• ### First measurement of elastic, inelastic and total cross-section at $\sqrt{s}=13$ TeV by TOTEM and overview of cross-section data at LHC energies(1712.06153)

Jan. 31, 2018 hep-ex
The TOTEM collaboration has measured the proton-proton total cross section at $\sqrt{s}=13$ TeV with a luminosity-independent method. Using dedicated $\beta^{*}=90$ m beam optics, the Roman Pots were inserted very close to the beam. The inelastic scattering rate has been measured by the T1 and T2 telescopes during the same LHC fill. After applying the optical theorem the total proton-proton cross section is $\sigma_{\rm tot}=(110.6 \pm 3.4$) mb, well in agreement with the extrapolation from lower energies. This method also allows one to derive the luminosity-independent elastic and inelastic cross sections: $\sigma_{\rm el} = (31.0 \pm 1.7)$ mb and $\sigma_{\rm inel} = (79.5 \pm 1.8)$ mb.
• ### Diamond Detectors for the TOTEM Timing Upgrade(1701.05227)

Jan. 18, 2017 hep-ex, physics.ins-det
This paper describes the design and the performance of the timing detector developed by the TOTEM Collaboration for the Roman Pots (RPs) to measure the Time-Of-Flight (TOF) of the protons produced in central diffractive interactions at the LHC. The measurement of the TOF of the protons allows the determination of the longitudinal position of the proton interaction vertex and its association with one of the vertices reconstructed by the CMS detectors. The TOF detector is based on single crystal Chemical Vapor Deposition (scCVD) diamond plates and is designed to measure the protons TOF with about 50 ps time precision. This upgrade to the TOTEM apparatus will be used in the LHC run 2 and will tag the central diffractive events up to an interaction pileup of about 1. A dedicated fast and low noise electronics for the signal amplification has been developed. The digitization of the diamond signal is performed by sampling the waveform. After introducing the physics studies that will most profit from the addition of these new detectors, we discuss in detail the optimization and the performance of the first TOF detector installed in the LHC in November 2015.
• ### Measurement of Elastic pp Scattering at $\sqrt{s}$ = 8 TeV in the Coulomb-Nuclear Interference Region - Determination of the $\rho$-Parameter and the Total Cross-Section(1610.00603)

Oct. 3, 2016 hep-ph, hep-ex, nucl-ex
The TOTEM experiment at the CERN LHC has measured elastic proton-proton scattering at the centre-of-mass energy $\sqrt{s}$ = 8 TeV and four-momentum transfers squared, |t|, from 6 x $10^{-4}$ GeV$^2$ to 0.2 GeV$^2$. Near the lower end of the |t|-interval the differential cross-section is sensitive to the interference between the hadronic and the electromagnetic scattering amplitudes. This article presents the elastic cross-section measurement and the constraints it imposes on the functional forms of the modulus and phase of the hadronic elastic amplitude. The data exclude the traditional Simplified West and Yennie interference formula that requires a constant phase and a purely exponential modulus of the hadronic amplitude. For parametrisations of the hadronic modulus with second- or third-order polynomials in the exponent, the data are compatible with hadronic phase functions giving either central or peripheral behaviour in the impact parameter picture of elastic scattering. In both cases, the $\rho$-parameter is found to be 0.12 $\pm$ 0.03. The results for the total hadronic cross-section are $\sigma_{tot}$ = (102.9 $\pm$ 2.3) mb and (103.0 $\pm$ 2.3) mb for central and peripheral phase formulations, respectively. Both are consistent with previous TOTEM measurements.
• ### Evidence for non-exponential elastic proton-proton differential cross-section at low |t| and sqrt(s) = 8 TeV by TOTEM(1503.08111)

Sept. 12, 2015 hep-ex
The TOTEM experiment has made a precise measurement of the elastic proton-proton differential cross-section at the centre-of-mass energy sqrt(s) = 8 TeV based on a high-statistics data sample obtained with the beta* = 90 optics. Both the statistical and systematic uncertainties remain below 1%, except for the t-independent contribution from the overall normalisation. This unprecedented precision allows to exclude a purely exponential differential cross-section in the range of four-momentum transfer squared 0.027 < |t| < 0.2 GeV^2 with a significance greater than 7 sigma. Two extended parametrisations, with quadratic and cubic polynomials in the exponent, are shown to be well compatible with the data. Using them for the differential cross-section extrapolation to t = 0, and further applying the optical theorem, yields total cross-section estimates of (101.5 +- 2.1) mb and (101.9 +- 2.1) mb, respectively, in agreement with previous TOTEM measurements.
• ### Measurement of the forward charged particle pseudorapidity density in pp collisions at sqrt(s) = 8 TeV using a displaced interaction point(1411.4963)

