• ### Multichannel FPGA based MVT system for high precision time (20~ps~RMS) and charge measurement(1707.03565)

In this article it is presented an FPGA based $M$ulti-$V$oltage $T$hreshold (MVT) system which allows of sampling fast signals ($1-2$ ns rising and falling edge) in both voltage and time domain. It is possible to achieve a precision of time measurement of $20$ ps RMS and reconstruct charge of signals, using a simple approach, with deviation from real value smaller than 10$\%$. Utilization of the differential inputs of an FPGA chip as comparators together with an implementation of a TDC inside an FPGA allowed us to achieve a compact multi-channel system characterized by low power consumption and low production costs. This paper describes realization and functioning of the system comprising 192-channel TDC board and a four mezzanine cards which split incoming signals and discriminate them. The boards have been used to validate a newly developed Time-of-Flight Positron Emission Tomography system based on plastic scintillators. The achieved full system time resolution of $\sigma$(TOF) $\approx 68$ ps is by factor of two better with respect to the current TOF-PET systems.
• ### Measurement of gamma quantum interaction point in plastic scintillator with WLS strips(1612.08571)

Dec. 27, 2016 physics.ins-det
We demonstrated the feasibility of measuring the axial coordinate of a gamma quantum interaction point in a plastic scintillator bar via the detection of scintillation photons escaping from the scintillator with an array of wavelength-shifting (WLS) strips. Using a test set-up comprising a BC-420 scintillator bar and an~array of sixteen BC-482A WLS strips we achieved a spatial resolution of~5\,mm~($\sigma$) for annihilation photons from a $^{22}$Na isotope. The studied method can be used to improve the spatial resolution of a plastic-scintillator-based PET scanner which is being developed by the J-PET collaboration.
• ### TOF-PET detector concept based on organic scintillators(1305.5559)

In this contribution we present a new concept of the large acceptance detector systems based on organic scintillators which may allow for simultaneous diagnostic of large fraction of the human body. Novelty of the concept lies in employing large blocks of polymer scintillators instead of crystals as detectors of annihilation quanta, and in using predominantly the timing of signals instead of their amplitudes.
• ### STRIP-PET: a novel detector concept for the TOF-PET scanner(1305.5562)

We briefly present a design of a new PET scanner based on strips of polymer scintillators arranged in a barrel constituting a large acceptance detector. The solution proposed is based on the superior timing properties of the polymer scintillators. The position and time of the reaction of the gamma quanta in the detector material will be determined based on the time of arrival of light signals to the edges of the scintillator strips.
• This document describes the technical layout and the expected performance of the Straw Tube Tracker (STT), the main tracking detector of the PANDA target spectrometer. The STT encloses a Micro-Vertex-Detector (MVD) for the inner tracking and is followed in beam direction by a set of GEM-stations. The tasks of the STT are the measurement of the particle momentum from the reconstructed trajectory and the measurement of the specific energy-loss for a particle identification. Dedicated simulations with full analysis studies of certain proton-antiproton reactions, identified as being benchmark tests for the whole PANDA scientific program, have been performed to test the STT layout and performance. The results are presented, and the time lines to construct the STT are described.
• ### The TRB for HADES and FAIR experiments at GSI(0810.4723)

Nov. 6, 2008 nucl-ex
The TRB hardware module is a multi-purpose Trigger and Readout Board with on-board DAQ functionality developed for the upgrade of the HADES experiment. It contains a single computer chip (Etrax) running Linux as a well as a 100 Mbit/s Ethernet interface. It has been orginally designed to work as a 128-channel Time to Digital Converter based on the HPTDC chip from CERN. The new version contains a 2 Gbit/s optical link and an interface connector (15 Gbit/s) in order to realize an add-on card concept which makes the hardware very flexible. Moreover, an FPGA chip (Xilinx, Virtex 4 LX 40) and a TigerSharc DSP provide new computing resources which can be used to run on-line analysis algorithms. The TRB is proposed as a prototype for new modules for the planned detector systems PANDA and CBM at the future FAIR facility at GSI-Darmstadt.