• ### Reduction of stored-particle background by a magnetic pulse method at the KATRIN experiment(1805.01163)

May 3, 2018 physics.ins-det
The KATRIN experiment aims to determine the effective electron neutrino mass with a sensitivity of $0.2\,{\text{eV}/c^2}$ (90\% C.L.) by precision measurement of the shape of the tritium \textbeta-spectrum in the endpoint region. The energy analysis of the decay electrons is achieved by a MAC-E filter spectrometer. A common background source in this setup is the decay of short-lived isotopes, such as $\textsuperscript{219}$Rn and $\textsuperscript{220}$Rn, in the spectrometer volume. Active and passive countermeasures have been implemented and tested at the KATRIN main spectrometer. One of these is the magnetic pulse method, which employs the existing air coil system to reduce the magnetic guiding field in the spectrometer on a short timescale in order to remove low- and high-energy stored electrons. Here we describe the working principle of this method and present results from commissioning measurements at the main spectrometer. Simulations with the particle-tracking software Kassiopeia were carried out to gain a detailed understanding of the electron storage conditions and removal processes.
• ### Calibration of high voltages at the ppm level by the difference of $^{83\mathrm{m}}$Kr conversion electron lines at the KATRIN experiment(1802.05227)

Feb. 14, 2018 physics.ins-det
The neutrino mass experiment KATRIN requires a stability of 3 ppm for the retarding potential at -18.6 kV of the main spectrometer. To monitor the stability, two custom-made ultra-precise high-voltage dividers were developed and built in cooperation with the German national metrology institute Physikalisch-Technische Bundesanstalt (PTB). Until now, regular absolute calibration of the voltage dividers required bringing the equipment to the specialised metrology laboratory. Here we present a new method based on measuring the energy difference of two $^{83\mathrm{m}}$Kr conversion electron lines with the KATRIN setup, which was demonstrated during KATRIN's commissioning measurements in July 2017. The measured scale factor $M=1972.449(10)$ of the high-voltage divider K35 is in agreement with the last PTB calibration four years ago. This result demonstrates the utility of the calibration method, as well as the long-term stability of the voltage divider.
• ### First transmission of electrons and ions through the KATRIN beamline(1802.04167)

Feb. 12, 2018 nucl-ex, physics.ins-det
The Karlsruhe Tritium Neutrino (KATRIN) experiment is a large-scale effort to probe the absolute neutrino mass scale with a sensitivity of 0.2 eV (90% confidence level), via a precise measurement of the endpoint spectrum of tritium beta decay. This work documents several KATRIN commissioning milestones: the complete assembly of the experimental beamline, the successful transmission of electrons from three sources through the beamline to the primary detector, and tests of ion transport and retention. In the First Light commissioning campaign of Autumn 2016, photoelectrons were generated at the rear wall and ions were created by a dedicated ion source attached to the rear section; in July 2017, gaseous Kr-83m was injected into the KATRIN source section, and a condensed Kr-83m source was deployed in the transport section. In this paper we describe the technical details of the apparatus and the configuration for each measurement, and give first results on source and system performance. We have successfully achieved transmission from all four sources, established system stability, and characterized many aspects of the apparatus.
• ### A White Paper on keV Sterile Neutrino Dark Matter(1602.04816)

We present a comprehensive review of keV-scale sterile neutrino Dark Matter, collecting views and insights from all disciplines involved - cosmology, astrophysics, nuclear, and particle physics - in each case viewed from both theoretical and experimental/observational perspectives. After reviewing the role of active neutrinos in particle physics, astrophysics, and cosmology, we focus on sterile neutrinos in the context of the Dark Matter puzzle. Here, we first review the physics motivation for sterile neutrino Dark Matter, based on challenges and tensions in purely cold Dark Matter scenarios. We then round out the discussion by critically summarizing all known constraints on sterile neutrino Dark Matter arising from astrophysical observations, laboratory experiments, and theoretical considerations. In this context, we provide a balanced discourse on the possibly positive signal from X-ray observations. Another focus of the paper concerns the construction of particle physics models, aiming to explain how sterile neutrinos of keV-scale masses could arise in concrete settings beyond the Standard Model of elementary particle physics. The paper ends with an extensive review of current and future astrophysical and laboratory searches, highlighting new ideas and their experimental challenges, as well as future perspectives for the discovery of sterile neutrinos.
• ### Commissioning of the vacuum system of the KATRIN Main Spectrometer(1603.01014)

