• ### When "Optimal Filtering" Isn't(1611.07856)

The so-called "optimal filter" analysis of a microcalorimeter's x-ray pulses is statistically optimal only if all pulses have the same shape, regardless of energy. The shapes of pulses from a nonlinear detector can and do depend on the pulse energy, however. A pulse-fitting procedure that we call "tangent filtering" accounts for the energy dependence of the shape and should therefore achieve superior energy resolution. We take a geometric view of the pulse-fitting problem and give expressions to predict how much the energy resolution stands to benefit from such a procedure. We also demonstrate the method with a case study of K-line fluorescence from several 3d transition metals. The method improves the resolution from 4.9 eV to 4.2 eV at the Cu K$\alpha$ line (8.0keV).
• ### First application of superconducting transition-edge-sensor microcalorimeters to hadronic-atom x-ray spectroscopy(1608.05436)

High-resolution pionic-atom x-ray spectroscopy was performed with an x-ray spectrometer based on a 240-pixel array of superconducting transition-edge-sensor (TES) microcalorimeters at the piM1 beam line of the Paul Scherrer Institute. X-rays emitted by pionic carbon via the 4f->3d transition and the parallel 4d->3p transition were observed with a full-width-at-half-maximum energy resolution of 6.8 eV at 6.4 keV. Measured x-ray energies are consistent with calculated electromagnetic values which considered the strong-interaction effect assessed via the Seki-Masutani potential for the 3p energy level, and favor the electronic population of two filled 1s electrons in the K-shell. Absolute energy calibration with an uncertainty of 0.1 eV was demonstrated under a high-rate hadron beam condition of 1.45 MHz. This is the first application of a TES spectrometer to hadronic-atom x-ray spectroscopy and is an important milestone towards next-generation high-resolution kaonic-atom x-ray spectroscopy.
• ### The Practice of Pulse Processing(1511.03950)

Nov. 12, 2015 physics.ins-det
The analysis of data from x-ray microcalorimeters requires great care; their excellent intrinsic energy resolution cannot usually be achieved in practice without a statistically near-optimal pulse analysis and corrections for important systematic errors. We describe the essential parts of a pulse-analysis pipeline for data from x-ray microcalorimeters, including steps taken to reduce systematic gain variation and the unwelcome dependence of filtered pulse heights on the exact pulse-arrival time. We find these steps collectively to be essential tools for getting the best results from a microcalorimeter-based x-ray spectrometer.
• ### Microcalorimeter Spectroscopy at High Pulse Rates: a Multi-Pulse Fitting Technique(1503.05989)

March 31, 2015 astro-ph.IM
Transition edge sensor microcalorimeters can measure x-ray and gamma-ray energies with very high energy resolution and high photon-collection efficiency. For this technology to reach its full potential in future x-ray observatories, each sensor must be able to measure hundreds or even thousands of photon energies per second. Current "optimal filtering" approaches to achieve the best possible energy resolution work only for photons well isolated in time, a requirement in direct conflict with the need for high-rate measurements. We describe a new analysis procedure to allow fitting for the pulse height of all photons even in the presence of heavy pulse pile-up. In the limit of isolated pulses, the technique reduces to the standard optimal filtering with long records. We employ reasonable approximations to the noise covariance function in order to render multi-pulse fitting computationally viable even for very long data records. The technique is employed to analyze x-ray emission spectra at 600 eV and 6 keV at rates up to 250 counts per second in microcalorimeters having exponential signal decay times of approximately 1.2 ms.
• ### High-resolution kaonic-atom x-ray spectroscopy with transition-edge-sensor microcalorimeters(1401.8039)

Feb. 19, 2014 nucl-ex, physics.ins-det
We are preparing for an ultra-high resolution x-ray spectroscopy of kaonic atoms using an x-ray spectrometer based on an array of superconducting transition-edge-sensor microcalorimeters developed by NIST. The instrument has excellent energy resolutions of 2 - 3 eV (FWHM) at 6 keV and a large collecting area of about 20 mm^2. This will open new door to investigate kaon-nucleus strong interaction and provide new accurate charged-kaon mass value.
• ### Transition-Edge Sensors for Particle Induced X-ray Emission Measurements(1310.1733)

