• A new pixel detector for the CMS experiment was built in order to cope with the instantaneous luminosities anticipated for the Phase~I Upgrade of the LHC. The new CMS pixel detector provides four-hit tracking with a reduced material budget as well as new cooling and powering schemes. A new front-end readout chip mitigates buffering and bandwidth limitations, and allows operation at low comparator thresholds. In this paper, comprehensive test beam studies are presented, which have been conducted to verify the design and to quantify the performance of the new detector assemblies in terms of tracking efficiency and spatial resolution. Under optimal conditions, the tracking efficiency is $99.95\pm0.05\,\%$, while the intrinsic spatial resolutions are $4.80\pm0.25\,\mu \mathrm{m}$ and $7.99\pm0.21\,\mu \mathrm{m}$ along the $100\,\mu \mathrm{m}$ and $150\,\mu \mathrm{m}$ pixel pitch, respectively. The findings are compared to a detailed Monte Carlo simulation of the pixel detector and good agreement is found.
• The degradation of signal in silicon sensors is studied under conditions expected at the CERN High-Luminosity LHC. 200 $\mu$m thick n-type silicon sensors are irradiated with protons of different energies to fluences of up to $3 \cdot 10^{15}$ neq/cm$^2$. Pulsed red laser light with a wavelength of 672 nm is used to generate electron-hole pairs in the sensors. The induced signals are used to determine the charge collection efficiencies separately for electrons and holes drifting through the sensor. The effective trapping rates are extracted by comparing the results to simulation. The electric field is simulated using Synopsys device simulation assuming two effective defects. The generation and drift of charge carriers are simulated in an independent simulation based on PixelAV. The effective trapping rates are determined from the measured charge collection efficiencies and the simulated and measured time-resolved current pulses are compared. The effective trapping rates determined for both electrons and holes are about 50% smaller than those obtained using standard extrapolations of studies at low fluences and suggests an improved tracker performance over initial expectations.
• ### The STAR Vertex Position Detector(1403.6855)

March 26, 2014 nucl-ex, physics.ins-det
The 2x3 channel pseudo Vertex Position Detector (pVPD) in the STAR experiment at RHIC has been upgraded to a 2x19 channel detector in the same acceptance, called the Vertex Position Detector (VPD). This detector is fully integrated into the STAR trigger system and provides the primary input to the minimum-bias trigger in Au+Au collisions. The information from the detector is used both in the STAR Level-0 trigger and offline to measure the location of the primary collision vertex along the beam pipe and the event "start time" needed by other fast-timing detectors in STAR. The offline timing resolution of single detector channels in full-energy Au+Au collisions is ~100 ps, resulting in a start time resolution of a few tens of picoseconds and a resolution on the primary vertex location of ~1 cm.
• ### Calibration and performance of the STAR Muon Telescope Detector using cosmic rays(1402.1078)

We report the timing and spatial resolution from the Muon Telescope Detector (MTD) installed in the STAR experiment at RHIC. Cosmic ray muons traversing the STAR detector have an average transverse momentum of 6 GeV/c. Due to their very small multiple scattering, these cosmic muons provide an ideal tool to calibrate the detectors and measure their timing and spatial resolution. The values obtained were ~100 ps and ~1-2 cm, respectively. These values are comparable to those obtained from cosmic-ray bench tests and test beams.
• ### Perspectives of a Midrapidity Dimuon Program at RHIC: A Novel and Compact Muon Telescope Detector(0904.3774)

July 17, 2009 nucl-ex
We propose a large-area, cost-effective Muon Telescope Detector (MTD) for the Solenoidal Tracker at RHIC (STAR) at mid-rapidity and for the next generation of detectors at a possible electron-ion collider. We utilize Multi-gap Resistive Plate Chambers with large modules and long readout strips (Long-MRPC) in the detector design. The results from cosmic ray and beam tests show the intrinsic timing and spatial resolution for a Long-MRPC are 60-70 ps and $\sim1$ cm, respectively. The prototype performance of such a novel muon telescope detector at STAR indicates that muon identification at the transverse momentum of a few GeV/$c$ can be achieved through the combined information of track matching with the MTD, ionization energy loss in the Time Projection Chamber, and time-of-flight measurements. A primary muon over secondary muon ratio of better than 1/3 can be achieved. This provides a promising device for future quarkonium programs and primordial dilepton measurements at RHIC. Simulations of the muon efficiency, the signal-to-background ratio of $J/\psi$, the separation of $\Upsilon$ 1S from 2S+3S states, and the electron-muon correlation from charm pair production in the RHIC environment are presented.
• ### Simple Front End Electronics for Multigap Resistive Plate Chambers(0807.3016)

July 18, 2008 physics.ins-det
A simple circuit for the presentation of the signals from Multi-gap Resistive Plate Chambers (MRPCs) to standard existing digitization electronics is described. The circuit is based on "off-the-shelf" discrete components. An optimization of the values of specific components is required to match the aspects of the MRPCs for the given application. This simple circuit is an attractive option for the initial signal processing for MRPC prototyping and bench- or beam-testing efforts, as well as for final implementations of small-area Time-of-Flight systems with existing data acquisition systems.
• ### The TOFp/pVPD Time of Flight System for STAR(nucl-ex/0308022)

Aug. 22, 2003 nucl-ex
A time-of-flight system was constructed for the STAR Experiment for the direct identification of hadrons produced in 197Au+197Au collisions at RHIC. The system consists of two separate detector subsystems, one called the pVPD (the "start" detector) and the other called the TOFp tray (the "stop" detector). Each detector is based on conventional scintillator/phototube technology and includes custom high-performance front-end electronics and a common CAMAC-based digitization and read-out. The design of the system and its performance during the 2001 RHIC run will be described. The start resolution attained by the pVPD was 24 ps, implying a pVPD single-detector resolution of 58 ps. The total time resolution of the system averaged over all detector channels was 87 ps, allowing direct pi/K/p discrimination for momenta up to 1.8 GeV/c, and direct (pi+K)/p discrimination up to 3 GeV/c.