• ### Reduction of stored-particle background by a magnetic pulse method at the KATRIN experiment(1805.01163)

May 3, 2018 physics.ins-det
The KATRIN experiment aims to determine the effective electron neutrino mass with a sensitivity of $0.2\,{\text{eV}/c^2}$ (90\% C.L.) by precision measurement of the shape of the tritium \textbeta-spectrum in the endpoint region. The energy analysis of the decay electrons is achieved by a MAC-E filter spectrometer. A common background source in this setup is the decay of short-lived isotopes, such as $\textsuperscript{219}$Rn and $\textsuperscript{220}$Rn, in the spectrometer volume. Active and passive countermeasures have been implemented and tested at the KATRIN main spectrometer. One of these is the magnetic pulse method, which employs the existing air coil system to reduce the magnetic guiding field in the spectrometer on a short timescale in order to remove low- and high-energy stored electrons. Here we describe the working principle of this method and present results from commissioning measurements at the main spectrometer. Simulations with the particle-tracking software Kassiopeia were carried out to gain a detailed understanding of the electron storage conditions and removal processes.
• ### Calibration of high voltages at the ppm level by the difference of $^{83\mathrm{m}}$Kr conversion electron lines at the KATRIN experiment(1802.05227)

Feb. 14, 2018 physics.ins-det
The neutrino mass experiment KATRIN requires a stability of 3 ppm for the retarding potential at -18.6 kV of the main spectrometer. To monitor the stability, two custom-made ultra-precise high-voltage dividers were developed and built in cooperation with the German national metrology institute Physikalisch-Technische Bundesanstalt (PTB). Until now, regular absolute calibration of the voltage dividers required bringing the equipment to the specialised metrology laboratory. Here we present a new method based on measuring the energy difference of two $^{83\mathrm{m}}$Kr conversion electron lines with the KATRIN setup, which was demonstrated during KATRIN's commissioning measurements in July 2017. The measured scale factor $M=1972.449(10)$ of the high-voltage divider K35 is in agreement with the last PTB calibration four years ago. This result demonstrates the utility of the calibration method, as well as the long-term stability of the voltage divider.
• ### First transmission of electrons and ions through the KATRIN beamline(1802.04167)

Feb. 12, 2018 nucl-ex, physics.ins-det
The Karlsruhe Tritium Neutrino (KATRIN) experiment is a large-scale effort to probe the absolute neutrino mass scale with a sensitivity of 0.2 eV (90% confidence level), via a precise measurement of the endpoint spectrum of tritium beta decay. This work documents several KATRIN commissioning milestones: the complete assembly of the experimental beamline, the successful transmission of electrons from three sources through the beamline to the primary detector, and tests of ion transport and retention. In the First Light commissioning campaign of Autumn 2016, photoelectrons were generated at the rear wall and ions were created by a dedicated ion source attached to the rear section; in July 2017, gaseous Kr-83m was injected into the KATRIN source section, and a condensed Kr-83m source was deployed in the transport section. In this paper we describe the technical details of the apparatus and the configuration for each measurement, and give first results on source and system performance. We have successfully achieved transmission from all four sources, established system stability, and characterized many aspects of the apparatus.
• ### Technical design and commissioning of the KATRIN large-volume air coil system(1712.01078)

Dec. 4, 2017 physics.ins-det
The KATRIN experiment is a next-generation direct neutrino mass experiment with a sensitivity of 0.2 eV (90% C.L.) to the effective mass of the electron neutrino. It measures the tritium $\beta$-decay spectrum close to its endpoint with a spectrometer based on the MAC-E filter technique. The $\beta$-decay electrons are guided by a magnetic field that operates in the mT range in the central spectrometer volume; it is fine-tuned by a large-volume air coil system surrounding the spectrometer vessel. The purpose of the system is to provide optimal transmission properties for signal electrons and to achieve efficient magnetic shielding against background. In this paper we describe the technical design of the air coil system, including its mechanical and electrical properties. We outline the importance of its versatile operation modes in background investigation and suppression techniques. We compare magnetic field measurements in the inner spectrometer volume during system commissioning with corresponding simulations, which allows to verify the system's functionality in fine-tuning the magnetic field configuration. This is of major importance for a successful neutrino mass measurement at KATRIN.
• ### A White Paper on keV Sterile Neutrino Dark Matter(1602.04816)

