• Electrical Induced Annealing technique for neutron radiation damage on SiPMs(1804.09792)

April 27, 2018 physics.ins-det
The use of Silicon Photo-Multipliers (SiPMs) has become popular in the design of High Energy Physics experimental apparatus with a growing interest for their application in detector area where a significant amount of non-ionising dose is delivered. For these devices, the main effect caused by the neutron flux is a linear increase of the leakage current. In this paper, we present a technique that provides a partial recovery of the neutron damage on SiPMs by means of an Electrical Induced Annealing. Tests were performed on a sample of three SiPM arrays (2 $\times$ 3) of 6 mm$^2$ cells with 50 {\mu}m pixel sizes: two from Hamamatsu and one from SensL. These SiPMs were irradiated up to an integrated neutron flux up to 8 $\times$ 10$^{11}$ n$_{1MeV-eq}$/cm$^2$. Our techniques allowed to reduced the leakage current of a factor ranging between 15-20 depending on the overbias used and the SiPM vendor.
• Design, status and perspective of the Mu2e crystal calorimeter(1801.03159)

April 18, 2018 hep-ex, physics.ins-det
The Mu2e experiment at Fermilab will search for the charged lepton flavor violating process of neutrino-less $\mu \to e$ coherent conversion in the field of an aluminum nucleus. Mu2e will reach a single event sensitivity of about $2.5\cdot 10^{-17}$ that corresponds to four orders of magnitude improvements with respect to the current best limit. The detector system consists of a straw tube tracker and a crystal calorimeter made of undoped CsI coupled with Silicon Photomultipliers. The calorimeter was designed to be operable in a harsh environment where about 10 krad/year will be delivered in the hottest region and work in presence of 1 T magnetic field. The calorimeter role is to perform $\mu$/e separation to suppress cosmic muons mimiking the signal, while providing a high level trigger and a seeding the track search in the tracker. In this paper we present the calorimeter design and the latest R$\&$D results.

March 15, 2018 hep-ex, physics.ins-det
In this paper, we report the measurement of the neutron radiation hardness of custom Silicon Photomultipliers arrays (SiPMs) manufactured by three companies: Hamamatsu (Japan), AdvanSiD (Italy) and SensL (Ireland). These custom SiPMs consist of a 2 x 3 array of 6 x 6 mm^2 monolithic cells with pixel sizes of respectively 50 um (Hamamatsu and SensL) and 30 um (AdvanSid). A sample from each vendor has been exposed to neutrons generated by the Elbe Positron Source facility (Dresden), up to a total fluence of ~ 8.5 x 10^11 n_(1 MeV)/cm^2. Test results show that the dark current increases almost linearly with the neutron fluence. The room temperature annealing was quantified by measuring the dark current two months after the irradiation test. The dependence of the dark current on the device temperature and on the applied bias have been also evaluated.
• Results of the first user program on the Homogenous Thermal Neutron Source HOTNES (ENEA / INFN)(1802.08132)

Feb. 22, 2018 nucl-ex, physics.ins-det
The HOmogeneous Thermal NEutron Source (HOTNES) is a new type of thermal neutron irradiation assembly developed by the ENEA-INFN collaboration. The facility is fully characterized in terms of neutron field and dosimetric quantities, by either computational and experimental methods. This paper reports the results of the first "HOTNES users program", carried out in 2016, and covering a variety of thermal neutron active detectors such as scintillators, solid-state, single crystal diamond and gaseous detectors.
• Quality Assurance on Un-Doped CsI Crystals for the Mu2e Experiment(1802.08247)

Feb. 22, 2018 hep-ex, physics.ins-det
The Mu2e experiment is constructing a calorimeter consisting of 1,348 undoped CsI crystals in two disks. Each crystal has a dimension of 34 x 34 x 200 mm, and is readout by a large area silicon PMT array. A series of technical specifications was defined according to physics requirements. Preproduction CsI crystals were procured from three firms: Amcrys, Saint-Gobain and Shanghai Institute of Ceramics. We report the quality assurance on crystal's scintillation properties and their radiation hardness against ionization dose and neutrons. With a fast decay time of 30 ns and a light output of more than 100 p.e./MeV measured with a bi-alkali PMT, undoped CsI crystals provide a cost-effective solution for the Mu2e experiment.
• The Mu2e undoped CsI crystal calorimeter(1801.02237)

Feb. 22, 2018 physics.ins-det
The Mu2e experiment at Fermilab will search for Charged Lepton Flavor Violating conversion of a muon to an electron in an atomic field. The Mu2e detector is composed of a tracker, an electromagnetic calorimeter and an external system, surrounding the solenoid, to veto cosmic rays. The calorimeter plays an important role to provide: a) excellent particle identification capabilities; b) a fast trigger filter; c) an easier tracker track reconstruction. Two disks, located downstream of the tracker, contain 674 pure CsI crystals each. Each crystal is read out by two arrays of UV-extended SiPMs. The choice of the crystals and SiPMs has been finalized after a thorough test campaign. A first small scale prototype consisting of 51 crystals and 102 SiPM arrays has been exposed to an electron beam at the BTF (Beam Test Facility) in Frascati. Although the readout electronics were not the final, results show that the current design is able to meet the timing and energy resolution required by the Mu2e experiment.
• Expression of Interest for Evolution of the Mu2e Experiment(1802.02599)

