• ### PandaX-III: Searching for Neutrinoless Double Beta Decay with High Pressure $^{136}$Xe Gas Time Projection Chambers(1610.08883)

Oct. 28, 2016 hep-ex, nucl-ex, physics.ins-det
Searching for the Neutrinoless Double Beta Decay (NLDBD) is now regarded as the topmost promising technique to explore the nature of neutrinos after the discovery of neutrino masses in oscillation experiments. PandaX-III (Particle And Astrophysical Xenon Experiment III) will search for the NLDBD of $^{136}$Xe at the China Jin Ping underground Laboratory (CJPL). In the first phase of the experiment, a high pressure gas Time Projection Chamber (TPC) will contain 200 kg, 90% $^{136}$Xe enriched gas operated at 10 bar. Fine pitch micro-pattern gas detector (Microbulk Micromegas) will be used at both ends of the TPC for the charge readout with a cathode in the middle. Charge signals can be used to reconstruct tracks of NLDBD events and provide good energy and spatial resolution. The detector will be immersed in a large water tank to ensure $\sim$5 m of water shielding in all directions. The second phase, a ton-scale experiment, will consist of five TPCs in the same water tank, with improved energy resolution and better control over backgrounds.
• ### Detailed study of the column-based priority logic readout of Topmetal-II- CMOS pixel direct charge sensor(1608.06399)

Aug. 23, 2016 physics.ins-det
We present the detailed study of the digital readout of Topmetal-II- CMOS pixel direct charge sensor. Topmetal-II- is an integrated sensor with an array of 72X72 pixels each capable of directly collecting external charge through exposed metal electrodes in the topmost metal layer. In addition to the time-shared multiplexing readout of the analog output from Charge Sensitive Amplifiers in each pixel, hits are also generated through comparators with individually DAC settable thresholds in each pixel. The hits are read out via a column-based priority logic structure, retaining both hit location and time information. The in-array column-based priority logic is fully combinational hence there is no clock distributed in the pixel array. Sequential logic and clock are placed on the peripheral of the array. We studied the detailed working behavior and performance of this readout, and demonstrated its potential in imaging applications.
• ### A Low-Noise CMOS Pixel Direct Charge Sensor, Topmetal-II-(1509.08611)

Dec. 22, 2015 physics.ins-det
We report the design and characterization of a CMOS pixel direct charge sensor, Topmetal-II-, fabricated in a standard 0.35um CMOS Integrated Circuit process. The sensor utilizes exposed metal patches on top of each pixel to directly collect charge. Each pixel contains a low-noise charge-sensitive preamplifier to establish the analog signal and a discriminator with tunable threshold to generate hits. The analog signal from each pixel is accessible through time-shared multiplexing over the entire array. Hits are read out digitally through a column-based priority logic structure. Tests show that the sensor achieved a <15e- analog noise and a 200e- minimum threshold for digital readout per pixel. The sensor is capable of detecting both electrons and ions drifting in gas. These characteristics enable its use as the charge readout device in future Time Projection Chambers without gaseous gain mechanism, which has unique advantages in low background and low rate-density experiments.
• ### Development of a highly pixelated direct charge sensor, Topmetal-I, for ionizing radiation imaging(1407.3712)

Using industrial standard 0.35{\mu}m CMOS Integrated Circuit process, we realized a highly pixelated sensor that directly collects charge via metal nodes placed on the top of each pixel and forms two dimensional images of charge cloud distribution. The first version, Topmetal-I, features a 64x64 pixel array of 80{\mu}m pitch size. Direct charge calibration reveals an average capacitance of 210fF per pixel. The charge collection noise is near the thermal noise limit. With the readout, individual pixel channels exhibit a most probable equivalent noise charge of 330e-.