We present a comprehensive review of keV-scale sterile neutrino Dark Matter, collecting views and insights from all disciplines involved - cosmology, astrophysics, nuclear, and particle physics - in each case viewed from both theoretical and experimental/observational perspectives. After reviewing the role of active neutrinos in particle physics, astrophysics, and cosmology, we focus on sterile neutrinos in the context of the Dark Matter puzzle. Here, we first review the physics motivation for sterile neutrino Dark Matter, based on challenges and tensions in purely cold Dark Matter scenarios. We then round out the discussion by critically summarizing all known constraints on sterile neutrino Dark Matter arising from astrophysical observations, laboratory experiments, and theoretical considerations. In this context, we provide a balanced discourse on the possibly positive signal from X-ray observations. Another focus of the paper concerns the construction of particle physics models, aiming to explain how sterile neutrinos of keV-scale masses could arise in concrete settings beyond the Standard Model of elementary particle physics. The paper ends with an extensive review of current and future astrophysical and laboratory searches, highlighting new ideas and their experimental challenges, as well as future perspectives for the discovery of sterile neutrinos.
• ### Commissioning of the vacuum system of the KATRIN Main Spectrometer(1603.01014)

March 3, 2016 physics.ins-det
The KATRIN experiment will probe the neutrino mass by measuring the beta-electron energy spectrum near the endpoint of tritium beta-decay. An integral energy analysis will be performed by an electro-static spectrometer (Main Spectrometer), an ultra-high vacuum vessel with a length of 23.2 m, a volume of 1240 m^3, and a complex inner electrode system with about 120000 individual parts. The strong magnetic field that guides the beta-electrons is provided by super-conducting solenoids at both ends of the spectrometer. Its influence on turbo-molecular pumps and vacuum gauges had to be considered. A system consisting of 6 turbo-molecular pumps and 3 km of non-evaporable getter strips has been deployed and was tested during the commissioning of the spectrometer. In this paper the configuration, the commissioning with bake-out at 300{\deg}C, and the performance of this system are presented in detail. The vacuum system has to maintain a pressure in the 10^{-11} mbar range. It is demonstrated that the performance of the system is already close to these stringent functional requirements for the KATRIN experiment, which will start at the end of 2016.
• ### Focal-plane detector system for the KATRIN experiment(1404.2925)

Jan. 28, 2015 hep-ex, nucl-ex, physics.ins-det
The focal-plane detector system for the KArlsruhe TRItium Neutrino (KATRIN) experiment consists of a multi-pixel silicon p-i-n-diode array, custom readout electronics, two superconducting solenoid magnets, an ultra high-vacuum system, a high-vacuum system, calibration and monitoring devices, a scintillating veto, and a custom data-acquisition system. It is designed to detect the low-energy electrons selected by the KATRIN main spectrometer. We describe the system and summarize its performance after its final installation.
• ### Modeling of electron emission processes accompanying Radon-$\alpha$-decays within electrostatic spectrometers(1304.1375)

April 4, 2013 physics.ins-det
Electrostatic spectrometers utilized in high-resolution beta-spectroscopy studies such as in the Karlsruhe Tritium Neutrino (KATRIN) experiment have to operate with a background level of less than 10^(-2) counts per second. This limit can be exceeded by even a small number of Rn-219 or Rn-220 atoms being emanated into the volume and undergoing alpha-decay there. In this paper we present a detailed model of the underlying background-generating processes via electron emission by internal conversion, shake-off and relaxation processes in the atomic shells of the Po-215 and Po-216 daughters. The model yields electron energy spectra up to 400 keV and electron multiplicities of up to 20 which are compared to experimental data.
• ### Validation of a model for Radon-induced background processes in electrostatic spectrometers(1304.1379)

