• ### Towards 2+1 photon tomography: Energy-based selection of two 511 keV photons and a prompt photon with the J-PET scanner(1803.00996)

The possibility to separate signals caused by 511 keV photons created in annihilation of electron-positron pairs and the so-called prompt photons from nuclei de- excitation is investigated. It could potentially be used to improve the quality of reconstructed images in the J-PET scanner in 2+1 photon tomography. Firstly, a research is conducted for several radioisotopes that decay via \b{eta}+ decay followed by de-excitation of an excited nucleus. Efficiency, purity and false positive rate are calculated for each isotope as a function of energy deposited threshold, with a hypothesis that signals caused by 511 keV photons deposit smaller values of energy than 1 z 13the selected threshold, while prompt photons deposit larger energy than the threshold. Analysis of the results accompanied with physical properties of radioisotopes suggests using 44 Sc, which is the most promising candidate for medical applications. With the use of GATE and J-POS simulation software, in-phantom scattering was introduced and the best energy deposited threshold value was estimated to be approximately 375 keV. It corresponds to almost 100% efficiency for 511 keV signals, 75% purity for 511 keV photons, and approximately 70% efficiency and purity for prompt photons.
• ### Multichannel FPGA based MVT system for high precision time (20~ps~RMS) and charge measurement(1707.03565)

In this article it is presented an FPGA based $M$ulti-$V$oltage $T$hreshold (MVT) system which allows of sampling fast signals ($1-2$ ns rising and falling edge) in both voltage and time domain. It is possible to achieve a precision of time measurement of $20$ ps RMS and reconstruct charge of signals, using a simple approach, with deviation from real value smaller than 10$\%$. Utilization of the differential inputs of an FPGA chip as comparators together with an implementation of a TDC inside an FPGA allowed us to achieve a compact multi-channel system characterized by low power consumption and low production costs. This paper describes realization and functioning of the system comprising 192-channel TDC board and a four mezzanine cards which split incoming signals and discriminate them. The boards have been used to validate a newly developed Time-of-Flight Positron Emission Tomography system based on plastic scintillators. The achieved full system time resolution of $\sigma$(TOF) $\approx 68$ ps is by factor of two better with respect to the current TOF-PET systems.
• ### Calculation of time resolution of the J-PET tomograph using the Kernel Density Estimation(1706.00924)

In this paper we estimate the time resolution of the J-PET scanner built from plastic scintillators. We incorporate the method of signal processing using the Tikhonov regularization framework and the Kernel Density Estimation method. We obtain simple, closed-form analytical formulas for time resolutions. The proposed method is validated using signals registered by means of the single detection unit of the J-PET tomograph built out from 30 cm long plastic scintillator strip. It is shown that the experimental and theoretical results, obtained for the J-PET scanner equipped with vacuum tube photomultipliers, are consistent.
• ### Time resolution of the plastic scintillator strips with matrix photomultiplier readout for J-PET tomograph(1602.02058)

Recent tests of a single module of the Jagiellonian Positron Emission Tomography system (J-PET) consisting of 30 cm long plastic scintillator strips have proven its applicability for the detection of annihilation quanta (0.511 MeV) with a coincidence resolving time (CRT) of 0.266 ns. The achieved resolution is almost by a factor of two better with respect to the current TOF-PET detectors and it can still be improved since, as it is shown in this article, the intrinsic limit of time resolution for the determination of time of the interaction of 0.511 MeV gamma quanta in plastic scintillators is much lower. As the major point of the article, a method allowing to record timestamps of several photons, at two ends of the scintillator strip, by means of matrix of silicon photomultipliers (SiPM) is introduced. As a result of simulations, conducted with the number of SiPM varying from 4 to 42, it is shown that the improvement of timing resolution saturates with the growing number of photomultipliers, and that the 2 x 5 configuration at two ends allowing to read twenty timestamps, constitutes an optimal solution. The conducted simulations accounted for the emission time distribution, photon transport and absorption inside the scintillator, as well as quantum efficiency and transit time spread of photosensors, and were checked based on the experimental results. Application of the 2 x 5 matrix of SiPM allows for achieving the coincidence resolving time in positron emission tomography of $\approx$ 0.170 ns for 15 cm axial field-of-view (AFOV) and $\approx$ 0.365 ns for 100 cm AFOV. The results open perspectives for construction of a cost-effective TOF-PET scanner with significantly better TOF resolution and larger AFOV with respect to the current TOF-PET modalities.
• ### Sampling FEE and Trigger-less DAQ for the J-PET Scanner(1602.05251)

