• ### Innovative design and construction technique for the Cylindrical GEM detector for the BESIII experiment(1803.07797)

March 21, 2018 physics.ins-det
Gas detector are very light instrument used in high energy physics to measure the particle properties: position and momentum. Through high electric field is possible to use the Gas Electron Multiplier (GEM) technology to detect the particles and to exploit the its properties to construct a large area detector, such as the new IT for BESIII. The state of the art in the GEM production allow to create very large area GEM foils (up to 50x100 cm2) and thanks to the small thickness of these foil is it possible to shape it to the desired form: a Cylindrical Gas Electron Multiplier (CGEM) is then proposed. The innovative construction technique based on Rohacell, a PMI foam, will give solidity to cathode and anode with a very low impact on material budget. The entire detector is sustained by permaglass rings glued at the edges. These rings are use to assembly the CGEM together with a dedicated Vertical Insertion System and moreover there is placed the On-Detector electronic. The anode has been improved w.r.t. the state of the art through a jagged readout that minimize the inter-strip capacitance. The mechanical challenge of this detector requires a precision of the entire geometry within few hundreds of microns in the whole area. In this presentation will be presented an overview of the construction technique and the validation of this technique through the realization of a CGEM and its first tests. These activities are performed within the framework of the BESIIICGEM Project (645664), funded by the European Commission in the action H2020-RISE-MSCA-2014.
• ### The new cylindrical GEM inner tracker of BESIII(1803.07258)

March 20, 2018 hep-ex, physics.ins-det
The Cylindrical GEM-Inner Tracker (CGEM-IT) is the upgrade of the internal tracking system of the BESIII experiment. It consists of three layers of cylindrically-shaped triple GEMs, with important innovations with respect to the existing GEM detectors, in order to achieve the best performance with the lowest material budget. It will be the first cylindrical GEM running with analog readout inside a 1T magnetic field. The simultaneous measurement of both the deposited charge and the signal time will permit to use a combination of two algorithms to evaluate the spatial position of the charged tracks inside the CGEM-IT: the charge centroid and the micro time projection chamber modes. They are complementary and can cope with the asymmetry of the electron avalanche when running in magnetic field and with non-orthogonal incident tracks. To evaluate the behavior under different working settings, both planar chambers and the first cylindrical prototype have been tested during various test beams at CERN with 150 GeV/c muons and pions. This paper reports the results obtained with the two reconstruction methods and a comparison between the planar and cylindrical chambers.
• ### Test beam results with prototypes for the new Cylindrical GEM Inner Tracker of the BESIII experiment(1803.07263)

March 20, 2018 physics.ins-det
A cylindrical GEM tracker is under construction in order to replace and improve the inner tracking system of the BESIII experiment. Tests with planar chamber prototypes were carried out on the H4 beam line of SPS (CERN) with muons of 150 GeV/c momentum, to evaluate the efficiency and resolution under different working conditions. The obtained efficiency was in the 96 - 98% range. Two complementary algorithms for the position determination were developed: the charge centroid and the micro-TPC methods. With the former, resolutions <100 micron and <200 micron were achieved without and with magnetic field, respectively. The micro-TPC improved these results. By the end of 2016, the first cylindrical prototype was tested on the same beam line. It showed optimal stability under different settings. The comparison of its performance with respect to the planar chambers is ongoing. Here, the results of the planar prototype tests will be addressed.
• ### Performance of the micro-TPC Reconstruction for GEM Detectors at High Rate(1803.07266)

March 20, 2018 physics.ins-det
Gas detectors are one of the pillars of the research in fundamental physics. Since many years, a new concept of detectors, the Micro Pattern Gas Detectors, allows to overcome many of the problems of other types of commonly used detectors, as drift chambers and microstrips, reducing the discharge rate and increasing the radiation tolerance. Among these, one of the most commonly used is the Gas Electron Multiplier. Commonly deployed as fast timing detectors and triggers, due to their fast response, high rate capability and high radiation hardness, they can also be used as trackers. The center of gravity readout technique allows to overcome the limit of the digital pads, whose spatial resolution is constrained by the pitch size. The presence of a high external magnetic field can distort the electronic cloud and affect the spatial resolution. The micro-TPC reconstruction method allows to reconstruct the three dimensional particle position as in a traditional Time Projection Chamber, but within a drift gap of a few millimeters. This method brings these detectors into a new perspective for what concerns the spatial resolution in strong magnetic field. In this report, the basis of this new technique will be shown and it will be compared to the traditional center of gravity. The results of a series of test beam performed with 10 x 10 cm2 planar prototypes in magnetic field will also be presented. This is one of the first implementations of this technique for GEM detectors in magnetic field and allows to reach unprecedented performance for gas detectors, up to a limit of 120 micron at 1T, one of the world's best results for MPGDs in strong magnetic field. The micro-TPC reconstruction has been recently tested at very high rates in a test beam at the MAMI facility; preliminary results of the test will be presented.
• ### Test beam results of a Cylindrical GEM detector for the BESIII experiment(1803.07489)