April 4, 2013 physics.ins-det
The Karlsruhe Tritium Neutrino (KATRIN) experiment investigating tritium beta-decay close to the endpoint with unprecedented precision has stringent requirements on the background level of less than 10^(-2) counts per second. Electron emission during the alpha-decay of Rn-219 and Rn-220 atoms in the electrostatic spectrometers of KATRIN is a serious source of background exceeding this limit. In this paper we compare extensive simulations of Rn-induced background to specific measurements with the KATRIN pre-spectrometer to fully characterize the observed Rn-background rates and signatures and determine generic Rn emanation rates from the pre-spectrometer bulk material and its vacuum components.
• ### Neutron Beta Decay Studies with Nab(1209.4663)

Sept. 20, 2012 nucl-ex
Precision measurements in neutron beta decay serve to determine the coupling constants of beta decay and allow for several stringent tests of the standard model. This paper discusses the design and the expected performance of the Nab spectrometer.
• ### Monitoring of the operating parameters of the KATRIN Windowless Gaseous Tritium Source(1205.5421)

Sept. 16, 2012 physics.ins-det
The Karlsruhe Tritium Neutrino (KATRIN) experiment will measure the absolute mass scale of neutrinos with a sensitivity of $\m_{\nu}$ = 200 meV/c$^2$ by high-precision spectroscopy close to the tritium beta-decay endpoint at 18.6 keV. Its Windowless Gaseous Tritium Source (WGTS) is a beta-decay source of high intensity ($10^{11}$/s) and stability, where high-purity molecular tritium at 30 K is circulated in a closed loop with a yearly throughput of 10 kg. To limit systematic effects the column density of the source has to be stabilised at the 0.1% level. This requires extensive sensor instrumentation and dedicated control and monitoring systems for parameters such as the beam tube temperature, injection pressure, gas composition and others. Here we give an overview of these systems including a dedicated Laser-Raman system as well as several beta-decay activity monitors. We also report on results of the WGTS demonstrator and other large-scale test experiments giving proof-of-principle that all parameters relevant to the systematics can be controlled and monitored on the 0.1% level or better. As a result of these works, the WGTS systematics can be controlled within stringent margins, enabling the KATRIN experiment to explore the neutrino mass scale with the design sensitivity.
• ### Stochastic Heating by ECR as a Novel Means of Background Reduction in the KATRIN Spectrometers(1205.3729)

July 13, 2012 physics.ins-det
The primary objective of the KATRIN experiment is to probe the absolute neutrino mass scale with a sensitivity of 200 meV (90% C.L.) by precision spectroscopy of tritium beta-decay. To achieve this, a low background of the order of 10^(-2) cps in the region of the tritium beta-decay endpoint is required. Measurements with an electrostatic retarding spectrometer have revealed that electrons, arising from nuclear decays in the volume of the spectrometer, are stored over long time periods and thereby act as a major source of background exceeding this limit. In this paper we present a novel active background reduction method based on stochastic heating of stored electrons by the well-known process of electron cyclotron resonance (ECR). A successful proof-of-principle of the ECR technique was demonstrated in test measurements at the KATRIN pre-spectrometer, yielding a large reduction of the background rate. In addition, we have carried out extensive Monte Carlo simulations to reveal the potential of the ECR technique to remove all trapped electrons within negligible loss of measurement time in the main spectrometer. This would allow the KATRIN experiment attaining its full physics potential.
• ### The KATRIN Pre-Spectrometer at reduced Filter Energy(1203.2444)

March 12, 2012 physics.ins-det
The KArlsruhe TRItium Neutrino experiment, KATRIN, will determine the mass of the electron neutrino with a sensitivity of 0.2 eV (90% C.L.) via a measurement of the beta-spectrum of gaseous tritium near its endpoint of E_0 =18.57 keV. An ultra-low background of about b = 10 mHz is among the requirements to reach this sensitivity. In the KATRIN main beam-line two spectrometers of MAC-E filter type are used in a tandem configuration. This setup, however, produces a Penning trap which could lead to increased background. We have performed test measurements showing that the filter energy of the pre-spectrometer can be reduced by several keV in order to diminish this trap. These measurements were analyzed with the help of a complex computer simulation, modeling multiple electron reflections both from the detector and the photoelectric electron source used in our test setup.
• ### Solving for Micro- and Macro- Scale Electrostatic Configurations Using the Robin Hood Algorithm(1111.5035)