Nov. 18, 2014 hep-ex
The pseudorapidity density of charged particles dN(ch)/deta is measured by the TOTEM experiment in pp collisions at sqrt(s) = 8 TeV within the range 3.9 < eta < 4.7 and -6.95 < eta < -6.9. Data were collected in a low intensity LHC run with collisions occurring at a distance of 11.25 m from the nominal interaction point. The data sample is expected to include 96-97\% of the inelastic proton-proton interactions. The measurement reported here considers charged particles with p_T > 0 MeV/c, produced in inelastic interactions with at least one charged particle in -7 < eta < -6 or 3.7 < eta <4.8 . The dN(ch)/deta has been found to decrease with |eta|, from 5.11 +- 0.73 at eta = 3.95 to 1.81 +- 0.56 at eta= - 6.925. Several MC generators are compared to the data and are found to be within the systematic uncertainty of the measurement.
• ### LHC Optics Measurement with Proton Tracks Detected by the Roman Pots of the TOTEM Experiment(1406.0546)

June 2, 2014 hep-ex, physics.acc-ph
Precise knowledge of the beam optics at the LHC is crucial to fulfil the physics goals of the TOTEM experiment, where the kinematics of the scattered protons is reconstructed with the near-beam telescopes -- so-called Roman Pots (RP). Before being detected, the protons' trajectories are influenced by the magnetic fields of the accelerator lattice. Thus precise understanding of the proton transport is of key importance for the experiment. A novel method of optics evaluation is proposed which exploits kinematical distributions of elastically scattered protons observed in the RPs. Theoretical predictions, as well as Monte Carlo studies, show that the residual uncertainty of this optics estimation method is smaller than 0.25 percent.
• ### Performance of the TOTEM Detectors at the LHC(1310.2908)

Oct. 10, 2013 hep-ex, physics.ins-det
The TOTEM Experiment is designed to measure the total proton-proton cross-section with the luminosity-independent method and to study elastic and diffractive pp scattering at the LHC. To achieve optimum forward coverage for charged particles emitted by the pp collisions in the interaction point IP5, two tracking telescopes, T1 and T2, are installed on each side of the IP in the pseudorapidity region 3.1 < = |eta | < = 6.5, and special movable beam-pipe insertions - called Roman Pots (RP) - are placed at distances of +- 147 m and +- 220 m from IP5. This article describes in detail the working of the TOTEM detector to produce physics results in the first three years of operation and data taking at the LHC.
• ### Double diffractive cross-section measurement in the forward region at LHC(1308.6722)

Aug. 30, 2013 hep-ex
The first double diffractive cross-section measurement in the very forward region has been carried out by the TOTEM experiment at the LHC with center-of-mass energy of sqrt(s)=7 TeV. By utilizing the very forward TOTEM tracking detectors T1 and T2, which extend up to |eta|=6.5, a clean sample of double diffractive pp events was extracted. From these events, we measured the cross-section sigma_DD =(116 +- 25) mub for events where both diffractive systems have 4.7 <|eta|_min < 6.5 .
• 2nd workshop on the implications of HERA for LHC physics. Working groups: Parton Density Functions Multi-jet final states and energy flows Heavy quarks (charm and beauty) Diffraction Cosmic Rays Monte Carlos and Tools
• ### Diffraction at TOTEM(0812.3338)

