• ### Measurement and analysis of the Am-243 neutron capture cross section at the n_TOF facility at CERN(1412.1707)

Dec. 4, 2014 nucl-ex
Background:The design of new nuclear reactors and transmutation devices requires to reduce the present neutron cross section uncertainties of minor actinides. Purpose: Reduce the $^{243}$Am(n,$\gamma$) cross section uncertainty. Method: The $^{243}$Am(n,$\gamma$) cross section has been measured at the n_TOF facility at CERN with a BaF$_{2}$ Total Absorption Calorimeter, in the energy range between 0.7 eV and 2.5 keV. Results: The $^{243}$Am(n,$\gamma$) cross section has been successfully measured in the mentioned energy range. The resolved resonance region has been extended from 250 eV up to 400 eV. In the unresolved resonance region our results are compatible with one of the two incompatible capture data sets available below 2.5 keV. The data available in EXFOR and in the literature has been used to perform a simple analysis above 2.5 keV. Conclusions: The results of this measurement contribute to reduce the $^{243}$Am(n,$\gamma$) cross section uncertainty and suggest that this cross section is underestimated up to 25% in the neutron energy range between 50 eV and a few keV in the present evaluated data libraries.
• ### A Study of the Operation of Especially Designed Photosensitive Gaseous Detectors at Cryogenic Temperatures(physics/0512007)

Dec. 1, 2005 physics.ins-det
In some experiments and applications there is need for large-area photosensitive detectors to operate at cryogenic temperatures. Nowadays, vacuum PMs are usually used for this purpose. We have developed special designs of planar photosensitive gaseous detectors able to operate at cryogenic temperatures. Such detectors are much cheaper PMs and are almost insensitive to magnetic fields. Results of systematic measurements of their quantum efficiencies, the maximum achievable gains and long-term stabilities will be presented. The successful operation of these detectors open realistic possibilities in replacing PMs by photosensitive gaseous detectors in some applications dealing with cryogenic liquids; for example in experiments using noble liquid TPCs or noble liquid scintillating calorimeters.
• ### A High Position Resolution X-ray Detector: an Edge on Illuminated Capillary Plate Combined with a Gas Amplification Structure(physics/0411082)

Nov. 8, 2004 physics.ins-det
We have developed and successfully tested a prototype of a new type of high position resolution hybrid X-ray detector. It contains a thin wall lead glass capillary plate converter of X-rays combined with a microgap parallel-plate avalanche chamber filled with gas at 1 atm. The operation of these converters was studied in a wide range of X-ray energies (from 6 to 60 keV) at incident angles varying from 0-90 degree. The detection efficiency, depending on the geometry, photon energy, incident angle and the mode of operation, was between 5-30 percent in a single step mode and up to 50 percent in a multi-layered combination. Depending on the capillary geometry, the position resolution achieved was between 0.050-0.250 mm in digital form and was practically independent of the photon energy or gas mixture. The usual lead glass capillary plates operated without noticeable charging up effects at counting rates of 50 Hz/mm2, and hydrogen treated capillaries up to 10E5 Hz/mm2. The developed detector may open new possibilities for medical imaging, for example in mammography, portal imaging, radiography (including security devices), crystallography and many other applications.
• ### The Successful Operation of Hole-type Gaseous Detectors at Cryogenic Temperatures(physics/0410280)

Oct. 29, 2004 physics.ins-det
We have demonstrated that hole-type gaseous detectors, GEMs and capillary plates, can operate up to 77 K. For example, a single capillary plate can operate at gains of above 10E3 in the entire temperature interval between 300 until 77 K. The same capillary plate combined with CsI photocathodes could operate perfectly well at gains (depending on gas mixtures) of 100-1000. Obtained results may open new fields of applications for capillary plates as detectors of UV light and charge particles at cryogenic temperatures: noble liquid TPCs, WIMP detectors or LXe scintillating calorimeters and cryogenic PETs.
• ### Evaluation of various planar gaseous detectors with CsI photocathodes for the detection of primary scintillation light from noble gases(physics/0106070)

June 21, 2001 physics.ins-det
Noble gases and liquids are excellent scintillators and this opens a unique opportunity to directly detect the primary scintillation light produced in these media by photons or particles. This signal can be used for several purposes, for example as a start signal for TPCs or for particles identification. Usually photomultipliers (PMs) are used for the detection of the scintillation light. In our previous work we have demonstrated that costly PMs could be replaced by gaseous detectors with CsI photocathodes . Such detectors have the same quantum efficiency as the best PMs but at the same time are cheap, simple and have high position and time resolutions. The aim of this work is to evaluate various planar type gaseous detectors with CsI photocahodes in order to choose the best one for the detection of the primary scintillation light from noble gases and liquids.
• ### Direct T-violation measurements and T-odd effects in decay experiments(hep-ph/9903458)

March 23, 1999 hep-ph
Motivated by the recent experimental announcements for direct measurements of time-reversal non-invariance in the neutral kaon system, we make a comparative discussion of the CPLEAR and KTeV measurements. The most suitable way to consistently incorporate the mixing, the time evolution and the decays of kaons, is to describe the neutral kaon system as a system with a non-Hermitean Hamiltonian. In this framework, the physical (decaying) incoming and outgoing states are distinct and belong to dual spaces. Moreover, since they are eigenstates of the full Hamiltonian, they never oscillate. This is directly manifest in the orthogonality conditions of the physical states, which entirely determine the evolution of the kaon system. Along these lines we conclude: CPLEAR studies K0-bar{K0} oscillations, a process where initial and final states can be reversed, the CPLEAR asymmetry being an effect directly related to the definition of time-reversal. Conclusively, CPLEAR provides a direct measurement of T-violation without any assumption either on unitarity or on CPT-invariance. The KTeV experiment studies in particular the process KL -> pi+ pi- e+ e- where they measure a T-odd effect. However, using unitarity together with estimates of the final state interactions, it should be possible to determine whether this effect can be identified with a genuine T-reversal violation.