• ### Slow Liquid Scintillator Candidates for MeV-scale Neutrino Experiments(1708.07781)

Feb. 15, 2019 physics.ins-det
Slow liquid scintillator Cherenkov detectors have been proposed as part of several future neutrino experiments because they can provide both directionality and energy measurements. This feature is expected to enhance the sensitivities for MeV-scale neutrino physics, including solar physics, the search for supernova relic neutrino, and the study of geo-sciences. In this study, the characteristics of a slow liquid scintillator were investigated, along with the light yields and decay time constants for various combinations of linear alkylbenzene (LAB), 2,5-diphenyloxazole (PPO), and 1,4-bis (2-methylstyryl)-benzene (bis-MSB). The results of our study indicated that LAB with 0.07 g/L of PPO and 13 mg/L of bis-MSB was the best candidate for an effective separation between Cherenkov and scintillation lights with a reasonably high light yield.
• ### Light Absorption Properties of the High Quality Linear Alkylbenzene for the JUNO Experiment(1801.08363)

Jan. 25, 2018 hep-ex, physics.ins-det
The Jiangmen Underground Neutrino Observatory (JUNO), a 20 kton multi-purpose underground liquid scintillator detector designed to determine the neutrino mass hierarchy, and measure the neutrino oscillation parameters. The excellent energy resolution and the large fiducial volume anticipated for the JUNO detector offer exciting opportunities for addressing many important topics in neutrino and astro-particle physics. Linear alkylbenzene (LAB) will be used as the solvent for the liquid scintillation system in the central detector of JUNO. The light attenuation lengths of LAB should be comparable to the diameter of the JUNO detector, hence very good optical transparency is required. However, the presence of impurities in the LAB renders an intrinsic limit for the transparency. This work focuses on the study of the effects of organic impurities in the LAB, and their light absorption properties particularly in the wavelength region of 350 to 450 nm. we have prepared LAB samples and measured their light attenuation lengths. These samples were then analyzed by a gas chromatography mass spectrometry, and the structure formulas of organic impurities were ascertained. These impurities' light absorption properties in the wavelength region of 350 to 550 nm were theoretically investigated with PCM TDDFT. The overall optical transparency of the LAB samples was studied, which would further help us in promoting the LAB preparation technique for the mass production thereof, thus improving the transparency of the high quality LAB samples in the near future.
• ### Deep Learning the Effects of Photon Sensors on the Event Reconstruction Performance in an Antineutrino Detector(1711.00607)

Dec. 24, 2017 hep-ex, physics.data-an
We provide a fast approach incorporating the usage of deep learning for evaluating the effects of photon sensors in an antineutrino detector on the event reconstruction performance therein. This work is an attempt to harness the power of deep learning for detector designing and upgrade planning. Using the Daya Bay detector as a benchmark case and the vertex reconstruction performance as the objective for the deep neural network, we find that the photomultiplier tubes (PMTs) have different relative importance to the vertex reconstruction. More importantly, the vertex position resolutions for the Daya Bay detector follow approximately a multi-exponential relationship with respect to the number of PMTs and hence, the coverage. This could also assist in deciding on the merits of installing additional PMTs for future detector plans. The approach could easily be used with other objectives in place of vertex reconstruction.
• ### Light Attenuation Length of High Quality Linear Alkyl Benzene as Liquid Scintillator Solvent for the JUNO Experiment(1703.01867)

March 6, 2017 hep-ex, physics.ins-det
The Jiangmen Underground Neutrino Observatory (JUNO) is a multipurpose neutrino experiment with a 20 kt liquid scintillator detector designed to determine the neutrino mass hierarchy, and measure the neutrino oscillation parameters. Linear alkyl benzene (LAB) will be used as the solvent for the liquid scintillation system in the central detector of JUNO. For this purpose, we have prepared LAB samples, and have measured their light attenuation lengths, with one achieving a length of 25.8 m, comparable to the diameter of the JUNO detector.
• The Jiangmen Underground Neutrino Observatory (JUNO), a 20 kton multi-purpose underground liquid scintillator detector, was proposed with the determination of the neutrino mass hierarchy as a primary physics goal. It is also capable of observing neutrinos from terrestrial and extra-terrestrial sources, including supernova burst neutrinos, diffuse supernova neutrino background, geoneutrinos, atmospheric neutrinos, solar neutrinos, as well as exotic searches such as nucleon decays, dark matter, sterile neutrinos, etc. We present the physics motivations and the anticipated performance of the JUNO detector for various proposed measurements. By detecting reactor antineutrinos from two power plants at 53-km distance, JUNO will determine the neutrino mass hierarchy at a 3-4 sigma significance with six years of running. The measurement of antineutrino spectrum will also lead to the precise determination of three out of the six oscillation parameters to an accuracy of better than 1\%. Neutrino burst from a typical core-collapse supernova at 10 kpc would lead to ~5000 inverse-beta-decay events and ~2000 all-flavor neutrino-proton elastic scattering events in JUNO. Detection of DSNB would provide valuable information on the cosmic star-formation rate and the average core-collapsed neutrino energy spectrum. Geo-neutrinos can be detected in JUNO with a rate of ~400 events per year, significantly improving the statistics of existing geoneutrino samples. The JUNO detector is sensitive to several exotic searches, e.g. proton decay via the $p\to K^++\bar\nu$ decay channel. The JUNO detector will provide a unique facility to address many outstanding crucial questions in particle and astrophysics. It holds the great potential for further advancing our quest to understanding the fundamental properties of neutrinos, one of the building blocks of our Universe.
• ### Some new progress on the light absorption properties of linear alkyl benzene solvent(1504.05444)

