• ### The calibration of read-out-streak photometry in the XMM-Newton Optical Monitor and the construction of a bright-source catalogue(1612.03506)

Dec. 12, 2016 astro-ph.IM
The dynamic range of the XMM-Newton Optical Monitor (XMM-OM) is limited at the bright end by coincidence loss, the superposition of multiple photons in the individual frames recorded from its micro-channel-plate (MCP) intensified charge-coupled device (CCD) detector. One way to overcome this limitation is to use photons that arrive during the frame transfer of the CCD, forming vertical read-out streaks for bright sources. We calibrate these read-out streaks for photometry of bright sources observed with XMM-OM. The bright source limit for read-out streak photometry is set by the recharge time of the MCPs. For XMM-OM we find that the MCP recharge time is 0.55 ms. We determine that the effective bright limits for read-out streak photometry with XMM-OM are approximately 1.5 magnitudes brighter than the bright source limits for normal aperture photometry in full-frame images. This translates into bright-source limits in Vega magnitudes of UVW2=7.1, UVM2=8.0, UVW1=9.4, U=10.5, B=11.5, V=10.2 and White=12.5 for data taken early in the mission. The limits brighten by up to 0.2 magnitudes, depending on filter, over the course of the mission as the detector ages. The method is demonstrated by deriving UVW1 photometry for the symbiotic nova RR Telescopii, and the new photometry is used to constrain the e-folding time of its decaying UV emission. Using the read-out streak method, we obtain photometry for 50 per cent of the missing UV source measurements in version 2.1 of the XMM-Newton Serendipitous UV Source Survey (XMM-SUSS 2.1) catalogue.
• ### PSR J1930-1852: a pulsar in the widest known orbit around another neutron star(1503.06276)

March 21, 2015 astro-ph.HE
In the summer of 2012, during a Pulsar Search Collaboratory workshop, two high-school students discovered J1930$-$1852, a pulsar in a double neutron star (DNS) system. Most DNS systems are characterized by short orbital periods, rapid spin periods and eccentric orbits. However, J1930$-$1852 has the longest spin period ($P_{\rm spin}\sim$185 ms) and orbital period ($P_{\rm b}\sim$45 days) yet measured among known, recycled pulsars in DNS systems, implying a shorter than average and/or inefficient recycling period before its companion went supernova. We measure the relativistic advance of periastron for J1930$-$1852, $\dot{\omega}=0.00078$(4) deg/yr, which implies a total mass (M$_{\rm{tot}}=2.59$(4) M$_{\odot}$) consistent with other DNS systems. The $2\sigma$ constraints on M$_{\rm{tot}}$ place limits on the pulsar and companion masses ($m_{\rm p}<1.32$ M$_{\odot}$ and $m_{\rm c}>1.30$ M$_{\odot}$ respectively). J1930$-$1852's spin and orbital parameters challenge current DNS population models and make J1930$-$1852 an important system for further investigation.
• ### The Swift-UVOT ultraviolet and visible grism calibration(1501.02433)

Feb. 23, 2015 astro-ph.IM
We present the calibration of the Swift UVOT grisms, of which there are two, providing low-resolution field spectroscopy in the ultraviolet and optical bands respectively. The UV grism covers the range 1700-5000 Angstrom with a spectral resolution of 75 at 2600 Angstrom for source magnitudes of u=10-16 mag, while the visible grism covers the range 2850-6600 Angstrom with a spectral resolution of 100 at 4000 Angstrom for source magnitudes of b=12-17 mag. This calibration extends over all detector positions, for all modes used during operations. The wavelength accuracy (1-sigma) is 9 Angstrom in the UV grism clocked mode, 17 Angstrom in the UV grism nominal mode and 22 Angstrom in the visible grism. The range below 2740 Angstrom in the UV grism and 5200 Angstrom in the visible grism never suffers from overlapping by higher spectral orders. The flux calibration of the grisms includes a correction we developed for coincidence loss in the detector. The error in the coincidence loss correction is less than 20%. The position of the spectrum on the detector only affects the effective area (sensitivity) by a few percent in the nominal modes, but varies substantially in the clocked modes. The error in the effective area is from 9% in the UV grism clocked mode to 15% in the visible grism clocked mode .
• ### The use and calibration of read-out streaks to increase the dynamic range of the Swift Ultraviolet/Optical Telescope(1309.4816)