In this paper we present a new measurement setup, where a transitionedge sensor detector array is used to detect X-rays in particle induced X-ray emission measurements with a 2 MeV proton beam. Transition-edge sensors offer orders of magnitude improvement in energy resolution compared to conventional silicon or germanium detectors, making it possible to recognize spectral lines in materials analysis that have previously been impossible to resolve, and to get chemical information from the elements. Our sensors are cooled to the operation temperature (65 mK) with a cryogen-free adiabatic demagnetization refrigerator, which houses a specially designed X-ray snout that has a vacuum tight window to couple in the radiation. For the best pixel, the measured instrumental energy resolution was 3.06 eV full width at half maximum at 5.9 keV.We discuss the current status of the project, benefits of transition-edge sensors when used in particle induced X-ray emission spectroscopy, and the results from the first measurements.
• ### SCUBA-2: The 10000 pixel bolometer camera on the James Clerk Maxwell Telescope(1301.3650)

Jan. 16, 2013 physics.ins-det, astro-ph.IM
SCUBA-2 is an innovative 10000 pixel bolometer camera operating at submillimetre wavelengths on the James Clerk Maxwell Telescope (JCMT). The camera has the capability to carry out wide-field surveys to unprecedented depths, addressing key questions relating to the origins of galaxies, stars and planets. With two imaging arrays working simultaneously in the atmospheric windows at 450 and 850 microns, the vast increase in pixel count means that SCUBA-2 maps the sky 100-150 times faster than the previous SCUBA instrument. In this paper we present an overview of the instrument, discuss the physical characteristics of the superconducting detector arrays, outline the observing modes and data acquisition, and present the early performance figures on the telescope. We also showcase the capabilities of the instrument via some early examples of the science SCUBA-2 has already undertaken. In February 2012, SCUBA-2 began a series of unique legacy surveys for the JCMT community. These surveys will take 2.5 years and the results are already providing complementary data to the shorter wavelength, shallower, larger-area surveys from Herschel. The SCUBA-2 surveys will also provide a wealth of information for further study with new facilities such as ALMA, and future telescopes such as CCAT and SPICA.
• ### Code-division multiplexing for x-ray microcalorimeters(1201.6289)

Jan. 30, 2012 physics.ins-det, astro-ph.IM
We demonstrate the code-division multiplexed (CDM) readout of eight transition-edge sensor microcalorimeters. The energy resolution is 3.0 eV (full width at half-maximum) or better at 5.9 keV, with a best resolution of 2.3 eV and a mean of 2.6 eV over the seven modulated detectors. The flux-summing CDM system is described and compared with similar time-division multiplexed (TDM) readout. We show that the sqrt(Npixels) multiplexing disadvantage associated with TDM is not present in CDM. This demonstration establishes CDM as both a simple route to higher performance in existing TDM microcalorimetric experiments and a long-term approach to reaching higher multiplexing factors.
• ### Optimization and analysis of code-division multiplexed TES microcalorimeters(1110.2531)

Oct. 11, 2011 physics.ins-det, astro-ph.IM
We are developing code-division multiplexing (CDM) systems for transition-edge sensor arrays with the goal of reaching multiplexing factors in the hundreds. We report on x-ray measurements made with a four-channel prototype CDM system that employs a flux-summing architecture, emphasizing data-analysis issues. We describe an empirical method to determine the demodulation matrix that minimizes cross-talk. This CDM system achieves energy resolutions of between 2.3 eV and 3.0 eV FWHM at 5.9 keV.
• ### Advanced code-division multiplexers for superconducting detector arrays(1110.1608)

Multiplexers based on the modulation of superconducting quantum interference devices are now regularly used in multi-kilopixel arrays of superconducting detectors for astrophysics, cosmology, and materials analysis. Over the next decade, much larger arrays will be needed. These larger arrays require new modulation techniques and compact multiplexer elements that fit within each pixel. We present a new in-focal-plane code-division multiplexer that provides multiplexing elements with the required scalability. This code-division multiplexer uses compact lithographic modulation elements that simultaneously multiplex both signal outputs and superconducting transition-edge sensor (TES) detector bias voltages. It eliminates the shunt resistor used to voltage bias TES detectors, greatly reduces power dissipation, allows different dc bias voltages for each TES, and makes all elements sufficiently compact to fit inside the detector pixel area. These in-focal-plane code-division multiplexers can be combined with multi-gigahertz readout based on superconducting microresonators to scale to even larger arrays.
• ### The Atacama Cosmology Telescope: Sunyaev Zel'dovich Selected Galaxy Clusters at 148 GHz in the 2008 Survey(1010.1065)

