• During the shutdown of the CERN Large Hadron Collider in 2013-2014, an additional pixel layer was installed between the existing Pixel detector of the ATLAS experiment and a new, smaller radius beam pipe. The motivation for this new pixel layer, the Insertable B-Layer (IBL), was to maintain or improve the robustness and performance of the ATLAS tracking system, given the higher instantaneous and integrated luminosities realised following the shutdown. Because of the extreme radiation and collision rate environment, several new radiation-tolerant sensor and electronic technologies were utilised for this layer. This paper reports on the IBL construction and integration prior to its operation in the ATLAS detector.
• Quadratic Hyperbolicity Preservers & Multiplier Sequences(1304.1841)

April 6, 2013 math.CA, math.CV
It is known (see \cite[Br\"and\'en, Lemma 2.7]{B10}) that a necessary condition for $T:=\sum Q_k(x) D^k$ to be hyperbolicity preserving is that $Q_k(x)$ and $Q_{k-1}(x)$ have interlacing zeros. We characterize all quadratic linear operators, as a consequence we find several classes of $P_n$-multiplier sequence.
• Development of a custom on-line ultrasonic vapour analyzer/flowmeter for the ATLAS inner detector, with application to gaseous tracking and Cherenkov detectors(1210.8045)

Oct. 30, 2012 hep-ex, physics.ins-det
Precision sound velocity measurements can simultaneously determine binary gas composition and flow. We have developed an analyzer with custom electronics, currently in use in the ATLAS inner detector, with numerous potential applications. The instrument has demonstrated ~0.3% mixture precision for C3F8/C2F6 mixtures and < 10-4 resolution for N2/C3F8 mixtures. Moderate and high flow versions of the instrument have demonstrated flow resolutions of +/- 2% F.S. for flows up to 250 l.min-1, and +/- 1.9% F.S. for linear flow velocities up to 15 ms-1; the latter flow approaching that expected in the vapour return of the thermosiphon fluorocarbon coolant recirculator being built for the ATLAS silicon tracker.
• A combined ultrasonic flow meter and binary vapour mixture analyzer for the ATLAS silicon tracker(1210.4835)

Oct. 17, 2012 hep-ex, physics.ins-det
An upgrade to the ATLAS silicon tracker cooling control system may require a change from C3F8 (octafluoro-propane) evaporative coolant to a blend containing 10-25% of C2F6 (hexafluoro-ethane). Such a change will reduce the evaporation temperature to assure thermal stability following radiation damage accumulated at full LHC luminosity. Central to this upgrade is a new ultrasonic instrument in which sound transit times are continuously measured in opposite directions in flowing gas at known temperature and pressure to deduce the C3F8/C2F6 flow rate and mixture composition. The instrument and its Supervisory, Control and Data Acquisition (SCADA) software are described in this paper. Several geometries for the instrument are in use or under evaluation. An instrument with a pinched axial geometry intended for analysis and measurement of moderate flow rates has demonstrated a mixture resolution of 3.10-3 for C3F8/C2F6 molar mixtures with 20%C2F6, and a flow resolution of 2% of full scale for mass flows up to 30gs-1. In mixtures of widely-differing molecular weight (mw), higher mixture precision is possible: a sensitivity of <5.10-5 to leaks of C3F8 into part of the ATLAS tracker nitrogen envelope (mw difference 160) has been seen. An instrument with an angled sound path geometry has been developed for use at high fluorocarbon mass flow rates of around 1.2 kgs-1 - corresponding to full flow in a new 60kW thermosiphon recirculator under construction for the ATLAS silicon tracker. Extensive computational fluid dynamics studies were performed to determine the preferred geometry (ultrasonic transducer spacing and placement, together with the sound crossing angle with respect to the vapour flow direction). A prototype with 45deg crossing angle has demonstrated a flow resolution of 1.9% of full scale for linear flow velocities up to 15 ms-1. The instrument has many potential applications.
• Test Beam Results of 3D Silicon Pixel Sensors for the ATLAS upgrade(1101.4203)

Jan. 21, 2011 physics.ins-det
Results on beam tests of 3D silicon pixel sensors aimed at the ATLAS Insertable-B-Layer and High Luminosity LHC (HL-LHC)) upgrades are presented. Measurements include charge collection, tracking efficiency and charge sharing between pixel cells, as a function of track incident angle, and were performed with and without a 1.6 T magnetic field oriented as the ATLAS Inner Detector solenoid field. Sensors were bump bonded to the front-end chip currently used in the ATLAS pixel detector. Full 3D sensors, with electrodes penetrating through the entire wafer thickness and active edge, and double-sided 3D sensors with partially overlapping bias and read-out electrodes were tested and showed comparable performance.
• Dialogue Act Modeling for Automatic Tagging and Recognition of Conversational Speech(cs/0006023)

Oct. 26, 2000 cs.CL
We describe a statistical approach for modeling dialogue acts in conversational speech, i.e., speech-act-like units such as Statement, Question, Backchannel, Agreement, Disagreement, and Apology. Our model detects and predicts dialogue acts based on lexical, collocational, and prosodic cues, as well as on the discourse coherence of the dialogue act sequence. The dialogue model is based on treating the discourse structure of a conversation as a hidden Markov model and the individual dialogue acts as observations emanating from the model states. Constraints on the likely sequence of dialogue acts are modeled via a dialogue act n-gram. The statistical dialogue grammar is combined with word n-grams, decision trees, and neural networks modeling the idiosyncratic lexical and prosodic manifestations of each dialogue act. We develop a probabilistic integration of speech recognition with dialogue modeling, to improve both speech recognition and dialogue act classification accuracy. Models are trained and evaluated using a large hand-labeled database of 1,155 conversations from the Switchboard corpus of spontaneous human-to-human telephone speech. We achieved good dialogue act labeling accuracy (65% based on errorful, automatically recognized words and prosody, and 71% based on word transcripts, compared to a chance baseline accuracy of 35% and human accuracy of 84%) and a small reduction in word recognition error.
• Can Prosody Aid the Automatic Classification of Dialog Acts in Conversational Speech?(cs/0006024)

June 11, 2000 cs.CL
Identifying whether an utterance is a statement, question, greeting, and so forth is integral to effective automatic understanding of natural dialog. Little is known, however, about how such dialog acts (DAs) can be automatically classified in truly natural conversation. This study asks whether current approaches, which use mainly word information, could be improved by adding prosodic information. The study is based on more than 1000 conversations from the Switchboard corpus. DAs were hand-annotated, and prosodic features (duration, pause, F0, energy, and speaking rate) were automatically extracted for each DA. In training, decision trees based on these features were inferred; trees were then applied to unseen test data to evaluate performance. Performance was evaluated for prosody models alone, and after combining the prosody models with word information -- either from true words or from the output of an automatic speech recognizer. For an overall classification task, as well as three subtasks, prosody made significant contributions to classification. Feature-specific analyses further revealed that although canonical features (such as F0 for questions) were important, less obvious features could compensate if canonical features were removed. Finally, in each task, integrating the prosodic model with a DA-specific statistical language model improved performance over that of the language model alone, especially for the case of recognized words. Results suggest that DAs are redundantly marked in natural conversation, and that a variety of automatically extractable prosodic features could aid dialog processing in speech applications.
• Preliminary Results for LP VPE X-Ray Detectors(physics/9709035)

Sept. 24, 1997 physics.ins-det
Thick epitaxial layers have been grown using Low Pressure Vapour Phase Epitaxy techniques with low free carrier concentrations . This type of material is attractive as a medium for X-ray detection, because of its high conversion efficiency for X-rays in the medically interesting energy range.