Dec. 17, 2008 hep-ex
The TOTEM experiment at the LHC measures the total proton-proton cross section with the luminosity-independent method and the elastic proton-proton cross-section over a wide |t|-range. It also performs a comprehensive study of diffraction, spanning from cross-section measurements of individual diffractive processes to the analysis of their event topologies. Hard diffraction will be studied in collaboration with CMS taking advantage of the large common rapidity coverage for charged and neutral particle detection and the large variety of trigger possibilities even at large luminosities. TOTEM will take data under all LHC beam conditions including standard high luminosity runs to maximize its physics reach. This contribution describes the main features of the TOTEM physics programme including measurements to be made in the early LHC runs. In addition, a novel scheme to extend the diffractive proton acceptance for high luminosity runs by installing proton detectors at IP3 is described.
• ### CADe tools for early detection of breast cancer(physics/0410082)

Oct. 13, 2004 physics.med-ph
A breast neoplasia is often marked by the presence of microcalcifications and massive lesions in the mammogram: hence the need for tools able to recognize such lesions at an early stage. Our collaboration, among italian physicists and radiologists, has built a large distributed database of digitized mammographic images and has developed a Computer Aided Detection (CADe) system for the automatic analysis of mammographic images and installed it in some Italian hospitals by a GRID connection. Regarding microcalcifications, in our CADe digital mammogram is divided into wide windows which are processed by a convolution filter; after a self-organizing map analyzes each window and produces 8 principal components which are used as input of a neural network (FFNN) able to classify the windows matched to a threshold. Regarding massive lesions we select all important maximum intensity position and define the ROI radius. From each ROI found we extract the parameters which are used as input in a FFNN to distinguish between pathological and non-pathological ROI. We present here a test of our CADe system, used as a second reader and a comparison with another (commercial) CADe system.
• ### GPCALMA: A Tool For Mammography With A GRID-Connected Distributed Database(physics/0410084)

Oct. 13, 2004 physics.med-ph
The GPCALMA (Grid Platform for Computer Assisted Library for MAmmography) collaboration involves several departments of physics, INFN sections, and italian hospitals. The aim of this collaboration is developing a tool that can help radiologists in early detection of breast cancer. GPCALMA has built a large distributed database of digitised mammographic images (about 5500 images corresponding to 1650 patients) and developed a CAD (Computer Aided Detection) software which is integrated in a station that can also be used for acquire new images, as archive and to perform statistical analysis. The images are completely described: pathological ones have a consistent characterization with radiologist's diagnosis and histological data, non pathological ones correspond to patients with a follow up at least three years. The distributed database is realized throught the connection of all the hospitals and research centers in GRID tecnology. In each hospital local patients digital images are stored in the local database. Using GRID connection, GPCALMA will allow each node to work on distributed database data as well as local database data. Using its database the GPCALMA tools perform several analysis. A texture analysis, i.e. an automated classification on adipose, dense or glandular texture, can be provided by the system. GPCALMA software also allows classification of pathological features, in particular massive lesions analysis and microcalcification clusters analysis. The performance of the GPCALMA system will be presented in terms of the ROC (Receiver Operating Characteristic) curves. The results of GPCALMA system as "second reader" will also be presented.
• ### GPCALMA: a Grid-based tool for Mammographic Screening(physics/0410085)

Oct. 13, 2004 physics.med-ph
The next generation of High Energy Physics (HEP) experiments requires a GRID approach to a distributed computing system and the associated data management: the key concept is the Virtual Organisation (VO), a group of distributed users with a common goal and the will to share their resources. A similar approach is being applied to a group of Hospitals which joined the GPCALMA project (Grid Platform for Computer Assisted Library for MAmmography), which will allow common screening programs for early diagnosis of breast and, in the future, lung cancer. HEP techniques come into play in writing the application code, which makes use of neural networks for the image analysis and proved to be useful in improving the radiologists' performances in the diagnosis. GRID technologies allow remote image analysis and interactive online diagnosis, with a potential for a relevant reduction of the delays presently associated to screening programs. A prototype of the system, based on AliEn GRID Services, is already available, with a central Server running common services and several clients connecting to it. Mammograms can be acquired in any location; the related information required to select and access them at any time is stored in a common service called Data Catalogue, which can be queried by any client. The result of a query can be used as input for analysis algorithms, which are executed on nodes that are in general remote to the user (but always local to the input images) thanks to the PROOF facility. The selected approach avoids data transfers for all the images with a negative diagnosis (about 95% of the sample) and allows an almost real time diagnosis for the 5% of images with high cancer probability.
• ### GPCALMA: a Grid Approach to Mammographic Screening(physics/0307157)

