• A new pixel detector for the CMS experiment was built in order to cope with the instantaneous luminosities anticipated for the Phase~I Upgrade of the LHC. The new CMS pixel detector provides four-hit tracking with a reduced material budget as well as new cooling and powering schemes. A new front-end readout chip mitigates buffering and bandwidth limitations, and allows operation at low comparator thresholds. In this paper, comprehensive test beam studies are presented, which have been conducted to verify the design and to quantify the performance of the new detector assemblies in terms of tracking efficiency and spatial resolution. Under optimal conditions, the tracking efficiency is $99.95\pm0.05\,\%$, while the intrinsic spatial resolutions are $4.80\pm0.25\,\mu \mathrm{m}$ and $7.99\pm0.21\,\mu \mathrm{m}$ along the $100\,\mu \mathrm{m}$ and $150\,\mu \mathrm{m}$ pixel pitch, respectively. The findings are compared to a detailed Monte Carlo simulation of the pixel detector and good agreement is found.
• ### Measurements of the Rate Capability of Various Resistive Plate Chambers(1507.06968)

Sept. 7, 2015 physics.ins-det
Resistive Plate Chambers (RPCs) exhibit a significant loss of efficiency for the detection of particles, when subjected to high particle fluxes. This rate limitation is related to the usually high resistivity of the resistive plates used in their construction. This paper reports on measurements of the performance of three different glass RPC designs featuring a different total resistance of the resistive plates. The measurements were performed with 120 GeV protons at varying beam intensities
• The degradation of signal in silicon sensors is studied under conditions expected at the CERN High-Luminosity LHC. 200 $\mu$m thick n-type silicon sensors are irradiated with protons of different energies to fluences of up to $3 \cdot 10^{15}$ neq/cm$^2$. Pulsed red laser light with a wavelength of 672 nm is used to generate electron-hole pairs in the sensors. The induced signals are used to determine the charge collection efficiencies separately for electrons and holes drifting through the sensor. The effective trapping rates are extracted by comparing the results to simulation. The electric field is simulated using Synopsys device simulation assuming two effective defects. The generation and drift of charge carriers are simulated in an independent simulation based on PixelAV. The effective trapping rates are determined from the measured charge collection efficiencies and the simulated and measured time-resolved current pulses are compared. The effective trapping rates determined for both electrons and holes are about 50% smaller than those obtained using standard extrapolations of studies at low fluences and suggests an improved tracker performance over initial expectations.
• ### Tests of a Novel Design of Resistive Plate Chambers(1501.05907)

March 4, 2015 hep-ex, physics.ins-det
A novel design of Resistive Plate Chambers (RPCs), using only a single resistive plate, is being proposed. Based on this design, two large size prototype chambers were constructed and were tested with cosmic rays and in particle beams. The tests confirmed the viability of this new approach. In addition to showing an improved single-particle response compared to the traditional 2-plate design, the novel chambers also prove to be suitable for calorimetric applications.