• The NA62 experiment will begin taking data in 2015. Its primary purpose is a 10% measurement of the branching ratio of the ultrarare kaon decay $K^+ \to \pi^+ \nu \bar{ \nu }$, using the decay in flight of kaons in an unseparated beam with momentum 75 GeV/c.The detector and analysis technique are described here.
• A new method for calibrating the hadron response of a segmented calorimeter is developed and successfully applied to beam test data. It is based on a principal component analysis of energy deposits in the calorimeter layers, exploiting longitudinal shower development information to improve the measured energy resolution. Corrections for invisible hadronic energy and energy lost in dead material in front of and between the calorimeters of the ATLAS experiment were calculated with simulated Geant4 Monte Carlo events and used to reconstruct the energy of pions impinging on the calorimeters during the 2004 Barrel Combined Beam Test at the CERN H8 area. For pion beams with energies between 20 GeV and 180 GeV, the particle energy is reconstructed within 3% and the energy resolution is improved by between 11% and 25% compared to the resolution at the electromagnetic scale.
• ### Performance of the Fast Beam Conditions Monitor BCM1F of CMS in the first running periods of LHC(1012.3580)

Dec. 16, 2010 physics.ins-det
The Beam Conditions and Radiation Monitoring System, BRM, is implemented in CMS to protect the detector and provide an interface to the LHC. Seven sub-systems monitor beam conditions and the radiation level inside the detector on different time scales. They detect adverse beam conditions, facilitate beam tuning close to CMS, and measure the doses accumulated in different detector components. Data are taken and analysed independently of the CMS data acquisition, displayed in the control room, and provide inputs to the trigger system and the LHC operators. In case of beam conditions dangerous to the CMS detector, a beam abort is induced. The Fast Beam Conditions Monitor, BCM1F, is a flux counter close to the beam pipe inside the tracker volume. It uses single-crystal CVD diamond sensors, radiation-hard FE electronics, and optical signal transmission to measure the beam halo as well as collision products bunch by bunch. The system has been operational during the initiatory runs of LHC in September 2008. It works reliably since the restart in 2009 and is invaluable to CMS for everyday LHC operation. A characterisation of the system on the basis of data collected during LHC operation is presented.
• ### Fast Beam Conditions Monitor BCM1F for the CMS Experiment(0911.2480)

Dec. 18, 2009 physics.ins-det
The CMS Beam Conditions and Radiation Monitoring System, BRM, will support beam tuning, protect the CMS detector from adverse beam conditions, and measure the accumulated dose close to or inside all sub-detectors. It is composed of different sub-systems measuring either the particle flux near the beam pipe with time resolution between nano- and microseconds or the integrated dose over longer time intervals. This paper presents the Fast Beam Conditions Monitor, BCM1F, which is designed for fast flux monitoring measuring both beam halo and collision products. BCM1F is located inside the CMS pixel detector volume close to the beam-pipe. It uses sCVD diamond sensors and radiation hard front-end electronics, along with an analog optical readout of the signals. The commissioning of the system and its successful operation during the first be ams of the LHC are described.