• Report of the first workshop to identify approaches and techniques in the domain of quantum sensing that can be utilized by future High Energy Physics applications to further the scientific goals of High Energy Physics.
• ### Signal Optimization with HV divider of MCP-PMT for JUNO(1803.03746)

March 10, 2018 hep-ex, physics.ins-det
The Jiangmen Underground Neutrino Observatory (JUNO) is proposed to determine the neutrino mass hierarchy using a 20 kiloton underground liquid scintillator detector (CD). One of the keys is the energy resolution of the CD to reach <3% at 1 MeV, where totally 15,000 MCP-PMT will be used. The optimization of the 20-inch MCP-PMT is very important for better detection efficiency and stable performance. In this work, we will show the study to optimize the MCP-PMT working configuration for charge measurement. Particularly, the quality of PMT signal is another key for high-precision neutrino experiments while most of these experiments are affected by the overshoot of PMT signal from the positive HV scheme. The overshoot coupled with positive HV which is troubling trigger, dead time and precise charge measurement, we have studied to control it to less than 1% of signal amplitude for a better physics measurement. In this article, on the one hand, the optimized HV divider ratio will be presented here to improve its collection efficiency; on the other hand, we will introduce the method to reduce the ratio of overshoot from 10% to 1%.
• ### A study of the new hemispherical 9-inch PMT(1801.02737)

Jan. 9, 2018 hep-ex, physics.ins-det
The new hemispherical photomultiplier tubes (PMTs) with 9 inch diameter from Hainan Zhanchuang Photonics Technology Co.,Ltd (HZC) have been studied. Narrow transit time spread (FWHM=2.35 ns) accompanied by small nonlinearity (750 photoelectrons at 5%) and high gain (1E7 ) with good single photoelectron (PE) resolution have been observed. 11 PMTs of this type are deployed and studied in the prototype detector for JUNO at IHEP, China.
• ### First results of the new endcap TOF commissioning at BESIII(1605.01780)

May 5, 2016 hep-ex, physics.ins-det
The upgrade of the current BESIII Endcap TOF (ETOF) is carried out with the Multi-gap Resistive Plate Chamber (MRPC) technology. The installation of the new ETOF has been finished in October 2015. The first results of the MRPCs commissioning at BESIII are reported in this paper.
• ### The upgrade system of BESIII ETOF with MRPC technology(1604.02701)

April 10, 2016 hep-ex, physics.ins-det
The Beijing Spectrometer III (BESIII) endcap Time-Of-Filght (ETOF) was proposed to upgrade with Multigap Resistive Plate Chamber (MRPC) technology to substitute the current ETOF of scintillator+PMT for extending time resolutin better than 80 ps and enhance the particle identification capability to satisfy the higher precision requirement of physics. The ETOF system including MRPC modules, front end electronics (FEE), CLOCK module, fast control boards and time to digital modules (TDIG), has been designed, constructed and done some experimental tests seperately. Aiming at examining the quality of entire ETOF system and training the operation of all participated parts, a cosmic ray test system was built at the laboratory and underwent about three months to guarantee performance. In this paper the results will be presented indicating that the entire ETOF system works well and satisfies the requirements of the upgrade.
• ### The cosmic ray test of MRPCs for the BESIII ETOF upgrade(1511.08959)

