03565)

In this article it is presented an FPGA based $M$ulti-$V$oltage $T$hreshold (MVT) system which allows of sampling fast signals ($1-2$ ns rising and falling edge) in both voltage and time domain. It is possible to achieve a precision of time measurement of $20$ ps RMS and reconstruct charge of signals, using a simple approach, with deviation from real value smaller than 10$\%$. Utilization of the differential inputs of an FPGA chip as comparators together with an implementation of a TDC inside an FPGA allowed us to achieve a compact multi-channel system characterized by low power consumption and low production costs. This paper describes realization and functioning of the system comprising 192-channel TDC board and a four mezzanine cards which split incoming signals and discriminate them. The boards have been used to validate a newly developed Time-of-Flight Positron Emission Tomography system based on plastic scintillators. The achieved full system time resolution of $\sigma$(TOF) $\approx 68$ ps is by factor of two better with respect to the current TOF-PET systems.
• ### Calculation of time resolution of the J-PET tomograph using the Kernel Density Estimation(1706.00924)

In this paper we estimate the time resolution of the J-PET scanner built from plastic scintillators. We incorporate the method of signal processing using the Tikhonov regularization framework and the Kernel Density Estimation method. We obtain simple, closed-form analytical formulas for time resolutions. The proposed method is validated using signals registered by means of the single detection unit of the J-PET tomograph built out from 30 cm long plastic scintillator strip. It is shown that the experimental and theoretical results, obtained for the J-PET scanner equipped with vacuum tube photomultipliers, are consistent.
• ### Measurement of gamma quantum interaction point in plastic scintillator with WLS strips(1612.08571)

Dec. 27, 2016 physics.ins-det
We demonstrated the feasibility of measuring the axial coordinate of a gamma quantum interaction point in a plastic scintillator bar via the detection of scintillation photons escaping from the scintillator with an array of wavelength-shifting (WLS) strips. Using a test set-up comprising a BC-420 scintillator bar and an~array of sixteen BC-482A WLS strips we achieved a spatial resolution of~5\,mm~($\sigma$) for annihilation photons from a $^{22}$Na isotope. The studied method can be used to improve the spatial resolution of a plastic-scintillator-based PET scanner which is being developed by the J-PET collaboration.
• ### A feasibility study of ortho-positronium decays measurement with the J-PET scanner based on plastic scintillators(1607.08588)

Aug. 22, 2016 physics.ins-det
We present a study of the application of the Jagiellonian Positron Emission Tomograph (J-PET) for the registration of gamma quanta from decays of ortho-positronium (o-Ps). The J-PET is the first positron emission tomography scanner based on organic scintillators in contrast to all current PET scanners based on inorganic crystals. Monte Carlo simulations show that the J-PET as an axially symmetric and high acceptance scanner can be used as a multi-purpose detector well suited to pursue research including e.g. tests of discrete symmetries in decays of ortho-positronium in addition to the medical imaging. The gamma quanta originating from o-Ps decay interact in the plastic scintillators predominantly via the Compton effect, making the direct measurement of their energy impossible. Nevertheless, it is shown in this paper that the J-PET scanner will enable studies of the o-Ps$\to3\gamma$ decays with angular and energy resolution equal to $\sigma(\theta) \approx 0.4^{\circ}$ and $\sigma(E) \approx 4.1$ keV, respectively. An order of magnitude shorter decay time of signals from plastic scintillators with respect to the inorganic crystals results not only in better timing properties crucial for the reduction of physical and instrumental background, but also suppresses significantly the pileups, thus enabling compensation of the lower efficiency of the plastic scintillators by performing measurements with higher positron source activities.
• ### Sampling FEE and Trigger-less DAQ for the J-PET Scanner(1602.05251)

Feb. 17, 2016 physics.ins-det
In this paper, we present a complete Data Acquisition System (DAQ) together with the readout mechanisms for the J-PET tomography scanner. In general detector readout chain is constructed out of Front-End Electronics (FEE), measurement devices like Time-to-Digital or Analog-to-Digital Converters (TDCs or ADCs), data collectors and storage. We have developed a system capable for maintaining continuous readout of digitized data without preliminary selection. Such operation mode results in up to 8 Gbps data stream, therefore it is required to introduce a dedicated module for online event building and feature extraction. The Central Controller Module, equipped with Xilinx Zynq SoC and 16 optical transceivers serves as such true real time computing facility. Our solution for the continuous data recording (trigger-less) is a novel approach in such detector systems and assures that most of the information is preserved on the storage for further, high-level processing. Signal discrimination applies an unique method of using LVDS buffers located in the FPGA fabric.
• ### Determination of the 3\gamma fraction from positron annihilation in mesoporous materials for symmetry violation experiment with J-PET scanner(1602.05376)

Feb. 17, 2016 nucl-ex, physics.ins-det
Various mesoporous materials were investigated to choose the best material for experiments requiring high yield of long-lived positronium. We found that the fraction of 3\gamma annihilation determined using \gamma-ray energy spectra and positron annihilation lifetime spectra (PAL) changed from 20% to 25%. The 3gamma fraction and o-Ps formation probability in the polymer XAD-4 is found to be the largest. Elemental analysis performed using scanning electron microscop (SEM) equipped with energy-dispersive X-ray spectroscop EDS show high purity of the investigated materials.