A picoammeter system has been developed and engineering. It consists in a
current-voltage converter, based on an operational amplifier with very low
input current, a high precision ADC, a radio controlled data acquisition unit
and the computer-based control, visualization and storage. The precision is of
the order of a tenth of picoampers and it can measure currents between
electrodes at potentials up to 8 kV. The system is battery powered and a number
of strategies have been implemented to limit the power consumption. The system
is designed for multichannel applications, up to 256 parallel channels. The
overall implementation is cost-effective to make the availability of
multichannel setups easily affordable. The design, implementation and
performance of the picoammeter system are described in detail as well as a an
application: the measurement of ion backflow in MPGD-based photon detectors.
The architecture of the novel MPGD-based photon detectors of COMPASS RICH-1
consists in a large-size hybrid MPGD multilayer layout combining two layers of
Thick-GEMs and a bulk resistive MICROMEGAS. Concerning biasing voltage, the
Thick-GEMs are segmented in order to reduce the energy released in case of
occasional discharges, while the MICROMEGAS anode is segmented in pads
individually biased at positive voltage, while the micromesh is grounded. In
total, there are ten different electrode types and more than 20000 electrodes
supplied by more than 100 HV channels. Commercial power supply units are used.
The original elements of the power supply system are the architecture of the
voltage distribution net, the compensation, by voltage adjustment, of the
effects of pressure and temperature variation affecting the detector gain and a
sophisticated control software, which allows to protect the detectors against
errors by the operator, to monitor and log voltages and current at 1 Hz rate
and to automatically react to detector misbehaviors. The HV system and its
performance are described in detail as well as the electrical stability of the
detector during the operation at COMPASS.
A Set of new MPGD-based Photon Detectors is being built for the upgrade of
COMPASS RICH-1. The detectors cover a total active area of 1.4 m$^2$ and are
based on a hybrid architecture consisting of two THGEM layers and a Micromegas.
A CsI film on one THGEM acts as a reflective photocathode. The characteristics
of the detector, the production of the components and their validation tests
are described in detail.
The main characteristics of the COMPASS experimental setup for physics with
hadron beams are described. This setup was designed to perform exclusive
measurements of processes with several charged and/or neutral particles in the
final state. Making use of a large part of the apparatus that was previously
built for spin structure studies with a muon beam, it also features a new
target system as well as new or upgraded detectors. The hadron setup is able to
operate at the high incident hadron flux available at CERN. It is characterised
by large angular and momentum coverages, large and nearly flat acceptances, and
good two and three-particle mass resolutions. In 2008 and 2009 it was
successfully used with positive and negative hadron beams and with liquid
hydrogen and solid nuclear targets. This article describes the new and upgraded
detectors and auxiliary equipment, outlines the reconstruction procedures used,
and summarises the general performance of the setup.