Jan. 20, 2016 hep-ex, physics.ins-det
In order to improve the particle identification capability of the Beijing Spectrometer III (BESIII),t is proposed to upgrade the current endcap time-of-flight (ETOF) detector with multi-gap resistive plate chamber (MRPC) technology. Aiming at extending ETOF overall time resolution better than 100ps, the whole system including MRPC detectors, new-designed Front End Electronics (FEE), CLOCK module, fast control boards and time to digital modules (TDIG), was built up and operated online 3 months under the cosmic ray. The main purposes of cosmic ray test are checking the detectors' construction quality, testing the joint operation of all instruments and guaranteeing the performance of the system. The results imply MRPC time resolution better than 100$ps$, efficiency is about 98$\%$ and the noise rate of strip is lower than 1$Hz/$($scm^{2}$) at normal threshold range, the details are discussed and analyzed specifically in this paper. The test indicates that the whole ETOF system would work well and satisfy the requirements of upgrade.
• The Jiangmen Underground Neutrino Observatory (JUNO), a 20 kton multi-purpose underground liquid scintillator detector, was proposed with the determination of the neutrino mass hierarchy as a primary physics goal. It is also capable of observing neutrinos from terrestrial and extra-terrestrial sources, including supernova burst neutrinos, diffuse supernova neutrino background, geoneutrinos, atmospheric neutrinos, solar neutrinos, as well as exotic searches such as nucleon decays, dark matter, sterile neutrinos, etc. We present the physics motivations and the anticipated performance of the JUNO detector for various proposed measurements. By detecting reactor antineutrinos from two power plants at 53-km distance, JUNO will determine the neutrino mass hierarchy at a 3-4 sigma significance with six years of running. The measurement of antineutrino spectrum will also lead to the precise determination of three out of the six oscillation parameters to an accuracy of better than 1\%. Neutrino burst from a typical core-collapse supernova at 10 kpc would lead to ~5000 inverse-beta-decay events and ~2000 all-flavor neutrino-proton elastic scattering events in JUNO. Detection of DSNB would provide valuable information on the cosmic star-formation rate and the average core-collapsed neutrino energy spectrum. Geo-neutrinos can be detected in JUNO with a rate of ~400 events per year, significantly improving the statistics of existing geoneutrino samples. The JUNO detector is sensitive to several exotic searches, e.g. proton decay via the $p\to K^++\bar\nu$ decay channel. The JUNO detector will provide a unique facility to address many outstanding crucial questions in particle and astrophysics. It holds the great potential for further advancing our quest to understanding the fundamental properties of neutrinos, one of the building blocks of our Universe.
• ### The high-speed after pulse measurement system for PMT(1404.6125)

April 24, 2014 hep-ex, physics.ins-det
A system employing a desktop FADC has been developed to investigate the features of 8 inches Hamamatsu PMT. The system stands out for its high-speed and informative results as a consequence of adopting fast waveform sampling technology. Recording full waveforms allows us to perform digital signal processing, pulse shape analysis, and precision timing extraction. High precision after pulse time and charge distribution characteristics are presented in this manuscript. Other photomultipliers characteristics, such as dark rate and transit time spread, can also be obtained by exploiting waveform analysis using this system.
• ### A GEANT4 Simulation Studyof BESIII endcap TOF Upgrade(1208.3049)

Aug. 15, 2012 hep-ex, physics.ins-det
A GEANT4-based Monte-Carlo model is developed to study the performance of Endcap Time-Of-Flight (ETOF) at BESIII. It's found that the multiple scattering effects, mainly from the materials at the MDC endcap, can cause multi-hit on the ETOF's readout cell and significantly influence the timing property of ETOF. Multi-gap Resistive Plate Chamber (MRPC) with a smaller readout cell structure is more suitable for ETOF detector due to significantly reduced multi-hit rate, from 71.5% for currently-used scintillator-based ETOF to 21.8% or 16.7% for MRPC-based ETOF, depending on the readout pad size used. The timing performance of a MRPC ETOF is also improved. These simulation results suggest and guide an ETOF upgrade effort at BESIII.
• ### Leakage Tests of the Stainless Steel Vessels of the Antineutrino Detectors in the Daya Bay Reactor Neutrino Experiment(1203.0346)

March 2, 2012 hep-ex, nucl-ex, physics.ins-det
The antineutrino detectors in the Daya Bay reactor neutrino experiment are liquid scintillator detectors designed to detect low energy particles from antineutrino interactions with high efficiency and low backgrounds. Since the antineutrino detector will be installed in a water Cherenkov cosmic ray veto detector and will run for 3 to 5 years, ensuring water tightness is critical to the successful operation of the antineutrino detectors. We choose a special method to seal the detector. Three leak checking methods have been employed to ensure the seal quality. This paper will describe the sealing method and leak testing results.