• During the shutdown of the CERN Large Hadron Collider in 2013-2014, an additional pixel layer was installed between the existing Pixel detector of the ATLAS experiment and a new, smaller radius beam pipe. The motivation for this new pixel layer, the Insertable B-Layer (IBL), was to maintain or improve the robustness and performance of the ATLAS tracking system, given the higher instantaneous and integrated luminosities realised following the shutdown. Because of the extreme radiation and collision rate environment, several new radiation-tolerant sensor and electronic technologies were utilised for this layer. This paper reports on the IBL construction and integration prior to its operation in the ATLAS detector.
• ### Fiber Bragg Grating (FBG) sensors as flatness and mechanical stretching sensors(1512.08481)

Dec. 28, 2015 physics.ins-det
A novel approach which uses Fibre Bragg Grating (FBG) sensors has been utilised to assess and monitor the flatness of Gaseous Electron Multipliers (GEM) foils. The setup layout and preliminary results are presented.
• ### A novel application of Fiber Bragg Grating (FBG) sensors in MPGD(1512.08529)

Dec. 28, 2015 physics.ins-det
We present a novel application of Fiber Bragg Grating (FBG) sensors in the construction and characterisation of Micro Pattern Gaseous Detector (MPGD), with particular attention to the realisation of the largest triple (Gas electron Multiplier) GEM chambers so far operated, the GE1/1 chambers of the CMS experiment at LHC. The GE1/1 CMS project consists of 144 GEM chambers of about 0.5 m2 active area each, employing three GEM foils per chamber, to be installed in the forward region of the CMS endcap during the long shutdown of LHC in 2108-2019. The large active area of each GE1/1 chamber consists of GEM foils that are mechanically stretched in order to secure their flatness and the consequent uniform performance of the GE1/1 chamber across its whole active surface. So far FBGs have been used in high energy physics mainly as high precision positioning and re-positioning sensors and as low cost, easy to mount, low space consuming temperature sensors. FBGs are also commonly used for very precise strain measurements in material studies. In this work we present a novel use of FBGs as flatness and mechanical tensioning sensors applied to the wide GEM foils of the GE1/1 chambers. A network of FBG sensors have been used to determine the optimal mechanical tension applied and to characterise the mechanical tension that should be applied to the foils. We discuss the results of the test done on a full-sized GE1/1 final prototype, the studies done to fully characterise the GEM material, how this information was used to define a standard assembly procedure and possible future developments.
• ### Charge Transfer Properties Through Graphene for Applications in Gaseous Detectors(1512.05409)

Dec. 16, 2015 hep-ex, physics.ins-det
Graphene is a single layer of carbon atoms arranged in a honeycomb lattice with remarkable mechanical and electrical properties. Regarded as the thinnest and narrowest conductive mesh, it has drastically different transmission behaviours when bombarded with electrons and ions in vacuum. This property, if confirmed in gas, may be a definitive solution for the ion back-flow problem in gaseous detectors. In order to ascertain this aspect, graphene layers of dimensions of about 2x2cm$^2$, grown on a copper substrate, are transferred onto a flat metal surface with holes, so that the graphene layer is freely suspended. The graphene and the support are installed into a gaseous detector equipped with a triple Gaseous Electron Multiplier (GEM), and the transparency properties to electrons and ions are studied in gas as a function of the electric fields. The techniques to produce the graphene samples are described, and we report on preliminary tests of graphene-coated GEMs.
• ### Performance of a Large-Area GEM Detector Prototype for the Upgrade of the CMS Muon Endcap System(1412.0228)

