• ### New mass anchor points for neutron-deficient heavy nuclei from direct mass measurements of radium and actinium isotopes(1801.02823)

April 27, 2018 nucl-ex
The masses of the exotic isotopes $^{210-214}$Ac and $^{210-214}$Ra have been measured with a multi-reflection time-of-flight mass spectrograph. These isotopes were obtained in flight as fusion-evaporation products behind the gas-filled recoil ion separator GARIS-II at RIKEN. The new direct mass measurements serve as an independent and direct benchmark for existing $\alpha$-$\gamma$-spectroscopy data in this mass region. Further, new mass anchor points are set for U and Np nuclei close to the $N=126$ shell closure for a future benchmark of the $Z=92$ sub-shell for neutron-deficient heavy isotopes. Our mass results are in general in good agreement with the previously indirectly-determined mass values. Together with the measurement data, reasons for possible mass ambiguities from decay-data links between ground states are discussed.
• ### First direct mass measurements of nuclides around $Z=100$ with a Multireflection Time-of-Flight Mass Spectrograph(1709.06468)

April 3, 2018 nucl-ex
The masses of $^{246}$Es, $^{251}$Fm and the transfermium nuclei $^{249-252}$Md, and $^{254}$No, produced by hot- and cold-fusion reactions, in the vicinity of the deformed $N=152$ neutron shell closure, have been directly measured using a multireflection time-of-flight mass spectrograph. The masses of $^{246}$Es and $^{249,250,252}$Md were measured for the first time. Using the masses of $^{249,250}$Md as anchor points for $\alpha$ decay chains, the masses of heavier nuclei, up to $^{261}$Bh and $^{266}$Mt, were determined. These new masses were compared with theoretical global mass models and demonstrated to be in good agreement with macroscopic-microscopic models in this region. The empirical shell gap parameter $\delta_{2n}$ derived from three isotopic masses was updated with the new masses and corroborate the existence of the deformed $N=152$ neutron shell closure for Md and Lr.
• ### Atomic masses of intermediate-mass neutron-deficient nuclei with relative uncertainty down to 35-ppb via multireflection time-of-flight mass spectrograph(1706.00186)

March 1, 2018 nucl-ex
High-precision mass measurements of $^{63}$Cu, $^{64-66}$Zn, $^{65}$Ga, $^{65-67}$Ge, $^{67}$As, $^{78,81}$Br, $^{80}$Rb, and $^{79}$Sr were performed utilizing a multireflection time-of-flight mass spectrograph combined with the gas-filled recoil ion separator GARIS-II. In the case of $^{65}$Ga, a mass uncertainty of 2.1 keV, corresponding to a relative precision of $\delta m / m = 3.5\times10^{-8}$, was obtained and the mass value is in excellent agreement with the 2016 Atomic Mass Evaluation. For $^{67}$Ge and $^{81}$Br, where masses were previously deduced through indirect measurements, discrepancies with literature values were found. The feasibility of using this device for mass measurements of nuclides more neutron-deficient side, which have significant impact on the $rp$-process pathway, is discussed.
• ### A novel ion cooling trap for multi-reflection time-of-flight mass spectrograph(1309.0384)

A radiofrequency quadrupole ion trap system for use with a multi-reflection time-of-flight mass spectrograph (MRTOF) for short-lived nuclei has been developed. The trap system consists of two different parts, an asymmetric taper trap and a flat trap. The ions are cooled to a sufficient small bunch for precise mass measurement with MRTOF in only 2 ms cooling time in the flat trap, then orthogonally ejected to the MRTOF for mass analysis. A trapping efficiency of ~27% for 23Na+ and ~5.1% for 7Li+ has been achieved.
• ### Single-reference high-precision mass measurement with a multi-reflection time-of-flight mass spectrograph(1307.7373)

July 28, 2013 nucl-ex
A multi-reflection time-of-flight mass spectrograph, competitive with Penning trap mass spectrometers, has been built at RIKEN. We have performed a first online mass measurement, using 8Li+ (T1/2 = 838 ms). A new analysis method has been realized, with which, using only 12C+ references, the mass excess of 8Li was accurately determined to be 20947.6(15)(34) keV (dm/m = 6.6 x 10-7). The speed, precision and accuracy of this first online measurement exemplifies the potential for using this new type of mass spectrograph for precision measurements of short-lived nuclei.
• ### An rf-carpet electrospray ion source to provide isobaric mass calibrants for trans-uranium elements(1211.1448)

Dec. 16, 2012 nucl-ex, physics.atom-ph
For trans-uranium elements, stable atomic isobars do not exist. In order to provide isobaric reference ions for the mass measurement of trans-uranium elements, an electrospray ion source (ESI) was combined with an rf-carpet to collect molecular ions efficiently. The rf-carpet allows for simplification of the pumping system to transport ions from the ESI to a precision mass analyzer. Molecular ions appropriate for isobaric references of trans-uranium elements were extracted from the rf-carpet and analyzed by a multi-reflection time-of-flight mass spectrograph (MRTOF-MS) with a resolving power of $\rm{R_m} \gtrsim100,000$.
• ### Development of a resonant laser ionization gas cell for high-energy, short-lived nuclei(1210.6483)

Oct. 24, 2012 nucl-ex, physics.ins-det
A new laser ion source configuration based on resonant photoionization in a gas cell has been developed at RIBF RIKEN. This system is intended for the future PArasitic RI-beam production by Laser Ion-Source (PALIS) project which will be installed at RIKEN's fragment separator, BigRIPS. A novel implementation of differential pumping, in combination with a sextupole ion beam guide (SPIG), has been developed. A few small scroll pumps create a pressure difference from 1000 hPa - 10^-3 Pa within a geometry drastically miniaturized compared to conventional systems. This system can utilize a large exit hole for fast evacuation times, minimizing the decay loss for short-lived nuclei during extraction from a buffer gas cell, while sufficient gas cell pressure is maintained for stopping high energy RI-beams. In spite of the motion in a dense pressure gradient, the photo-ionized ions inside the gas cell are ejected with an assisting force gas jet and successfully transported to a high-vacuum region via SPIG followed by a quadrupole mass separator. Observed behaviors agree with the results of gas flow and Monte Carlo simulations.