Feb. 17, 2016 physics.ins-det
In this paper, we present a complete Data Acquisition System (DAQ) together with the readout mechanisms for the J-PET tomography scanner. In general detector readout chain is constructed out of Front-End Electronics (FEE), measurement devices like Time-to-Digital or Analog-to-Digital Converters (TDCs or ADCs), data collectors and storage. We have developed a system capable for maintaining continuous readout of digitized data without preliminary selection. Such operation mode results in up to 8 Gbps data stream, therefore it is required to introduce a dedicated module for online event building and feature extraction. The Central Controller Module, equipped with Xilinx Zynq SoC and 16 optical transceivers serves as such true real time computing facility. Our solution for the continuous data recording (trigger-less) is a novel approach in such detector systems and assures that most of the information is preserved on the storage for further, high-level processing. Signal discrimination applies an unique method of using LVDS buffers located in the FPGA fabric.
• ### Determination of the 3\gamma fraction from positron annihilation in mesoporous materials for symmetry violation experiment with J-PET scanner(1602.05376)

Feb. 17, 2016 nucl-ex, physics.ins-det
Various mesoporous materials were investigated to choose the best material for experiments requiring high yield of long-lived positronium. We found that the fraction of 3\gamma annihilation determined using \gamma-ray energy spectra and positron annihilation lifetime spectra (PAL) changed from 20% to 25%. The 3gamma fraction and o-Ps formation probability in the polymer XAD-4 is found to be the largest. Elemental analysis performed using scanning electron microscop (SEM) equipped with energy-dispersive X-ray spectroscop EDS show high purity of the investigated materials.
• ### Searches for discrete symmetries violation in ortho-positronium decay using the J-PET detector(1509.01114)

Sept. 3, 2015 hep-ex, physics.ins-det
In this paper we present prospects for using the J-PET detector to search for discrete symmetries violations in a purely leptonic system of the positronium atom. We discuss tests of CP and CPT symmetries by means of ortho-positronium decays into three photons. No zero expectation values for chosen correlations between ortho-positronium spin and momentum vectors of photons would imply the existence of physics phenomena beyond the Standard Model. Previous measurements resulted in violation amplitude parameters for CP and CPT symmetries consistent with zero, with an uncertainty of about 10-3. The J-PET detector allows to determine those values with better precision thanks to a unique time and angular esolution combined with a high geometrical acceptance. Achieving the aforementioned is possible due to application of polymer scintillators instead of crystals as detectors of annihilation quanta.
• ### Studies of unicellular micro-organisms Saccharomyces cerevisiae by means of Positron Annihilation Lifetime Spectroscopy(1509.00411)

Aug. 30, 2015 physics.ins-det, q-bio.OT
Results of Positron Annihilation Lifetime Spectroscopy (PALS) and microscopic studies on simple microorganisms: brewing yeasts are presented. Lifetime of ortho - positronium (o-Ps) were found to change from 2.4 to 2.9 ns (longer lived component) for lyophilised and aqueous yeasts, respectively. Also hygroscopicity of yeasts in time was examined, allowing to check how water - the main component of the cell - affects PALS parameters, thus lifetime of o-Ps were found to change from 1.2 to 1.4 ns (shorter lived component) for the dried yeasts. The time sufficient to hydrate the cells was found below 10 hours. In the presence of liquid water an indication of reorganization of yeast in the molecular scale was observed. Microscopic images of the lyophilised, dried and wet yeasts with best possible resolution were obtained using Inverted Microscopy (IM) and Environmental Scanning Electron Microscopy (ESEM) methods. As a result visible changes to the surface of the cell membrane were observed in ESEM images.
• ### PALS investigations of free volumes thermal expansion of J-PET plastic scintillator synthesized in polystyrene matrix(1508.06820)