Sept. 18, 2013 astro-ph.IM
The dynamic range of photon counting micro-channel-plate (MCP) intensified charged-coupled device (CCD) instruments such as the Swift Ultraviolet/Optical Telescope (UVOT) and the XMM-Newton Optical Monitor (XMM-OM) is limited at the bright end by coincidence loss, the superposition of multiple photons in the individual frames recorded by the CCD. Photons which arrive during the brief period in which the image frame is transferred for read out of the CCD are displaced in the transfer direction in the recorded images. For sufficiently bright sources, these displaced counts form read-out streaks. Using UVOT observations of Tycho-2 stars, we investigate the use of these read-out streaks to obtain photometry for sources which are too bright (and hence have too much coincidence loss) for normal aperture photometry to be reliable. For read-out-streak photometry, the bright-source limiting factor is coincidence loss within the MCPs rather than the CCD. We find that photometric measurements can be obtained for stars up to 2.4 magnitudes brighter than the usual full-frame coincidence-loss limit by using the read-out streaks. The resulting bright-limit Vega magnitudes in the UVOT passbands are UVW2=8.80, UVM2=8.27, UVW1=8.86, u=9.76, b=10.53, v=9.31 and White=11.71; these limits are independent of the windowing mode of the camera. We find that a photometric precision of 0.1 mag can be achieved through read-out streak measurements. A suitable method for the measurement of read-out streaks is described and all necessary calibration factors are given.
• ### Paper II: Calibration of the Swift ultraviolet/optical telescope(1004.2448)

April 14, 2010 astro-ph.IM
The Ultraviolet/Optical Telescope (UVOT) is one of three instruments onboard the Swift observatory. The photometric calibration has been published, and this paper follows up with details on other aspects of the calibration including a measurement of the point spread function with an assessment of the orbital variation and the effect on photometry. A correction for large scale variations in sensitivity over the field of view is described, as well as a model of the coincidence loss which is used to assess the coincidence correction in extended regions. We have provided a correction for the detector distortion and measured the resulting internal astrometric accuracy of the UVOT, also giving the absolute accuracy with respect to the International Celestial Reference System. We have compiled statistics on the background count rates, and discuss the sources of the background, including instrumental scattered light. In each case we describe any impact on UVOT measurements, whether any correction is applied in the standard pipeline data processing or whether further steps are recommended.
• ### The FP420 R&D Project: Higgs and New Physics with forward protons at the LHC(0806.0302)

Jan. 2, 2009 hep-ph, hep-ex, nucl-ex
We present the FP420 R&D project, which has been studying the key aspects of the development and installation of a silicon tracker and fast-timing detectors in the LHC tunnel at 420 m from the interaction points of the ATLAS and CMS experiments. These detectors would measure precisely very forward protons in conjunction with the corresponding central detectors as a means to study Standard Model (SM) physics, and to search for and characterise New Physics signals. This report includes a detailed description of the physics case for the detector and, in particular, for the measurement of Central Exclusive Production, pp --> p + phi + p, in which the outgoing protons remain intact and the central system phi may be a single particle such as a SM or MSSM Higgs boson. Other physics topics discussed are gamma-gamma and gamma-p interactions, and diffractive processes. The report includes a detailed study of the trigger strategy, acceptance, reconstruction efficiencies, and expected yields for a particular p p --> p H p measurement with Higgs boson decay in the b-bbar mode. The document also describes the detector acceptance as given by the LHC beam optics between the interaction points and the FP420 location, the machine backgrounds, the new proposed connection cryostat and the moving ("Hamburg") beam-pipe at 420 m, and the radio-frequency impact of the design on the LHC. The last part of the document is devoted to a description of the 3D silicon sensors and associated tracking performances, the design of two fast-timing detectors capable of accurate vertex reconstruction for background rejection at high-luminosities, and the detector alignment and calibration strategy.
• ### Photometric Calibration of the Swift Ultraviolet/Optical Telescope(0708.2259)

Nov. 9, 2007 astro-ph
We present the photometric calibration of the Swift UltraViolet/Optical Telescope (UVOT) which includes: optimum photometric and background apertures, effective area curves, colour transformations, conversion factors for count rates to flux, and the photometric zero points (which are accurate to better than 4 per cent) for each of the seven UVOT broadband filters. The calibration was performed with observations of standard stars and standard star fields that represent a wide range of spectral star types. The calibration results include the position dependent uniformity, and instrument response over the 1600-8000A operational range. Because the UVOT is a photon counting instrument, we also discuss the effect of coincidence loss on the calibration results. We provide practical guidelines for using the calibration in UVOT data analysis. The results presented here supersede previous calibration results.