Nov. 21, 2011 nucl-ex, physics.comp-ph
We present a novel technique by which highly-segmented electrostatic configurations can be solved. The Robin Hood method is a matrix-inversion algorithm optimized for solving high density boundary element method (BEM) problems. We illustrate the capabilities of this solver by studying two distinct geometry scales: (a) the electrostatic potential of a large volume beta-detector and (b) the field enhancement present at surface of electrode nano-structures. Geometries with elements numbering in the O(10^5) are easily modeled and solved without loss of accuracy. The technique has recently been expanded so as to include dielectrics and magnetic materials.
• ### Radon induced background processes in the KATRIN pre-spectrometer(1103.6238)

March 31, 2011 physics.ins-det
The KArlsruhe TRItium Neutrino (KATRIN) experiment is a next generation, model independent, large scale tritium beta-decay experiment to determine the effective electron anti-neutrino mass by investigating the kinematics of tritium beta-decay with a sensitivity of 200 meV/c2 using the MAC-E filter technique. In order to reach this sensitivity, a low background level of 0.01 counts per second (cps) is required. This paper describes how the decay of radon in a MAC-E filter generates background events, based on measurements performed at the KATRIN pre-spectrometer test setup. Radon (Rn) atoms, which emanate from materials inside the vacuum region of the KATRIN spectrometers, are able to penetrate deep into the magnetic flux tube so that the alpha-decay of Rn contributes to the background. Of particular importance are electrons emitted in processes accompanying the Rn alpha-decay, such as shake-off, internal conversion of excited levels in the Rn daughter atoms and Auger electrons. While low-energy electrons (< 100 eV) directly contribute to the background in the signal region, higher energy electrons can be stored magnetically inside the volume of the spectrometer. Depending on their initial energy, they are able to create thousands of secondary electrons via subsequent ionization processes with residual gas molecules and, since the detector is not able to distinguish these secondary electrons from the signal electrons, an increased background rate over an extended period of time is generated.
• ### Ion source for tests of ion behavior in the KATRIN beam line(1008.3342)

Jan. 21, 2011 physics.ins-det, astro-ph.IM
An electron-impact ion source based on photoelectron emission was developed for ionization of gases at pressures below 1e-4 mbar in an axial magnetic field in the order of 5 T. The ion source applies only DC fields, which makes it suitable for use in the presence of equipment sensitive to radio-frequency (RF) fields. The ion source was succesfully tested under varying conditions regarding pressure, magnetic field and magnetic-field gradient, and the results were studied with the help of simulations. The processes in the ion source are well understood and possibilities for further optimization of generated ion currents are clarified.
• ### Prototype of an angular-selective photoelectron calibration source for the KATRIN experiment(1011.6365)

Nov. 29, 2010 physics.ins-det
The method of direct neutrino mass determination based on the kinematics of tritium beta decay, which is adopted by the KATRIN experiment, makes use of a large, high-resolution electrostatic spectrometer with magnetic adiabatic collimation. In order to target a sensitivity on the neutrino mass of 0.2 eV/c^2, a detailed understanding of the electromagnetic properties of the electron spectrometer is essential, requiring comprehensive calibration measurements with dedicated electron sources. In this paper we report on a prototype of a photoelectron source providing a narrow energy spread and angular selectivity. Both are key properties for the characterisation of the spectrometer. The angular selectivity is achieved by applying non-parallel strong electric and magnetic fields: Directly after being created, photoelectrons are accelerated rapidly and non-adiabatically by a strong electric field before adiabatic magnetic guiding takes over.
• ### Impact of Neutron Decay Experiments on non-Standard Model Physics(1007.3027)

Oct. 8, 2010 hep-ph, hep-ex, nucl-ex
This paper gives a brief overview of the present and expected future limits on physics beyond the Standard Model (SM) from neutron beta decay, which is described by two parameters only within the SM. Since more than two observables are accessible, the problem is over-determined. Thus, precise measurements of correlations in neutron decay can be used to study the SM as well to search for evidence of possible extensions to it. Of particular interest in this context are the search for right-handed currents or for scalar and tensor interactions. Precision measurements of neutron decay observables address important open questions of particle physics and cosmology, and are generally complementary to direct searches for new physics beyond the SM in high-energy physics. Free neutron decay is therefore a very active field, with a number of new measurements underway worldwide. We present the impact of recent developments.
• ### Effect of a sweeping conductive wire on electrons stored in the Penning trap between the KATRIN spectrometers(0909.3337)