Aug. 27, 2015 nucl-ex, physics.ins-det
The polystyrene dopped with 2,5-diphenyloxazole as a primary fluor and 2-(4-styrylphenyl)benzoxazole as a wavelength shifter, prepared as a plastic scintillator was investigated using positronium probe in wide range of temperatures from 123 to 423 K. Three structural transitions at 260 K, 283 K and 370 K were found in the material. In the o-Ps intensity dependence on temperature, the significant hysteresis is observed. Heated to 370 K, the material exhibits the o-Ps intensity variations in time.
• ### Processing optimization with parallel computing for the J-PET tomography scanner(1508.02751)

The Jagiellonian-PET (J-PET) collaboration is developing a prototype TOF-PET detector based on long polymer scintillators. This novel approach exploits the excellent time properties of the plastic scintillators, which permit very precise time measurements. The very fast, FPGA-based front-end electronics and the data acquisition system, as well as, low- and high-level reconstruction algorithms were specially developed to be used with the J-PET scanner. The TOF-PET data processing and reconstruction are time and resource demanding operations, especially in case of a large acceptance detector, which works in triggerless data acquisition mode. In this article, we discuss the parallel computing methods applied to optimize the data processing for the J-PET detector. We begin with general concepts of parallel computing and then we discuss several applications of those techniques in the J-PET data processing.
• ### Application of the Compress Sensing Theory for Improvement of the TOF Resolution in a Novel J-PET Instrument(1507.03799)

Aug. 8, 2015 hep-ex, physics.ins-det
Nowadays, in Positron Emission Tomography (PET) systems, a Time of Flight information is used to improve the image reconstruction process. In Time of Flight PET (TOF-PET), fast detectors are able to measure the difference in the arrival time of the two gamma rays, with the precision enabling to shorten significantly a range along the line-of-response (LOR) where the annihilation occurred. In the new concept, called J-PET scanner, gamma rays are detected in plastic scintillators. In a single strip of J-PET system, time values are obtained by probing signals in the amplitude domain. Owing to Compress Sensing theory, information about the shape and amplitude of the signals is recovered. In this paper we demonstrate that based on the acquired signals parameters, a better signal normalization may be provided in order to improve the TOF resolution. The procedure was tested using large sample of data registered by a dedicated detection setup enabling sampling of signals with 50 ps intervals. Experimental setup provided irradiation of a chosen position in the plastic scintillator strip with annihilation gamma quanta.
• ### GPU accelerated image reconstruction in a two-strip J-PET tomograph(1502.07478)

Aug. 6, 2015 nucl-ex, physics.ins-det
We present a fast GPU implementation of the image reconstruction routine, for a novel two strip PET detector that relies solely on the time of flight measurements.
• ### A pilot study of the novel J-PET plastic scintillator with 2-(4-styrylphenyl)benzoxazole as a wavelength shifter(1502.02901)

July 4, 2015 nucl-ex, physics.ins-det
For the first time a molecule of 2-(4-styrylphenyl)benzoxazole containing benzoxazole and stilbene groups is applied as a scintillator dopant acting as a wavelength shifter. In this article a light yield of the plastic scintillator, prepared from styrene doped with 2 wt% of 2,5-diphenylbenzoxazole and 0.03 wt% of 2-(4-styrylphenyl)benzoxazole, is determined to be as large as 60% $\pm$ 2% of the anthracene light output. There is a potential to improve this value in the future by the optimization of the additives concentrations.
• ### Compressive Sensing of Signals Generated in Plastic Scintillators in a Novel J-PET Instrument(1503.05188)

April 4, 2015 physics.ins-det
The J-PET scanner, which allows for single bed imaging of the whole human body, is currently under development at the Jagiellonian University. The dis- cussed detector offers improvement of the Time of Flight (TOF) resolution due to the use of fast plastic scintillators and dedicated electronics allowing for sam- pling in the voltage domain of signals with durations of few nanoseconds. In this paper we show that recovery of the whole signal, based on only a few samples, is possible. In order to do that, we incorporate the training signals into the Tikhonov regularization framework and we perform the Principal Component Analysis decomposition, which is well known for its compaction properties. The method yields a simple closed form analytical solution that does not require iter- ative processing. Moreover, from the Bayes theory the properties of regularized solution, especially its covariance matrix, may be easily derived. This is the key to introduce and prove the formula for calculations of the signal recovery error. In this paper we show that an average recovery error is approximately inversely proportional to the number of acquired samples.
• ### Multiple scattering and accidental coincidences in the J-PET detector simulated using GATE package(1502.04532)