Dec. 8, 2014 physics.ins-det
Gas Electron Multiplier (GEM) technology is being considered for the forward muon upgrade of the CMS experiment in Phase 2 of the CERN LHC. Its first implementation is planned for the GE1/1 system in the $1.5 < \mid\eta\mid < 2.2$ region of the muon endcap mainly to control muon level-1 trigger rates after the second long LHC shutdown. A GE1/1 triple-GEM detector is read out by 3,072 radial strips with 455 $\mu$rad pitch arranged in eight $\eta$-sectors. We assembled a full-size GE1/1 prototype of 1m length at Florida Tech and tested it in 20-120 GeV hadron beams at Fermilab using Ar/CO$_{2}$ 70:30 and the RD51 scalable readout system. Four small GEM detectors with 2-D readout and an average measured azimuthal resolution of 36 $\mu$rad provided precise reference tracks. Construction of this largest GEM detector built to-date is described. Strip cluster parameters, detection efficiency, and spatial resolution are studied with position and high voltage scans. The plateau detection efficiency is [97.1 $\pm$ 0.2 (stat)]\%. The azimuthal resolution is found to be [123.5 $\pm$ 1.6 (stat)] $\mu$rad when operating in the center of the efficiency plateau and using full pulse height information. The resolution can be slightly improved by $\sim$ 10 $\mu$rad when correcting for the bias due to discrete readout strips. The CMS upgrade design calls for readout electronics with binary hit output. When strip clusters are formed correspondingly without charge-weighting and with fixed hit thresholds, a position resolution of [136.8 $\pm$ 2.5 stat] $\mu$rad is measured, consistent with the expected resolution of strip-pitch/$\sqrt{12}$ = 131.3 $\mu$rad. Other $\eta$-sectors of the detector show similar response and performance.
• ### The Resistive-WELL detector: a compact spark-protected single amplification-stage MPGD(1411.2466)

Nov. 10, 2014 hep-ex, physics.ins-det
In this work we present a novel idea for a compact spark-protected single amplification stage Micro-Pattern Gas Detector (MPGD). The detector amplification stage, realized with a structure very similar to a GEM foil, is embedded through a resistive layer in the readout board. A cathode electrode, defining the gas conversion/drift gap, completes the detector mechanics. The new structure, that we call Resistive-WELL (R-WELL), has some characteristics in common with previous MPGDs, such as C.A.T. and WELL, developed more than ten years ago. The prototype object of the present study has been realized in the 2009 by TE-MPE-EM Workshop at CERN. The new architecture is a very compact MPGD, robust against discharges and exhibiting a large gain ($\sim$6$\times$10$^3$), simple to construct and easy for engineering and then suitable for large area tracking devices as well as huge calorimetric apparata.
• ### Demonstration of new possibilities of multilayer technology on resistive microstrip/ microdot detectors(1406.6914)

June 26, 2014 physics.ins-det
The first successful attempts to optimize the electric field in Resistive Microstrip Gas Chamber and resistive microdot detectors using additional field shaping strips located inside the detector substrate are described
• ### Clifford Algebra and Space-Time Transformations: Lorentz Transformation and Inertial Transformation(1311.6378)

Nov. 25, 2013 gr-qc
We review the Inertial transformation and Lorentz transformation under a new context, by using Clifford Algebra or Geometric Algebra. The apparent contradiction between theses two approach is simply stems from different procedures for clock synchronization associated with different choices of the coordinates used to describe the physical world. We find the physical and coordinates components of both transformations. A important result is that in the case of Inertial transformation the physical components are exactly the Galilean transformations, but the speed of light is not c. Another interesting result is due to the fact the Lorentz transformations lead directly to physical components, and this case the speed of light is c. Finally e show that both scenarios, de-synchronization Einstein theory and synchronized theory, are all mathematically equivalent by means of Clifford Algebra Transformations.
• ### Piggyback resistive Micromegas(1310.1242)

Oct. 4, 2013 physics.ins-det
Piggyback Micromegas consists in a novel readout architecture where the anode element is made of a resistive layer on a ceramic substrate. The resistive layer is deposited on the thin ceramic substrate by an industrial process which provides large dynamic range of resistivity (10$^6$ to 10$^{10}$\,M$\Omega$/square). The particularity of this new structure is that the active part is entirely dissociated from the read-out element. This gives a large flexibility on the design of the anode structure and the readout scheme. Without significant loss, signals are transmitted by capacitive coupling to the read-out pads. The detector provides high gas gain, good energy resolution and the resistive layer assures spark protection for the electronics. This assembly could be combined with modern pixel array electronic ASICs. First tests with different Piggyback detectors and configurations will be presented. This structure is adequate for cost effective fabrication and low outgassing detectors. It was designed to perform in sealed mode and its long term stability has been extensively studied. In addition perspectives on the future developments will be evoked.
• ### Ageing studies of resistive Micromegas detectors for the HL-LHC(1304.2053)