March 20, 2018 hep-ex, physics.ins-det
Gas detector are very light instrument used in high energy physics to measure the particle properties: position and momentum. Through high electric field is possible to use the Gas Electron Multiplier (GEM) technology to detect the charged particles and to exploit their properties to construct a large area detector, such as the new IT for BESIII. The state of the art in the GEM production allows to create very large area GEM foils (up to 50x100 $\mathrm{cm}^2$) and thanks to the small thickness of these foils is it possible to shape it to the desired form: a Cylindrical Gas Electron Multiplier (CGEM) is then proposed. The innovative construction technique based on Rohacell, a PMI foam, will give solidity to cathode and anode with a very low impact on material budget. The entire detector is sustained by Permaglass rings glued at the edges. These rings are used to assembly the CGEM, together with a dedicated Vertical Insertion System and moreover they host the On-Detector electronic. The anode has been improved w.r.t. the state of the art through a jagged readout that minimize the inter-strip capacitance. The mechanical challenge of this detector requires a precision of the entire geometry within few hundreds of microns in the whole area. In this contribution an overview of the construction technique, the validation of this technique through the realization of a CGEM, and its first tests will be presented. These activities are performed within the framework of the BESIIICGEM Project (645664), funded by the European Commission in the action H2020-RISE-MSCA-2014.
• ### Status of COMPASS RICH-1 Upgrade with MPGD-based Photon Detectors(1710.09274)

Oct. 25, 2017 hep-ex, nucl-ex, physics.ins-det
A Set of new MPGD-based Photon Detectors is being built for the upgrade of COMPASS RICH-1. The detectors cover a total active area of 1.4 m$^2$ and are based on a hybrid architecture consisting of two THGEM layers and a Micromegas. A CsI film on one THGEM acts as a reflective photocathode. The characteristics of the detector, the production of the components and their validation tests are described in detail.
• ### Development and Test of a uTPC Cluster Reconstruction for a Triple GEM Detector in Strong Magnetic Field(1707.04456)

July 14, 2017 physics.ins-det
Performance of triple GEM prototypes has been evaluated by means of a muon beam at the H4 line of the SPS test area at CERN. The data from two planar prototypes have been reconstructed and analyzed offline with two clusterization methods: the enter of gravity of the charge distribution and the micro Time Projection Chamber (\muTPC). Concerning the spatial resolution, the charge centroid cluster reconstruction performs extremely well with no magnetic field: the resolution is well below 100 \mum . Increasing the magnetic field intensity, the resolution degrades almost linearly as effect of the Lorentz force that displaces, broadens and asymmetrizes the electron avalanche. Tuning the electric fields of the GEM prototype we could achieve the unprecedented spatial resolution of 190 \mum at 1 Tesla. In order to boost the spatial resolution with strong magnetic field and inclined tracks a \muTPC cluster reconstruction has been investigated. Such a readout mode exploits the good time resolution of the GEM detector and electronics to reconstruct the trajectory of the particle inside the conversion gap. Beside the improvement of the spatial resolution, information on the track angle can be also extracted. The new clustering algorithm has been tested with diagonal tracks with no magnetic field showing a resolution between 100 um and 150 um for the incident angle ranging from 10{\deg} to 45{\deg} . Studies show similar performance with 1 Tesla magnetic field. This is the first use of a \muTPC readout with a triple GEM detector in magnetic field. This study has shown that a combined readout is capable to guarantee stable performance over a broad spectrum of particle momenta and incident angles, up to a 1 Tesla magnetic field.
• ### The Cylindrical GEM Inner Tracker of the BESIII experiment: prototype test beam results(1706.02428)