April 2, 2015 physics.ins-det
Novel Positron Emission Tomography system, based on plastic scintillators, is developed by the J-PET collaboration. In order to optimize geometrical configuration of built device, advanced computer simulations are performed. Detailed study is presented of background given by accidental coincidences and multiple scattering of gamma quanta.
• ### A novel method for the line-of-response and time-of-flight reconstruction in TOF-PET detectors based on a library of synchronized model signals(1412.6963)

Feb. 10, 2015 nucl-ex, physics.ins-det
A novel method of hit time and hit position reconstruction in scintillator detectors is described. The method is based on comparison of detector signals with results stored in a library of synchronized model signals registered for a set of well-defined positions of scintillation points. The hit position is reconstructed as the one corresponding to the signal from the library which is most similar to the measurement signal. The time of the interaction is determined as a relative time between the measured signal and the most similar one in the library. A degree of similarity of measured and model signals is defined as the distance between points representing the measurement- and model-signal in the multi-dimensional measurement space. Novelty of the method lies also in the proposed way of synchronization of model signals enabling direct determination of the difference between time-of-flights (TOF) of annihilation quanta from the annihilation point to the detectors. The introduced method was validated using experimental data obtained by means of the double strip prototype of the J-PET detector and $^{22}$Na sodium isotope as a source of annihilation gamma quanta.The detector was built out from plastic scintillator strips with dimensions of 5 mm x 19 mm x 300 mm, optically connected at both sides to photomultipliers,from which signals were sampled by means of the Serial Data Analyzer.Using the introduced method, the spatial and TOF resolution of about 1.3 cm ($\sigma$) and 125 ps ($\sigma$) were established, respectively.
• ### Test of a single module of the J-PET scanner based on plastic scintillators(1407.7395)

Time of Flight Positron Emission Tomography scanner based on plastic scintillators is being developed at the Jagiellonian University by the J-PET collaboration. The main challenge of the conducted research lies in the elaboration of a method allowing application of plastic scintillators for the detection of low energy gamma quanta. In this article we report on tests of a single detection module built out from BC-420 plastic scintillator strip (with dimensions of 5x19x300mm^3) read out at two ends by Hamamatsu R5320 photomultipliers. The measurements were performed using collimated beam of annihilation quanta from the 68Ge isotope and applying the Serial Data Analyzer (Lecroy SDA6000A) which enabled sampling of signals with 50ps intervals. The time resolution of the prototype module was established to be better than 80ps (sigma) for a single level discrimination. The spatial resolution of the determination of the hit position along the strip was determined to be about 0.93cm (sigma) for the annihilation quanta. The fractional energy resolution for the energy E deposited by the annihilation quanta via the Compton scattering amounts to sigma(E)/E = 0.044/sqrt(E[MeV]) and corresponds to the sigma(E)/E of 7.5% at the Compton edge.
• ### Novel method for hit-position reconstruction using voltage signals in plastic scintillators and its application to Positron Emission Tomography(1407.8293)

July 31, 2014 hep-ex, physics.ins-det
Currently inorganic scintillator detectors are used in all commercial Time of Flight Positron Emission Tomograph (TOF-PET) devices. The J-PET collaboration investigates a possibility of construction of a PET scanner from plastic scintillators which would allow for single bed imaging of the whole human body. This paper describes a novel method of hit-position reconstruction based on sampled signals and an example of an application of the method for a single module with a 30 cm long plastic strip, read out on both ends by Hamamatsu R4998 photomultipliers. The sampling scheme to generate a vector with samples of a PET event waveform with respect to four user-defined amplitudes is introduced. The experimental setup provides irradiation of a chosen position in the plastic scintillator strip with an annihilation gamma quanta of energy 511~keV. The statistical test for a multivariate normal (MVN) distribution of measured vectors at a given position is developed, and it is shown that signals sampled at four thresholds in a voltage domain are approximately normally distributed variables. With the presented method of a vector analysis made out of waveform samples acquired with four thresholds, we obtain a spatial resolution of about 1 cm and a timing resolution of about 80 ps
• ### A novel method for calibration and monitoring of time synchronization of TOF-PET scanners by means of cosmic rays(1311.6152)