April 22, 2015 hep-ex, physics.ins-det
Linear alkyl benzene (LAB) will be used as the solvent of a liquid scintillator mixture for the JUNO antineutrino experiment in the near future. Its light absorption property should therefore be understood prior to its effective use in the experiment. Attenuation length measurements at a light wavelength of 430 nm have been performed on samples of LAB prepared for the purpose of the JUNO experiment. Inorganic impurities in LAB have also been studied for their possibilities of light absorption in our wavelength of interest. In view of a tentative plan by the JUNO collaboration to utilize neutron capture with hydrogen in the detector, we have also presented in this work, a study on the carbon-hydrogen ratio and the relationship thereof with the attenuation length of the samples.
• ### The study of the aging behavior on large area MCP-PMT(1503.02497)

March 9, 2015 hep-ex, physics.ins-det
Aging experiments of a novel type of large area MCP-PMT made by JUNO collaboration were conducted.In these aging experiments, the multi-photoelectron spectrum and single photoelectron spectrum were measured daily,as well as the MCP resistance of the second PMT before and after the experiment. Two PMTs were aged successivelyfor cross check. The first PMT was aged for 52 days, while the other one was aged for 84 days. In order to study the mechanism of the aging process, the high voltage on the second PMT was increased to accelerate its aging process when the cumulative output of charge from its anode was about 4 C. From our study, it can be known that large area MCP-PMT aging had a strong relationship with the related MCPs. In accordance with the PMT aging curve,a PMT aging model was setup and a general aging formula was given.
• ### Waterproofed Photomultiplier Tube Assemblies for the Daya Bay Reactor Neutrino Experiment(1502.06652)

Feb. 23, 2015 hep-ex, physics.ins-det
In the Daya Bay Reactor Neutrino Experiment 960 20-cm-diameter waterproof photomultiplier tubes are used to instrument three water pools as Cherenkov detectors for detecting cosmic-ray muons. Of these 960 photomultiplier tubes, 341 are recycled from the MACRO experiment. A systematic program was undertaken to refurbish them as waterproof assemblies. In the context of passing the water leakage check, a success rate better than 97% was achieved. Details of the design, fabrication, testing, operation, and performance of these waterproofed photomultiplier-tube assemblies are presented.
• ### The high-speed after pulse measurement system for PMT(1404.6125)

April 24, 2014 hep-ex, physics.ins-det
A system employing a desktop FADC has been developed to investigate the features of 8 inches Hamamatsu PMT. The system stands out for its high-speed and informative results as a consequence of adopting fast waveform sampling technology. Recording full waveforms allows us to perform digital signal processing, pulse shape analysis, and precision timing extraction. High precision after pulse time and charge distribution characteristics are presented in this manuscript. Other photomultipliers characteristics, such as dark rate and transit time spread, can also be obtained by exploiting waveform analysis using this system.
• ### Production of Gadolinium-loaded Liquid Scintillator for the Daya Bay Reactor Neutrino Experiment(1402.6694)

Feb. 27, 2014 hep-ex, physics.ins-det
We report on the production and characterization of liquid scintillators for the detection of electron antineutrinos by the Daya Bay Reactor Neutrino Experiment. One hundred eighty-five tons of gadolinium-loaded (0.1% by mass) liquid scintillator (Gd-LS) and two hundred tons of unloaded liquid scintillator (LS) were successfully produced from a linear-alkylbenzene (LAB) solvent in six months. The scintillator properties, the production and purification systems, and the quality assurance and control (QA/QC) procedures are described.