April 7, 2013 hep-ex, physics.ins-det
Resistive-anode Micromegas detectors are in development since several years, in an effort to solve the problem of sparks when working in high flux and high radiations environment like in the HL-LHC (ten times the luminosity of the LHC). They have been chosen as one of the technologies that will be part of the ATLAS New Small Wheel project (forward muon system). An ageing study is mandatory to assess their capabilities to handle the HL-LHC environment on a long-term period. A prototype has been exposed to several types of irradiations (X-rays, cold neutrons, 60 Co gammas) up to an equivalent HL-LHC time of more than five years without showing any degradation of the performances in terms of gain and energy resolution. Beam test studies took place in October 2012 to assess the tracking performances (efficiency, spatial resolution,...). Results of ageing studies and beam test performances are reported in this paper.
• ### A Piggyback resistive Micromegas(1208.6525)

Feb. 5, 2013 physics.ins-det
A novel read-out architecture has been developed for the Micromegas detector. The anode element is made of a resistive layer on a ceramic substrate. The detector part is entirely separated from the read-out element. Without significant loss, signals are transmitted by capacitive coupling to the read-out pads. The detector provides high gas gain, good energy resolution and the resistive layer assures spark protection to the electronics. This assembly could be combined with modern pixel array electronic ASICs. This readout organization is free on how the pixels are designed, arranged and connected. We present first results taken with a MediPix read-out chip.
• ### An ageing study of resistive micromegas for the HL-LHC environment(1301.7648)

Resistive-anode micromegas detectors are in development since several years, in an effort to solve the problem of sparks when working at high flux and high ionizing radiation like in the HL-LHC (up to ten times the luminosity of the LHC). They have been chosen as one of the technologies that will be part of the ATLAS New Small Wheel project (forward muon system). An ageing study is mandatory to assess their capabilities to handle the HL-LHC environment on a long-term period. A prototype has been exposed to several types of irradiation (X-rays, cold neutrons, $^{60}$Co gammas and alphas) above the equivalent charge produced at the detector in five HL-LHC running years without showing any degradation of the performances in terms of gain and energy resolution. This study has been completed with the characterization of the tracking performances in terms of efficiency and spatial resolution, verifying the compatibility of results obtained with both resistive micromegas detectors, irradiated and non-irradiated one.
• ### Beam Test Results for New Full-scale GEM Prototypes for a Future Upgrade of the CMS High-eta Muon System(1211.3939)

Nov. 16, 2012 physics.ins-det
The CMS GEM collaboration is considering Gas Electron Multipliers (GEMs) for upgrading the CMS forward muon system in the 1.5<|eta|<2.4 endcap region. GEM detectors can provide precision tracking and fast trigger information. They would improve the CMS muon trigger and muon momentum resolution and provide missing redundancy in the high-eta region. Employing a new faster construction and assembly technique, we built four full-scale Triple-GEM muon detectors for the inner ring of the first muon endcap station. We plan to install these or further improved versions in CMS during the first long LHC shutdown in 2013/14 for continued testing. These detectors are designed for the stringent rate and resolution requirements in the increasingly hostile environments expected at CMS after the second long LHC shutdown in 2018/19. The new prototypes were studied in muon/pion beams at the CERN SPS. We discuss our experience with constructing the new full-scale production prototypes and present preliminary performance results from the beam test. We also tested smaller Triple-GEM prototypes with zigzag readout strips with 2 mm pitch in these beams and measured a spatial resolution of 73 microns. This readout offers a potential reduction of channel count and consequently electronics cost for this system while maintaining high spatial resolution.
• ### Construction and Performance of Large-Area Triple-GEM Prototypes for Future Upgrades of the CMS Forward Muon System(1111.7249)

Nov. 30, 2011 hep-ex, physics.ins-det
At present, part of the forward RPC muon system of the CMS detector at the CERN LHC remains uninstrumented in the high-\eta region. An international collaboration is investigating the possibility of covering the 1.6 < |\eta| < 2.4 region of the muon endcaps with large-area triple-GEM detectors. Given their good spatial resolution, high rate capability, and radiation hardness, these micro-pattern gas detectors are an appealing option for simultaneously enhancing muon tracking and triggering capabilities in a future upgrade of the CMS detector. A general overview of this feasibility study will be presented. The design and construction of small (10\times10 cm2) and full-size trapezoidal (1\times0.5 m2) triple-GEM prototypes will be described. During detector assembly, different techniques for stretching the GEM foils were tested. Results from measurements with x-rays and from test beam campaigns at the CERN SPS will be shown for the small and large prototypes. Preliminary simulation studies on the expected muon reconstruction and trigger performances of this proposed upgraded muon system will be reported.
• ### Test beam results of the GE1/1 prototype for a future upgrade of the CMS high-$\eta$ muon system(1111.4883)