Feb. 7, 2018 hep-ex, physics.ins-det
We propose an evolution of the Mu2e experiment, called Mu2e-II, that would leverage advances in detector technology and utilize the increased proton intensity provided by the Fermilab PIP-II upgrade to improve the sensitivity for neutrinoless muon-to-electron conversion by one order of magnitude beyond the Mu2e experiment, providing the deepest probe of charged lepton flavor violation in the foreseeable future. Mu2e-II will use as much of the Mu2e infrastructure as possible, providing, where required, improvements to the Mu2e apparatus to accommodate the increased beam intensity and cope with the accompanying increase in backgrounds.
• The Mu2e crystal calorimeter(1801.10002)

Jan. 30, 2018 hep-ex, physics.ins-det
The Mu2e Experiment at Fermilab will search for coherent, neutrino-less conversion of negative muons into electrons in the field of an Aluminum nucleus, $\mu^- + Al \to e^- +Al$. Data collection start is planned for the end of 2021. The dynamics of such charged lepton flavour violating (CLFV) process is well modelled by a two-body decay, resulting in a mono-energetic electron with an energy slightly below the muon rest mass. If no events are observed in three years of running, Mu2e will set an upper limit on the ratio between the conversion and the capture rates %\convrate of $\leq 6\ \times\ 10^{-17}$ (@ 90$\%$ C.L.). R$_{\mu e} = \frac{\mu^- + A(Z,N) \to e^- +A(Z,N)}{\mu^- + A(Z,N) \to \nu_{\mu} ^- +A(Z-1,N)}$ of $\leq 6\ \times\ 10^{-17}$ (@ 90$\%$ C.L.). This will improve the current limit of four order of magnitudes with respect to the previous best experiment. Mu2e complements and extends the current search for $\mu \to e \gamma$ decay at MEG as well as the direct searches for new physics at the LHC. The observation of such CLFV process could be clear evidence for New Physics beyond the Standard Model. Given its sensitivity, Mu2e will be able to probe New Physics at a scale inaccessible to direct searches at either present or planned high energy colliders. To search for the muon conversion process, a very intense pulsed beam of negative muons ($\sim 10^{10} \mu/$ sec) is stopped on an Aluminum target inside a very long solenoid where the detector is also located. The Mu2e detector is composed of a straw tube tracker and a CsI crystals electromagnetic calorimeter. An overview of the physics motivations for Mu2e, the current status of the experiment and the required performances and design details of the calorimeter are presented.
• The calorimeter of the Mu2e experiment at Fermilab(1701.07975)

Jan. 27, 2017 physics.ins-det
The Mu2e experiment at Fermilab looks for Charged Lepton Flavor Violation (CLFV) improving by 4 orders of magnitude the current experimental sensitivity for the muon to electron conversion in a muonic atom. A positive signal could not be explained in the framework of the current Standard Model of particle interactions and therefore would be a clear indication of new physics. In 3 years of data taking, Mu2e is expected to observe less than one background event mimicking the electron coming from muon conversion. Achieving such a level of background suppression requires a deep knowledge of the experimental apparatus: a straw tube tracker, measuring the electron momentum and time, a cosmic ray veto system rejecting most of cosmic ray background and a pure CsI crystal calorimeter, that will measure time of flight, energy and impact position of the converted electron. The calorimeter has to operate in a harsh radiation environment, in a 10-4 Torr vacuum and inside a 1 T magnetic field. The results of the first qualification tests of the calorimeter components are reported together with the energy and time performances expected from the simulation and measured in beam tests of a small scale prototype.
• Irradiation study of UV Silicon Photomultipliers for the Mu2e Calorimeter(1701.06464)

Jan. 23, 2017 physics.ins-det
The Mu2e calorimeter is composed of 1400 un-doped CsI crystals, coupled to large area UV extended Silicon Photomultipliers (SiPMs), arranged in two annular disks. This calorimeter has to provide precise information on energy, timing and position resolutions. It should also be fast enough to handle the high rate background and it must operate and survive in the high radiation environment. Simulation studies estimated that, in the highest irradiated regions, each photo-sensor will absorb a dose of 20 krad and will be exposed to a neutron fluency of 5.5x10^11 n_(1MeV)/cm^2 in three years of running, with a safety factor of 3 included. At the end of 2015, we have concluded an irradiation campaign at the Frascati Neutron Generator (FNG, Frascati, Italy) measuring the response of two different 16 array models from Hamamatsu, which differ for the protection windows and a SiPM from FBK. In 2016, we have carried out two additional irradiation campaigns with neutrons and photons at the Helmholtz-Zentrum Dresden-Rossendorf (HZDR, Dresden, Germany) and at the Calliope gamma irradiation facility at ENEA-Casaccia, respectively. A negligible increment of the leakage current and no gain change have been observed with the dose irradiation. On the other hand, at the end of the neutron irradiation, the gain does not show large changes whilst the leakage current increases by around a factor of 2000. In these conditions, the too high leakage current makes problematic to bias the SiPMs, thus requiring to cool them down to a running temperature of ~0 {\deg}C.