June 23, 2011 astro-ph.CO
We report on twenty-three clusters detected blindly as Sunyaev-Zel'dovich (SZ) decrements in a 148 GHz, 455 square-degree map of the southern sky made with data from the Atacama Cosmology Telescope 2008 observing season. All SZ detections announced in this work have confirmed optical counterparts. Ten of the clusters are new discoveries. One newly discovered cluster, ACT-CL J0102-4915, with a redshift of 0.75 (photometric), has an SZ decrement comparable to the most massive systems at lower redshifts. Simulations of the cluster recovery method reproduce the sample purity measured by optical follow-up. In particular, for clusters detected with a signal-to-noise ratio greater than six, simulations are consistent with optical follow-up that demonstrated this subsample is 100% pure. The simulations further imply that the total sample is 80% complete for clusters with mass in excess of 6x10^14 solar masses referenced to the cluster volume characterized by five hundred times the critical density. The Compton y -- X-ray luminosity mass comparison for the eleven best detected clusters visually agrees with both self-similar and non-adiabatic, simulation-derived scaling laws.
• ### The Atacama Cosmology Telescope: Extragalactic Sources at 148 GHz in the 2008 Survey(1007.5256)

June 3, 2011 astro-ph.CO
We report on extragalactic sources detected in a 455 square-degree map of the southern sky made with data at a frequency of 148 GHz from the Atacama Cosmology Telescope 2008 observing season. We provide a catalog of 157 sources with flux densities spanning two orders of magnitude: from 15 to 1500 mJy. Comparison to other catalogs shows that 98% of the ACT detections correspond to sources detected at lower radio frequencies. Three of the sources appear to be associated with the brightest cluster galaxies of low redshift X-ray selected galaxy clusters. Estimates of the radio to mm-wave spectral indices and differential counts of the sources further bolster the hypothesis that they are nearly all radio sources, and that their emission is not dominated by re-emission from warm dust. In a bright (>50 mJy) 148 GHz-selected sample with complete cross-identifications from the Australia Telescope 20 GHz survey, we observe an average steepening of the spectra between 5, 20, and 148 GHz with median spectral indices of $\alpha_{\rm 5-20} = -0.07 \pm 0.06$, $\alpha_{\rm 20-148} = -0.39 \pm0.04$, and $\alpha_{\rm 5-148} = -0.20 \pm 0.03$. When the measured spectral indices are taken into account, the 148 GHz differential source counts are consistent with previous measurements at 30 GHz in the context of a source count model dominated by radio sources. Extrapolating with an appropriately rescaled model for the radio source counts, the Poisson contribution to the spatial power spectrum from synchrotron-dominated sources with flux density less than 20 mJy is $C^{\rm Sync} = (2.8 \pm 0.3) \times 10^{-6} \micro\kelvin^2$.
• ### The Atacama Cosmology Telescope (ACT): Beam Profiles and First SZ Cluster Maps(0907.0461)

Dec. 6, 2010 astro-ph.CO, astro-ph.IM
The Atacama Cosmology Telescope (ACT) is currently observing the cosmic microwave background with arcminute resolution at 148 GHz, 218 GHz, and 277 GHz. In this paper, we present ACT's first results. Data have been analyzed using a maximum-likelihood map-making method which uses B-splines to model and remove the atmospheric signal. It has been used to make high-precision beam maps from which we determine the experiment's window functions. This beam information directly impacts all subsequent analyses of the data. We also used the method to map a sample of galaxy clusters via the Sunyaev-Zel'dovich (SZ) effect, and show five clusters previously detected with X-ray or SZ observations. We provide integrated Compton-y measurements for each cluster. Of particular interest is our detection of the z = 0.44 component of A3128 and our current non-detection of the low-redshift part, providing strong evidence that the further cluster is more massive as suggested by X-ray measurements. This is a compelling example of the redshift-independent mass selection of the SZ effect.
• ### The Atacama Cosmology Telescope: A Measurement of the 600< ell <8000 Cosmic Microwave Background Power Spectrum at 148 GHz(1001.2934)