The next generation of High Energy Physics experiments requires a GRID approach to a distributed computing system and the associated data management: the key concept is the "Virtual Organisation" (VO), a group of geographycally distributed users with a common goal and the will to share their resources. A similar approach is being applied to a group of Hospitals which joined the GPCALMA project (Grid Platform for Computer Assisted Library for MAmmography), which will allow common screening programs for early diagnosis of breast and, in the future, lung cancer. HEP techniques come into play in writing the application code, which makes use of neural networks for the image analysis and shows performances similar to radiologists in the diagnosis. GRID technologies will allow remote image analysis and interactive online diagnosis, with a relevant reduction of the delays presently associated to screening programs.
• ### Diagnostic performance of radiologists with and without different CAD systems for mammography(physics/0307098)

July 19, 2003 physics.med-ph
The purpose of this study is the evaluation of the variation of performance in terms of sensitivity and specificity of two radiologists with different experience in mammography, with and without the assistance of two different CAD systems. The CAD considered are SecondLookTM (CADx Medical Systems, Canada), and CALMA (Computer Assisted Library in MAmmography). The first is a commercial system, the other is the result of a a research project, supported by INFN (Istituto Nazionale di Fisica Nucleare, Italy); their characteristics have been already reported in literature. To compare the results with and without these tools, a dataset composed by 70 images of patients with cancer (biopsy proven) and 120 images of healthy breasts (with a three years follow up) has been collected. All the images have been digitized and analysed by two CAD, then two radiologists with respectively 6 and 2 years of experience in mammography indipendently made their diagnosis without and with, the support of the two CAD systems. In this work sensitivity and specificity variation, the Az area under the ROC curve, are reported. The results show that the use of a CAD allows for a substantial increment in sensitivity and a less pronounced decrement in specificity. The extent of these effects depends on the experience of the readers and is comparable for the two CAD considered.
• ### A Computer Aided Detection system for mammographic images implemented on a GRID infrastructure(physics/0306198)

June 30, 2003 physics.med-ph
The use of an automatic system for the analysis of mammographic images has proven to be very useful to radiologists in the investigation of breast cancer, especially in the framework of mammographic-screening programs. A breast neoplasia is often marked by the presence of microcalcification clusters and massive lesions in the mammogram: hence the need for tools able to recognize such lesions at an early stage. In the framework of the GPCALMA (GRID Platform for Computer Assisted Library for MAmmography) project, the co-working of italian physicists and radiologists built a large distributed database of digitized mammographic images (about 5500 images corresponding to 1650 patients) and developed a CAD (Computer Aided Detection) system, able to make an automatic search of massive lesions and microcalcification clusters. The CAD is implemented in the GPCALMA integrated station, which can be used also for digitization, as archive and to perform statistical analyses. Some GPCALMA integrated stations have already been implemented and are currently on clinical trial in some italian hospitals. The emerging GRID technology can been used to connect the GPCALMA integrated stations operating in different medical centers. The GRID approach will support an effective tele- and co-working between radiologists, cancer specialists and epidemiology experts by allowing remote image analysis and interactive online diagnosis.
• ### GPCALMA, a mammographic CAD in a GRID connection(physics/0304019)

April 7, 2003 physics.med-ph
Purpose of this work is the development of an automatic system which could be useful for radiologists in the investigation of breast cancer. A breast neoplasia is often marked by the presence of microcalcifications and massive lesions in the mammogram: hence the need for tools able to recognize such lesions at an early stage. GPCALMA (Grid Platform Computer Assisted Library for MAmmography), a collaboration among italian physicists and radiologists, has built a large distributed database of digitized mammographic images (at this moment about 5500 images corresponding to 1650 patients). This collaboration has developed a CAD (Computer Aided Detection) system which, installed in an integrated station, can also be used for digitization, as archive and to perform statistical analysis. With a GRID configuration it would be possible for the clinicians tele- and co-working in new and innovative groupings ('virtual organisations') and, using the whole database, by the GPCALMA tools several analysis can be performed. Furthermore the GPCALMA system allows to be abreast of the CAD technical progressing into several hospital locations always with remote working by GRID connection. We report in this work the results obtained by the GPCALMA CAD software implemented with a GRID connection.