Nov. 30, 2011 hep-ex, physics.ins-det
Gas Electron Multipliers (GEM) are an interesting technology under consideration for the future upgrade of the forward region of the CMS muon system, specifically in the $1.6<| \eta |<2.4$ endcap region. With a sufficiently fine segmentation GEMs can provide precision tracking as well as fast trigger information. The main objective is to contribute to the improvement of the CMS muon trigger. The construction of large-area GEM detectors is challenging both from the technological and production aspects. In view of the CMS upgrade we have designed and built the largest full-size Triple-GEM muon detector, which is able to meet the stringent requirements given the hostile environment at the high-luminosity LHC. Measurements were performed during several test beam campaigns at the CERN SPS in 2010 and 2011. The main issues under study are efficiency, spatial resolution and timing performance with different inter-electrode gap configurations and gas mixtures. In this paper results of the performance of the prototypes at the beam tests will be discussed.
• ### Aging studies of Micromegas prototypes for the HL-LHC(1111.5998)

Nov. 25, 2011 hep-ex, physics.ins-det
The micromegas technology is a promising candidate to replace the forward muon chambers for the luminosity upgrade of ATLAS. The LHC accelerator luminosity will be five times the nominal one, increasing background and pile-up event probability. This requires detector performances which are currently under study in intensive R&D activities. Aging is one of the key issues for a high-luminosity LHC application. For this reason, we study the properties of resistive micromegas detectors under intense X-ray radiation and under thermal neutrons in different CEA-Saclay facilities. This study is complementary to those already performed using fast neutrons.
• ### A spark-resistant bulk-micromegas chamber for high-rate applications(1011.5370)

Nov. 24, 2010 hep-ex, physics.ins-det
We report on the design and performance of a spark-resistant bulk-micromegas chamber. The principle of this design lends itself to the construction of large-area muon chambers for the upgrade of the detectors at the Large Hadron Collider at CERN for luminosities in excess of 10**34/cm2/s or other high-rate applications.
• ### Radiopurity of Micromegas readout planes(1005.2022)

Aug. 2, 2010 nucl-ex, physics.ins-det
Micromesh Gas Amplification Structures (Micromegas) are being used in an increasing number of Particle Physics applications since their conception fourteen years ago. More recently, they are being used or considered as readout of Time Projection Chambers (TPCs) in the field of Rare Event searches (dealing with dark matter, axions or double beta decay). In these experiments, the radiopurity of the detector components and surrounding materials is measured and finely controlled in order to keep the experimental background as low as possible. In the present paper, the first measurement of the radiopurity of Micromegas planes obtained by high purity germanium spectrometry in the low background facilities of the Canfranc Underground Laboratory (LSC) is presented. The obtained results prove that Micromegas readouts of the microbulk type are currently manufactured with radiopurity levels below 30 microBq/cm2 for Th and U chains and ~60 microBq/cm2 for 40K, already comparable to the cleanest detector components of the most stringent low background experiments at present. Taking into account that the studied readouts were manufactured without any specific control of the radiopurity, it should be possible to improve these levels after dedicated development.
• ### Micromegas in a Bulk(physics/0501003)

Jan. 2, 2005 physics.ins-det
In this paper we present a novel way to manufacture the bulk Micromegas detector. A simple process based on the PCB (Printed Circuit Board) technology is employed to produce the entire sensitive detector. Such fabrication process could be extended to very large area detectors made by the industry. The low cost fabrication together with the robustness of the electrode materials will make it extremely attractive for several applications ranging from particle physics and astrophysics to medicine
• ### An insulating grid spacer for large-area MICROMEGAS chambers(physics/0103027)

March 9, 2001 physics.ins-det
We present an original design for large area gaseous detectors based on the MICROMEGAS technology. This technology incorporates an insulating grid, sandwiched between the micro-mesh and the anode-pad plane, which provides an uniform 200 $\mu$m amplification gap. The uniformity of the amplification gap thickness has been verified under several experimental conditions. The gain performances of the detector are presented and compared to the values obtained with detectors using cylindrical micro spacers. The new design presents several technical and financial advantages.