Aug. 25, 2010 astro-ph.CO
We present a measurement of the angular power spectrum of the cosmic microwave background (CMB) radiation observed at 148 GHz. The measurement uses maps with 1.4' angular resolution made with data from the Atacama Cosmology Telescope (ACT). The observations cover 228 square degrees of the southern sky, in a 4.2-degree-wide strip centered on declination 53 degrees South. The CMB at arcminute angular scales is particularly sensitive to the Silk damping scale, to the Sunyaev-Zel'dovich (SZ) effect from galaxy clusters, and to emission by radio sources and dusty galaxies. After masking the 108 brightest point sources in our maps, we estimate the power spectrum between 600 < \ell < 8000 using the adaptive multi-taper method to minimize spectral leakage and maximize use of the full data set. Our absolute calibration is based on observations of Uranus. To verify the calibration and test the fidelity of our map at large angular scales, we cross-correlate the ACT map to the WMAP map and recover the WMAP power spectrum from 250 < ell < 1150. The power beyond the Silk damping tail of the CMB is consistent with models of the emission from point sources. We quantify the contribution of SZ clusters to the power spectrum by fitting to a model normalized at sigma8 = 0.8. We constrain the model's amplitude ASZ < 1.63 (95% CL). If interpreted as a measurement of sigma8, this implies sigma8^SZ < 0.86 (95% CL) given our SZ model. A fit of ACT and WMAP five-year data jointly to a 6-parameter LCDM model plus terms for point sources and the SZ effect is consistent with these results.
• ### The Atacama Cosmology Telescope: The Receiver and Instrumentation(1007.0290)

July 2, 2010 astro-ph.CO, astro-ph.IM
The Atacama Cosmology Telescope was designed to measure small-scale anisotropies in the Cosmic Microwave Background and detect galaxy clusters through the Sunyaev-Zel'dovich effect. The instrument is located on Cerro Toco in the Atacama Desert, at an altitude of 5190 meters. A six-meter off-axis Gregorian telescope feeds a new type of cryogenic receiver, the Millimeter Bolometer Array Camera. The receiver features three 1000-element arrays of transition-edge sensor bolometers for observations at 148 GHz, 218 GHz, and 277 GHz. Each detector array is fed by free space mm-wave optics. Each frequency band has a field of view of approximately 22' x 26'. The telescope was commissioned in 2007 and has completed its third year of operations. We discuss the major components of the telescope, camera, and related systems, and summarize the instrument performance.
• ### CMB Observations with a Compact Heterogeneous 150 GHz Interferometer in Chile(astro-ph/0403137)

Jan. 27, 2005 astro-ph
We report on the design, first observing season, and analysis of data from a new prototype millimeter-wave interferometer, MINT. MINT consists of four 145 GHz SIS mixers operating in double-sideband mode in a compact heterogeneous configuration. The signal band is subdivided by a monolithic channelizer, after which the correlations between antennas are performed digitally. The typical receiver sensitivity in a 2 GHz band is 1.4 mK sqrt(s). MINT observed the cosmic microwave background (CMB) from the Chilean Altiplano. The site has a median nighttime atmospheric temperature of 9 K at zenith (exclusive of the CMB). Observations of Mars, Jupiter, and a telescope-mounted calibration source establish the system's phase and magnitude stability. MINT is the first CMB-dedicated interferometer to operate above 50 GHz. The same type of system can be used to probe the Sunyaev-Zel'dovich effect in galaxy clusters near the SZ null at 217 GHz. We present an analysis of sideband-separated, digitally sampled data recorded by the array. Based on 215 hours of data taken in late 2001, we set an upper limit on the CMB anisotropy in a band of width Delta ell=700 around ell=1540 of delta T < 105 microK (95% conf). Increased sensitivity can be achieved with more integration time, greater bandwidth, and more elements.