Oct. 3, 2010 nucl-ex, physics.ins-det
The KATRIN experiment is going to search for the mass of the electron antineutrino down to 0.2 eV/c^2. In order to reach this sensitivity the background rate has to be understood and minimised to 0.01 counts per second. One of the background sources is the unavoidable Penning trap for electrons due to the combination of the electric and magnetic fields between the pre- and the main spectrometer at KATRIN. In this article we will show that by sweeping a conducting wire periodically through such a particle trap stored particles can be removed, an ongoing discharge in the trap can be stopped, and the count rate measured with a detector looking at the trap is reduced.
• ### Measuring the proton spectrum in neutron decay - latest results with aSPECT(0811.3851)

Nov. 24, 2008 nucl-ex
The retardation spectrometer aSPECT was built to measure the shape of the proton spectrum in free neutron decay with high precision. This allows us to determine the antineutrino electron angular correlation coefficient a. We aim for a precision more than one order of magnitude better than the present best value, which is Delta_a /a = 5%. In a recent beam time performed at the Institut Laue-Langevin during April / May 2008 we reached a statistical accuracy of about 2% per 24 hours measurement time. Several systematic effects were investigated experimentally. We expect the total relative uncertainty to be well below 5%.
• ### The KATRIN sensitivity to the neutrino mass and to right-handed currents in beta decay(0704.3930)

April 30, 2007 hep-ph
The aim of the KArlsruhe TRItium Neutrino experiment KATRIN is the determination of the absolute neutrino mass scale down to 0.2 eV, with essentially smaller model dependence than from cosmology and neutrinoless double beta decay. For this purpose, the integral electron energy spectrum is measured close to the endpoint of molecular tritium beta decay. The endpoint, together with the neutrino mass, should be fitted from the KATRIN data as a free parameter. The right-handed couplings change the electron energy spectrum close to the endpoint, therefore they have some effect also to the precise neutrino mass determination. The statistical calculations show that, using the endpoint as a free parameter, the unaccounted right-handed couplings constrained by many beta decay experiments can change the fitted neutrino mass value, relative to the true neutrino mass, by not larger than about 5-10 %. Using, incorrectly, the endpoint as a fixed input parameter, the above change of the neutrino mass can be much larger, order of 100 %, and for some cases it can happen that for large true neutrino mass value the fitted neutrino mass squared is negative. Publications using fixed endpoint and presenting large right-handed coupling effects to the neutrino mass determination are not relevant for the KATRIN experiment.
• ### Scalar interaction limits from the beta-neutrino correlation of trapped radioactive atoms(nucl-ex/0412032)

April 14, 2005 nucl-ex
We have set limits on contributions of scalar interactions to nuclear beta decay. A magneto-optical trap (MOT) provides a localized source of atoms suspended in space, so the low-energy recoiling nuclei can freely escape and be detected in coincidence with the beta. This allows reconstruction of the neutrino momentum, and the measurement of the beta-neutrino correlation, in a more direct fashion than previously possible. The beta-neutrino correlation parameter of the 0+ to 0+ pure Fermi decay of 38mK is $\tilde{a}$=0.9981+-0.0030+-0.0037, consistent with the standard model prediction a=1.
• ### Is the Unitarity of the quark-mixing-CKM-matrix violated in neutron $\beta$-decay?(hep-ex/0206058)

June 23, 2002 hep-ex
We report on a new measurement of neutron $\beta$-decay asymmetry. From the result \linebreak $A_0$ = -0.1189(7), we derive the ratio of the axial vector to the vector coupling constant $\lambda$ = ${\it g_A/g_V}$ = -1.2739(19). When included in the world average for the neutron lifetime $\tau$ = 885.7(7)s, this gives the first element of the Cabibbo-Kobayashi-Maskawa (CKM) matrix $V_{ud}$. With this value and the Particle Data Group values for $V_{us}$ and $V_{ub}$, we find a deviation from the unitarity condition for the first row of the CKM matrix of $\Delta$ = 0.0083(28), which is 3.0 times the stated error.