All of the present methods for calibration and monitoring of TOF-PET scanner detectors utilize radioactive isotopes such as e.g. $^{22}$Na or $^{68}$Ge, which are placed or rotate inside the scanner. In this article we describe a novel method based on the cosmic rays application to the PET calibration and monitoring methods. The concept allows to overcome many of the drawbacks of the present methods and it is well suited for newly developed TOF-PET scanners with a large longitudinal field of view. The method enables also monitoring of the quality of the scintillator materials and in general allows for the continuous quality assurance of the PET detector performance.
• ### J-PET analysis framework for the prototype TOF-PET detector(1311.6153)

Novel TOF-PET scanner solutions demand, apart from the state of the art detectors, software for fast processing of the gathered data, monitoring of the whole scanner and reconstruction of the PET image. In this article we present an analysis framework for the novel STRIP-PET scanner developed by the J-PET collaboration in the Institute of Physics of the Jagiellonian University. This software is based on the ROOT package used in many particle physics experiments.
• ### Determination of the map of efficiency of the J-PET detector with the GATE package(1402.6840)

Feb. 27, 2014 physics.ins-det
A novel PET detector consisting of strips of polymer scintillators is being developed by the J-PET Collaboration. The map of efficiency and the map of geometrical acceptance of the 2-strip J-PET scanner are presented. Map of efficiency was determined using the Monte Carlo simulation software GATE based on GEANT4. Both maps were compared using method based on the chi2 test.
• ### Computing support for advanced medical data analysis and imaging(1401.6929)

We discuss computing issues for data analysis and image reconstruction of PET-TOF medical scanner or other medical scanning devices producing large volumes of data. Service architecture based on the grid and cloud concepts for distributed processing is proposed and critically discussed.
• ### Calibration of photomultipliers gain used in the J-PET detector(1312.2744)

Photomultipliers are commonly used in commercial PET scanner as devices which convert light produced in scintillator by gamma quanta from positron-electron annihilation into electrical signal. For proper analysis of obtained electrical signal, a photomultiplier gain curve must be known, since gain can be significantly different even between photomultipliers of the same model. In this article we describe single photoelectron method used for photomultipliers calibration applied for J-PET scanner, a novel PET detector being developed at the Jagiellonian University. Description of calibration method, an example of calibration curve and gain of few R4998 Hamamatsu photomultipliers are presented.
• ### Simulations of gamma quanta scattering in a single module of the J-PET detector(1312.0250)

This article describes simulations of scattering of annihilation gamma quanta in a strip of plastic scintillator. Such strips constitute basic detection modules in a newly proposed Positron Emission Tomography which utilizes plastic scintillators instead of inorganic crystals. An algorithm simulating chain of Compton scatterings was elaborated and series of simulations have been conducted for the scintillator strip with the cross section of 5 mm x 19 mm. Obtained results indicate that secondary interactions occur only in the case of about 8% of events and out of them only 25$\%$ take place in the distance larger than 0.5 cm from the primary interaction. It was also established that light signals produced at primary and secondary interactions overlap with the delay which distribution is characterized by FWHM of about 40 ps.
• ### A novel method based solely on FPGA units enabling measurement of time and charge of analog signals in Positron Emission Tomography(1311.6127)

This article presents a novel technique for precise measurement of time and charge based solely on FPGA (Field Programmable Gate Array) device and few satellite discrete electronic components used in Positron Emission Tomography (PET). Described approach simplifies electronic circuits, reduces the power consumption, lowers costs, merges front-end electronics with digital electronics and also makes more compact final design. Furthermore, it allows to measure time when analog signals cross a reference voltage at different threshold levels with a very high precision of $\sim$ 10ps (rms) and thus enables sampling of signals in a voltage domain.