• ### Demonstration of high contrast with an obscured aperture with the WFIRST-AFTA shaped pupil coronagraph(1511.01882)

Nov. 5, 2015 astro-ph.IM
The coronagraph instrument on the WFIRST-AFTA mission study has two coronagraphic architectures, shaped pupil and hybrid Lyot, which may be interchanged for use in different observing scenarios. Each architecture relies on newly-developed mask components to function in the presence of the AFTA aperture, and so both must be matured to a high technology readiness level (TRL) in advance of the mission. A series of milestones were set to track the development of the technologies required for the instrument; in this paper, we report on completion of WFIRST-AFTA Coronagraph Milestone 2---a narrowband $10^{-8}$ contrast test with static aberrations for the shaped pupil---and the plans for the upcoming broadband Coronagraph Milestone 5.
• ### A Multi-wavelength Differential Imaging Experiment for the High Contrast Imaging Testbed(0906.0395)

June 1, 2009 astro-ph.IM
We discuss the results of a multi-wavelength differential imaging lab experiment with the High Contrast Imaging Testbed (HCIT) at the Jet Propulsion Laboratory. The HCIT combines a Lyot coronagraph with a Xinetics deformable mirror in a vacuum environment to simulate a space telescope in order to test technologies and algorithms for a future exoplanet coronagraph mission. At present, ground based telescopes have achieved significant attenuation of speckle noise using the technique of spectral differential imaging (SDI). We test whether ground-based SDI can be generalized to a non-simultaneous spectral differential imaging technique (NSDI) for a space mission. In our lab experiment, a series of 5 filter images centered around the O2(A) absorption feature at 0.762 um were acquired at nominal contrast values of 10^-6, 10^-7, 10^-8, and 10^-9. Outside the dark hole, single differences of images improve contrast by a factor of ~6. Inside the dark hole, we found significant speckle chromatism as a function of wavelength offset from the nulling wavelength, leading to a contrast degradation by a factor of 7.2 across the entire ~80 nm bandwidth. This effect likely stems from the chromatic behavior of the current occulter. New, less chromatic occulters are currently in development; we expect that these new occulters will resolve the speckle chromatism issue.