The Imaging Magnetograph eXperiment (IMaX) is a spectropolarimeter built by
four institutions in Spain that flew on board the Sunrise balloon-borne
telesocope in June 2009 for almost six days over the Arctic Circle. As a
polarimeter IMaX uses fast polarization modulation (based on the use of two
liquid crystal retarders), real-time image accumulation, and dual beam
polarimetry to reach polarization sensitivities of 0.1%. As a spectrograph, the
instrument uses a LiNbO3 etalon in double pass and a narrow band pre-filter to
achieve a spectral resolution of 85 mAA. IMaX uses the high Zeeman sensitive
line of Fe I at 5250.2 AA and observes all four Stokes parameters at various
points inside the spectral line. This allows vector magnetograms, Dopplergrams,
and intensity frames to be produced that, after reconstruction, reach spatial
resolutions in the 0.15-0.18 arcsec range over a 50x50 arcsec FOV. Time
cadences vary between ten and 33 seconds, although the shortest one only
includes longitudinal polarimetry. The spectral line is sampled in various ways
depending on the applied observing mode, from just two points inside the line
to 11 of them. All observing modes include one extra wavelength point in the
nearby continuum. Gauss equivalent sensitivities are four Gauss for
longitudinal fields and 80 Gauss for transverse fields per wavelength sample.
The LOS velocities are estimated with statistical errors of the order of 5-40
m/s. The design, calibration and integration phases of the instrument, together
with the implemented data reduction scheme are described in some detail.
SICANE is a neutron scattering multidetector facility for the determination
of the quenching factor (ratio of the response to nuclear recoils and to
electrons) of cryogenic detectors used in direct WIMP searches. Well collimated
monoenergetic neutron beams are obtained with inverse (p,n) reactions. The
facility is described, and results obtained for the quenching factors of
scintillation in NaI(Tl) and of heat and ionization in Ge are presented.
The EDELWEISS Dark Matter Search uses low-temperature Ge detectors with heat
and ionisation read-out to identify nuclear recoils induced by elastic
collisions with WIMPs from the galactic halo. Preliminary results obtained with
320g bolometers are described. After a few weeks of data taking, data
accumulated with one of these detectors already allow to reach the upper part
of the DAMA region. Prospects for the present run and the second stage of the
experiment, EDELWEISS-II, using an innovative reversed cryostat allowing data
taking with 100 detectors, are briefly described.
Preliminary results obtained with 320g bolometers with simultaneous
ionization and heat measurements are described. After a few weeks of data
taking, data accumulated with one of these detectors are beginning to exclude
the upper part of the DAMA region. Prospects for the present run and the second
stage of the experiment, EDELWEISS-II, using an innovative reversed cryostat
allowing data taking with 100 detectors, are briefly described.
The EDELWEISS Dark Matter search uses low-temperature Ge detectors with heat
and ionisation read- out to identify nuclear recoils induced by elastic
collisions with WIMPs from the galactic halo. Results from the operation of 70
g and 320 g Ge detectors in the low-background environment of the Modane
Underground Laboratory (LSM) are presented.
The status of the EDELWEISS experiment (underground dark matter search with
heat-ionisation bolometers) is reviewed. Auspicious results achieved with a
prototype 70 g Ge heat-ionisation detector under a 2 V reverse bias tension are
discussed. Based on gamma and neutron calibrations, a best-case rejection
factor, over the 15-45 keV range, of 99.7 % for gammas, with an acceptance of
94 % for neutrons, is presented first. Some operational results of physical
interest obtained under poor low radioactivity conditions follow. They include
a raw event rate of around 30 events/day/kg/keV over the same energy range,
and, after rejection of part of the background, lead to a conservative upper
limit on the signal of approximately 1.6 events/day/kg/keV at a 90 % confidence
level. Performance degrading surface effects of the detector are speculated
upon; and planned upgrades are summarized.