July 3, 2017 physics.ins-det
A cylindrical GEM detector is under development, to serve as an upgraded inner tracker at the BESIII spectrometer. It will consist of three layers of cylindrically-shaped triple GEMs surrounding the interaction point. The experiment is taking data at the e+e- collider BEPCII in Beijing (China) and the GEM tracker will be installed in 2018. Tests on the performances of triple GEMs in strong magnetic field have been run by means of the muon beam available in the H4 line of SPS (CERN) with both planar chambers and the first cylindrical prototype. Efficiencies and resolutions have been evaluated using different gains, gas mixtures, with and without magnetic field. The obtained efficiency is 97-98% on single coordinate view, in many operational arrangements. The spatial resolution for planar GEMs has been evaluated with two different algorithms for the position determination: the charge centroid and the micro time projection chamber (mu-TPC) methods. The two modes are complementary and are able to cope with the asymmetry of the electron avalanche when running in magnetic field, and with non-orthogonal incident tracks. With the charge centroid, a resolution lower than 100 micron has been reached without magnetic field and lower than 200 micron with a magnetic field up to 1 T. The mu-TPC mode showed to be able to improve those results. In the first beam test with the cylindrical prototype, the detector had a very good stability under different voltage configurations and particle intensities. The resolution evaluation is in progress.
• ### A custom readout electronics for the BESIII CGEM detector(1706.02267)

June 28, 2017 physics.ins-det
For the upgrade of the inner tracker of the BESIII spectrometer, planned for 2018, a lightweight tracker based on an innovative Cylindrical Gas Electron Multiplier (CGEM) detector is now under development. The analogue readout of the CGEM enables the use of a charge centroid algorithm to improve the spatial resolution to better than 130 um while loosening the pitch strip to 650 um, which allows to reduce the total number of channels to about 10 000. The channels are readout by 160 dedicated integrated 64-channel front-end ASICs, providing a time and charge measurement and featuring a fully-digital output. The energy measurement is extracted either from the time-over-threshold (ToT) or the 10-bit digitisation of the peak amplitude of the signal. The time of the event is generated by quad-buffered low-power TDCs, allowing for rates in excess of 60 kHz per channel. The TDCs are based on analogue interpolation techniques and produce a time stamp (or two, if working in ToT mode) of the event with a time resolution better than 50 ps. The front-end noise, based on a CSA and CR-RC2 shapers, dominate the channel intrinsic time jitter, which is less than 5 ns r.m.s.. The time information of the hit can be used to reconstruct the track path, operating the detector as a small TPC and hence improving the position resolution when the distribution of the cloud, due to large incident angle or magnetic field, is very broad. Event data is collected by an off-detector motherboard, where each GEM-ROC readout card handles 4 ASIC carrier PCBs (512 channels). Configuration upload and data readout between the off-detector electronics and the VME-based data collector cards are managed by bi-directional fibre optical links.
• ### Formation of the $S=-1$ resonance X(2265) in the reaction $pp -> X + K^+$ at 2.50 and 2.85 GeV(1102.0482)

June 23, 2012 nucl-ex
Analyzing DISTO data of $pp -> p Lambda K^+$ at $T_p = 2.50$ and 2.85 GeV to populate a previously reported X(2265) resonance with $M_X = 2267$ MeV/$c^2$ and $Gamma_X = 118$ MeV at 2.85 GeV, we found that the production of X(2265) at 2.50 GeV is much less than that at 2.85 GeV (less than 10%), though it is expected from a kinematical consideration to be produced as much as 33% of that at 2.85 GeV. The small population of X(2265) at 2.50 GeV is consistent with the very weak production of Lambda (1405) at the same incident energy toward its production threshold, thus indicating that Lambda (1405) plays an important role as a doorway state for the formation of X(2265).
• ### Indication of a deeply bound compact K-pp state formed in the pp -> p Lambda K+ reaction at 2.85 GeV(1002.3526)

Feb. 18, 2010 nucl-ex
We have analyzed data of the DISTO experiment on the exclusive pp -> p Lambda K+ reaction at 2.85 GeV to search for a strongly bound compact K-pp (= X) state to be formed in the pp -> K+ + X reaction. The observed spectra of the K+ missing-mass and the p Lambda invariant-mass with high transverse momenta of p and K+ revealed a broad distinct peak with a mass M_X = 2265 +- 2 (stat) +- 5 (syst) MeV/c2 and a width Gamma_X = 118 +- 8 (stat) +- 10 (syst) MeV.
• ### Pattern recognition and PID for COMPASS RICH-1(0902.0444)

Feb. 3, 2009 hep-ex, physics.ins-det
A package for pattern recognition and PID by COMPASS RICH-1 has been developed and used for the analysis of COMPASS data collected in the years 2002 to 2004, and 2006-2007 with the upgraded RICH-1 photon detectors. It has allowed the full characterization of the detector in the starting version and in the upgraded one, as well as the PID for physics results. We report about the package structure and algorithms, and the detector characterization and PID results.