• ### Three-dimensional convolutional neural networks for neutrinoless double-beta decay signal/background discrimination in high-pressure gaseous Time Projection Chamber(1803.01482)

March 7, 2018 hep-ex, physics.data-an
A distinct advantage of high-pressure gaseous Time Projection Chamber in the search of neutrinoless double-beta decay is that the ionization charge tracks resulting from particle interactions are extended and the detector equipped with appropriate charge readout captures the full three-dimensional charge distribution. Such information provides a crucial extra-handle for discriminating signal events against backgrounds. We adapted 3-dimensional convolutional and residual neural networks on the simulated double-beta and background charge tracks and tested their capabilities in classifying the two types of events. We show that both the 3D structure and the overall depth of the neural networks significantly improve the accuracy of the classifier over previous work. We also studied their performance under various spatial granularity as well as charge diffusion and noise conditions. The results indicate that the methods are stable and generalize well despite varying experimental conditions.
• Substantial experimental and theoretical efforts worldwide are devoted to explore the phase diagram of strongly interacting matter. At LHC and top RHIC energies, QCD matter is studied at very high temperatures and nearly vanishing net-baryon densities. There is evidence that a Quark-Gluon-Plasma (QGP) was created at experiments at RHIC and LHC. The transition from the QGP back to the hadron gas is found to be a smooth cross over. For larger net-baryon densities and lower temperatures, it is expected that the QCD phase diagram exhibits a rich structure, such as a first-order phase transition between hadronic and partonic matter which terminates in a critical point, or exotic phases like quarkyonic matter. The discovery of these landmarks would be a breakthrough in our understanding of the strong interaction and is therefore in the focus of various high-energy heavy-ion research programs. The Compressed Baryonic Matter (CBM) experiment at FAIR will play a unique role in the exploration of the QCD phase diagram in the region of high net-baryon densities, because it is designed to run at unprecedented interaction rates. High-rate operation is the key prerequisite for high-precision measurements of multi-differential observables and of rare diagnostic probes which are sensitive to the dense phase of the nuclear fireball. The goal of the CBM experiment at SIS100 (sqrt(s_NN) = 2.7 - 4.9 GeV) is to discover fundamental properties of QCD matter: the phase structure at large baryon-chemical potentials (mu_B > 500 MeV), effects of chiral symmetry, and the equation-of-state at high density as it is expected to occur in the core of neutron stars. In this article, we review the motivation for and the physics programme of CBM, including activities before the start of data taking in 2022, in the context of the worldwide efforts to explore high-density QCD matter.
• ### A beam monitor using silicon pixel sensors for hadron therapy(1611.07147)

We report the design and test results of a beam monitor developed for online monitoring in hadron therapy. The beam monitor uses eight silicon pixel sensors, \textit{Topmetal-${II}^-$}, as the anode array. \textit{Topmetal-${II}^-$} is a charge sensor designed in a CMOS 0.35 $\mu$m technology. Each \textit{Topmetal-${II}^-$} sensor has $72\times72$ pixels and the pixel size is $83\times83$ $\mu$m$^2$. In our design, the beam passes through the beam monitor without hitting the electrodes, making the beam monitor especially suitable for monitoring heavy ion beams. This design also reduces radiation damage to the beam monitor itself. The beam monitor is tested with a carbon ion beam at the Heavy Ion Research Facility in Lanzhou (HIRFL). Results indicate that the beam monitor can measure position, incidence angle and intensity of the beam with a position resolution better than 20 $\mu$m, angular resolution about 0.5$^\circ$ and intensity statistical accuracy better than 2$\%$.
• ### A highly pixelated CdZnTe detector based on \textit{Topmetal-${II}^-$} sensor(1611.04142)

Jan. 11, 2017 physics.ins-det
\textit{Topmetal-${II}^-$} is a low noise CMOS pixel direct charge sensor with a pitch of 83$\mu m$. CdZnTe is an excellent semiconductor material for radiation detection. The combination of CdZnTe and the sensor makes it possible to build a detector with high spatial resolution. In our experiments, an epoxy adhesive is used as the conductive medium to connect the sensor and Cadmium Zinc Telluride (CdZnTe). The diffusion coefficient and charge efficiency of electrons are measured at a low bias voltage of -2 Volts, and the image of a single alpha is clear with a reasonable spatial resolution. The detector of such structure has the potential to be applied in X-ray imaging systems with a further improvements of the sensor.
• ### Detailed study of the column-based priority logic readout of Topmetal-II- CMOS pixel direct charge sensor(1608.06399)

Aug. 23, 2016 physics.ins-det
We present the detailed study of the digital readout of Topmetal-II- CMOS pixel direct charge sensor. Topmetal-II- is an integrated sensor with an array of 72X72 pixels each capable of directly collecting external charge through exposed metal electrodes in the topmost metal layer. In addition to the time-shared multiplexing readout of the analog output from Charge Sensitive Amplifiers in each pixel, hits are also generated through comparators with individually DAC settable thresholds in each pixel. The hits are read out via a column-based priority logic structure, retaining both hit location and time information. The in-array column-based priority logic is fully combinational hence there is no clock distributed in the pixel array. Sequential logic and clock are placed on the peripheral of the array. We studied the detailed working behavior and performance of this readout, and demonstrated its potential in imaging applications.
• ### Test of \textit{Topmetal-${II}^-$} In Liquid Nitrogen For Cryogenic Temperature TPCs(1601.06955)

June 7, 2016 hep-ex, physics.ins-det
\textit{Topmetal-${II}^-$} is a highly pixelated direct charge sensor that contains a 72${\times}$72 pixel array of 83${\mu}$m pitch size. The key feature of \textit{Topmetal-${II}^-$} is that it can directly collect charges via metal nodes of each pixel to form two-dimensional images of charge cloud distributions. \textit{Topmetal-${II}^-$} was proved to measure charged particles without amplification at room temperature. To measure its performance at cryogenic temperature, a \textit{Topmetal-${II}^-$} sensor is embedded into a liquid nitrogen dewar. The results presented in this paper show that \textit{Topmetal-${II}^-$} can also operate well at this low temperature with a noise (ENC) of 12 e$^-$ lower than that at room temperature (13 e$^-$). From the noise perspective, \textit{Topmetal-${II}^-$} is a promising candidate for the next generation readout of liquid argon and xenon Time Projection Chamber (TPC) used in experiments searching for neutrinoless double beta decay and dark matter.
• ### Development of a highly pixelated direct charge sensor, Topmetal-I, for ionizing radiation imaging(1407.3712)

Using industrial standard 0.35{\mu}m CMOS Integrated Circuit process, we realized a highly pixelated sensor that directly collects charge via metal nodes placed on the top of each pixel and forms two dimensional images of charge cloud distribution. The first version, Topmetal-I, features a 64x64 pixel array of 80{\mu}m pitch size. Direct charge calibration reveals an average capacitance of 210fF per pixel. The charge collection noise is near the thermal noise limit. With the readout, individual pixel channels exhibit a most probable equivalent noise charge of 330e-.