
Fast digitisers and digital pulse processing have been widely used for
spectral application and pulse shape discrimination (PSD) owing to their
advantages in terms of compactness, higher trigger rates, offline analysis,
etc. Meanwhile, the noise of readout electronics is usually trivial for
organic, plastic, or liquid scintillator with PSD ability because of their poor
intrinsic energy resolution. However, LaBr3(Ce) has been widely used for its
excellent energy resolution and has been proven to have PSD ability for
alpha/gamma particles. Therefore, designing a digital acquisition system for
such scintillators as LaBr3(Ce) with both optimal energy resolution and
promising PSD ability is worthwhile. Several experimental research studies
about the choice of digitiser properties for liquid scintillators have already
been conducted in terms of the sampling rate and vertical resolution.
Quantitative analysis on the influence of waveform digitisers, that is, fast
amplifier (optional), sampling rates, and vertical resolution, on both
applications is still lacking. The present paper provides quantitative analysis
of these factors and, hence, general rules about the optimal design of
digitisers for both energy resolution and PSD application according to the
noise analysis of timevariant gated charge integration.

In neutrinoless double beta (0{\nu}{\beta}{\beta}) decay experiments, the
diversity of topological signatures of different particles provides an
important tool to distinguish double beta events from background events and
reduce background rates. Aiming at suppressing the singleelectron backgrounds
which are most challenging, several groups have established Monte Carlo
simulation packages to study the topological characteristics of singleelectron
events and 0{\nu}{\beta}{\beta} events and develop methods to differentiate
them. In this paper, applying the knowledge of graph theory, a new topological
signature called REF track (Refined EnergyFiltered track) is proposed and
proven to be an accurate approximation of the real particle trajectory. Based
on the analysis of the energy depositions along the REF track of
singleelectron events and 0{\nu}{\beta}{\beta} events, the REF energy
deposition models for both events are proposed to indicate the significant
differences between them. With these differences, this paper presents a new
discrimination method, which, in the Monte Carlo simulation, achieved a
singleelectron rejection factor of 93.8+0.3 (stat.)% as well as a
0{\nu}{\beta}{\beta} efficiency of 85.6+0.4 (stat.)% with optimized parameters
in CdZnTe.

The key to photoelectric Xray polarimetry is the determination of the
emission direction of photoelectrons. Because of the low mass of an electron,
the ionisation trajectory is not straight and the useful information needed for
polarimetry is stored mostly in the initial part of the track where less energy
is deposited. We present a new algorithm, based on the shortest path problem in
graph theory, to reconstruct the 2D electron track from the measured image that
is blurred due to transversal diffusion along drift and multiplication in the
gas chamber. Compared with previous methods based on moment analysis, this
algorithm allows us to identify the photoelectric interaction point more
accurately and precisely for complicated tracks resulting from high energy
photons or low pressure chambers. This leads to a better position resolution
and a higher degree of modulation toward high energy Xrays. The new algorithm
is justified using simulations and measurements with the gas pixel detector
(GPD), and it should also work for other polarimetric techniques such as a time
projection chamber (TPC). As the improvement is restricted in the high energy
band, this new algorithm shows limited improvement for the sensitivity of GPD
polarimeters, but it may have a larger potential for lowpressure TPC
polarimeters.

An underwater in situ gamma ray spectrometer based on LaBr3 was developed and
optimized to monitor marine radioactivity. The intrinsic background mainly from
La138 and Ac227 of LaBr3 was well determined by low background measurement and
pulse shape discrimination method. A method of selfcalibration using three
internal contaminant peaks was proposed to eliminate the peak shift during long
term monitoring. With experiments under different temperatures, the method was
proved to be helpful for maintaining long term stability. To monitor the marine
radioactivity, the spectrometer efficiency was calculated via water tank
experiment as well as Monte Carlo simulation.

The LaBr3:Ce scintillator has been widely studied for nuclear spectroscopy
because of its optimal energy resolution (<3%@ 662 keV) and time resolution
(~300 ps). Despite these promising properties, the intrinsic radiation
background of LaBr3:Ce is a critical issue, and pulse shape discrimination
(PSD) has been shown to be an efficient potential method to suppress the alpha
background from the 227Ac. In this paper, the charge comparison method (CCM)
for alpha and gamma discrimination in LaBr3:Ce is quantitatively analysed and
compared with two other typical PSD methods using digital pulse processing. The
algorithm parameters and discrimination efficiency are calculated for each
method. Moreover, for the CCM, the correlation between the CCM feature value
distribution and the total charge (energy) is studied, and a fitting equation
for the correlation is inferred and experimentally verified. Using the
equations, an energydependent threshold can be chosen to optimize the
discrimination efficiency. Additionally, the experimental results show a
potential application in lowactivity highenergy {\gamma} measurement by
suppressing the alpha background.

A prototype of LaBr3:Ce in situ gammaray spectrometer for marine
environmental monitoring is developed and applied for in situ measurement. A
3inch LaBr3:Ce scintillator is used in the detector, and a digital pulse
process electronics is chosen as the pulse height analyzer. For this prototype,
the energy response of the spectrometer is linear and the energy resolution of
662keV is 2.6% (much better than NaI). With the measurement of the prototype in
a water tank filled with 137Cs, the detect efficiency for 137Cs is (0.288
0.01)cps/(Bq/L), which is close to the result of Monte Carlo simulation,
0.283cps/(Bq/L). With this measurement, the MDAC for 137Cs in one hour has been
calculated to 0.78Bq/L, better than that of NaI(Tl) insitu gamma spectrometer,
which is ~1.0Bq/L.