March 3, 2016 physics.ins-det
The KATRIN experiment will probe the neutrino mass by measuring the beta-electron energy spectrum near the endpoint of tritium beta-decay. An integral energy analysis will be performed by an electro-static spectrometer (Main Spectrometer), an ultra-high vacuum vessel with a length of 23.2 m, a volume of 1240 m^3, and a complex inner electrode system with about 120000 individual parts. The strong magnetic field that guides the beta-electrons is provided by super-conducting solenoids at both ends of the spectrometer. Its influence on turbo-molecular pumps and vacuum gauges had to be considered. A system consisting of 6 turbo-molecular pumps and 3 km of non-evaporable getter strips has been deployed and was tested during the commissioning of the spectrometer. In this paper the configuration, the commissioning with bake-out at 300{\deg}C, and the performance of this system are presented in detail. The vacuum system has to maintain a pressure in the 10^{-11} mbar range. It is demonstrated that the performance of the system is already close to these stringent functional requirements for the KATRIN experiment, which will start at the end of 2016.
• ### A novel $^{83\mathrm{m}}$Kr tracer method for characterizing xenon gas and cryogenic distillation systems(1407.3981)

July 15, 2014 physics.ins-det, astro-ph.IM
The radioactive isomer $^{83\mathrm{m}}$Kr has many properties that make it very useful for various applications. Its low energy decay products, like conversion, shake-off and Auger electrons as well as X- and $\gamma$-rays are used for calibration purposes in neutrino mass experiments and direct dark matter detection experiments. Thanks to the short half-life of 1.83 h and the decay to the ground state $^{83}$Kr, one does not risk contamination of any low-background experiment with long- lived radionuclides. In this paper, we present two new applications of $^{83\mathrm{m}}$Kr. It can be used as a radioactive tracer in noble gases to characterize the particle flow inside of gas routing systems. A method of doping $^{83\mathrm{m}}$Kr into xenon gas and its detection, using special custom-made detectors, based on a photomultiplier tube, is described. This technique has been used to determine the circulation speed of gas particles inside of a gas purification system for xenon. Furthermore, 83m Kr can be used to rapidly estimate separation performance of a distillation system.
• ### Ultra-stable implanted 83Rb/83mKr electron sources for the energy scale monitoring in the KATRIN experiment(1212.5016)

Feb. 5, 2013 nucl-ex, physics.ins-det
The KATRIN experiment aims at the direct model-independent determination of the average electron neutrino mass via the measurement of the endpoint region of the tritium beta decay spectrum. The electron spectrometer of the MAC-E filter type is used, requiring very high stability of the electric filtering potential. This work proves the feasibility of implanted 83Rb/83mKr calibration electron sources which will be utilised in the additional monitor spectrometer sharing the high voltage with the main spectrometer of KATRIN. The source employs conversion electrons of 83mKr which is continuously generated by 83Rb. The K-32 conversion line (kinetic energy of 17.8 keV, natural line width of 2.7 eV) is shown to fulfill the KATRIN requirement of the relative energy stability of +/-1.6 ppm/month. The sources will serve as a standard tool for continuous monitoring of KATRIN's energy scale stability with sub-ppm precision. They may also be used in other applications where the precise conversion lines can be separated from the low energy spectrum caused by the electron inelastic scattering in the substrate.
• ### Limits on the release of Rb isotopes from a zeolite based 83mKr calibration source for the XENON project(1109.4270)

The isomer 83mKr with its half-life of 1.83 h is an ideal calibration source for a liquid noble gas dark matter experiment like the XENON project. However, the risk of contamination of the detector with traces of the much longer lived mother isotop 83Rb (86.2 d half-life) has to be ruled out. In this work the release of 83Rb atoms from a 1.8 MBq 83Rb source embedded in zeolite beads has been investigated. To do so, a cryogenic trap has been connected to the source for about 10 days, after which it was removed and probed for the strongest 83Rb gamma-rays with an ultra-sensitive Germanium detector. No signal has been found. The corresponding upper limit on the released 83Rb activity means that the investigated type of source can be used in the XENON project and similar low-background experiments as 83mKr generator without a significant risk of contaminating the detector. The measurements also allow to set upper limits on the possible release of the isotopes 84Rb and 86Rb, traces of which were created alongside the production of 83Rb at the Rez cyclotron.
• ### Magnetic moment of Ag-104(m) and the hyperfine magnetic field of Ag in Fe using nuclear magnetic resonance on oriented nuclei(1006.5885)

June 30, 2010 nucl-ex
Nuclear magnetic resonance (NMR/ON) measurements with beta- and gamma-ray detection have been performed on oriented Ag-104(g,m) nuclei with the NICOLE He-3-He-4 dilution refrigerator setup at ISOLDE/CERN. For Ag-104(g) (I-pi = 5(+)) the gamma-NMR/ON resonance signal was found at nu = 266.70(5) MHz. Combining this result with the known magnetic moment for this isotope, the magnetic hyperfine field of Ag impurities in an Fe host at low temperature (< 1 K) is found to be vertical bar B-hf(AgFe)vertical bar = 44.709(35) T. A detailed analysis of other relevant data available in the literature yields three more values for this hyperfine field. Averaging all four values yields a new and precise value for the hyperfine field of Ag in Fe; that is, vertical bar B-hf(AgFe)vertical bar = 44.692(30) T. For Ag-104(m) (I-pi = 2(+)), the anisotropy of the beta particles provided the NMR/ON resonance signal at nu = 627.7(4) MHz. Using the new value for the hyperfine field of Ag in Fe, this frequency corresponds to the magnetic moment mu(Ag-104m) = +3.691(3) mu(N), which is significantly more precise than previous results. The magnetic moments of the even-A Ag102 -110 isotopes are discussed in view of the competition between the (pi g(9/2))(7/2+)(-3)(nu d(5/2)nu g(7/2))(5/2+) and the (pi g(9/2))(9/2+)(-3)(nu d(5/2)nu g(7/2))(5/2+) configurations. The magnetic moments of the ground and isomeric states of Ag-104 can be explained by an almost complete mixing of these two configurations.
• ### Spatially uniform calibration of a liquid xenon detector at low energies using 83m-Kr(0908.0616)

May 16, 2010 astro-ph.IM
A difficult task with many particle detectors focusing on interactions below ~100 keV is to perform a calibration in the appropriate energy range that adequately probes all regions of the detector. Because detector response can vary greatly in various locations within the device, a spatially uniform calibration is important. We present a new method for calibration of liquid xenon (LXe) detectors, using the short-lived 83m-Kr. This source has transitions at 9.4 and 32.1 keV, and as a noble gas like Xe, it disperses uniformly in all regions of the detector. Even for low source activities, the existence of the two transitions provides a method of identifying the decays that is free of background. We find that at decreasing energies, the LXe light yield increases, while the amount of electric field quenching is diminished. Additionally, we show that if any long-lived radioactive backgrounds are introduced by this method, they will present less than 67E-6 events/kg/day in the next generation of LXe dark matter direct detection searches
• ### Feasibility of photoelectron sources for testing the energy scale stablity of the KATRIN beta-ray spectrometer(1003.3758)

March 19, 2010 nucl-ex
Photoabsorption of nuclear gamma-rays in thin metallic convertors was examined with the aim to produce monoenergetic photoelectrons of kinetic energy around 18.6 keV and natural width of about 1 eV. Calculations were carried out for commercial photon sources of 241Am (1.1 GBq) and 119mSn (0.5 GBq) irradiating Co and Ti convertors. Photoelectrons ejected by 241Am gamma- and X-rays from Co convertors of various thickness were measured with two electrostatic spectrometers.
• ### Long term stability of the energy of conversion electrons emitted from solid 83Rb/83mKr source(0902.0291)

Feb. 2, 2009 physics.ins-det
The mono-energetic conversion electrons from the decay of 83mKr represent a unique tool for the energy calibration, energy scale monitoring and systematic studies of the tritium beta spectrum measurement in the neutrino mass experiment KATRIN. For this reason, the long term stability of energy of the 7.5 keV and 17.8 keV conversion electrons populated in the decay of solid 83Rb/83mKr vacuum evaporated sources was examined by means of two electron spectrometers.
• ### Distribution of the 83Rb/83mKr activity on vacuum evaporated samples examined with the Timepix position sensitive detector(0712.3860)

Dec. 22, 2007 nucl-ex
Properties of vacuum evaporated 83Rb/83mKr sources of low-energy conversion electrons, which are under development for monitoring the energy scale stability of the Karlsruhe Tritium Neutrino experiment KATRIN, were examined by the Timepix pixel detector exhibiting the position resolution of at least 55 microm. No distinct local inhomogeneities in the surface distribution of 83Rb/83mKr were observed. The source diameter derived from the recorded image agrees within 5 % with that expected from evaporation geometry. More precise determination of the actual source diameter is complicated by Compton scattered electrons caused by 83Rb gamma-rays.